Instruction Manual Professional Weather Center model 01301CCDI / 393242 CONTENTS Unpacking Instructions...................2 Package Contents..........................2 Product Registration.......................2 Features & Benefits: 3-in-1.............3 Features: Display Unit....................4 Setup...........................................6 3-in-1 Sensor Setup.......................6 Display Unit Setup.........................7 Set Language, Time, Date & Units...8 Placement Guidelines..................9 3-in-1 Sensor Installation...........10 Using the Weather Center..........11 Troubleshooting.........................13 Care & Maintenance.................15 Calibration...............................15 Specifications............................16 FCC Information........................16 Customer Support.....................17 Warranty..................................17 Questions? Contact Customer Support at (877) 221-1252 or visit www.AcuRite.com. SAVE THIS MANUAL FOR FUTURE REFERENCE. Congratulations on your new AcuRite product. To ensure the best possible product performance, please read this manual in its entirety and retain it for future reference. Unpacking Instructions Remove the protective film that is applied to the LCD screen prior to using this product. Locate the tab and peel off to remove. Package Contents 1. Display unit with tabletop stand 2. 3-in-1 sensor 3. Sensor mounting bracket 4. Mounting hardware 5. Power adapter 6. Instruction Manual IMPORTANT PRODUCT MUST BE REGISTERED TO RECEIVE WARRANTY SERVICE PRODUCT REGISTRATION Register online to receive 1 year warranty protection www.AcuRite.com ►Register a Product 2 Features & Benefits 4 1 6 2 5 3 3-in-1 SENSOR 5. M ounting Hardware Includes anchors and screws. Qty Diameter Length 4 #4 1” 1 #6 3/8” 1. T emperature & Humidity Sensors (internal) 2. Release Lever for Battery Access 3. Mounting Point 6. M ounting Bracket 4. Wind Speed Anemometer 7. A -B-C Switch (not shown) ID code that must match display’s A-B-C switch to ensure units synchronize. 3 Features & Benefits 1 2 3 4 5 6 7 Display Unit 8 28 27 26 25 24 23 22 9 10 11 12 13 14 15 21 20 19 18 17 16 29 34 30 31 32 33 35 4 DISPLAY UNIT 19. – Button 1. Current Outdoor Temperature Arrow icon indicates direction 20. For setup preferences. Button Press to change the Weather Select (#23) category being displayed. temperature is trending. 2.Sensor Low Battery Indicator 21. Barometric Pressure 3.Average Wind Speed Average of all speeds over last 30 minutes or calibrated time interval. rrow icon indicates direction A pressure is trending. 22.Learning Mode Icon Disappears after weather forecast self-calibration is complete. 4.Backlight Button 5.Current Wind Speed 23.Weather Select Displays heat index, dew point or wind 6.Wind Gust Icon 7. Peak Wind Speed 8.Current Indoor Temperature 24.Language Highest speed from last 30 minutes or calibrated time interval. For setup preferences. Arrow icon indicates direction temperature is trending. 25.AUTO DIM Indicator 9.Display Low Battery Indicator 26.Current Outdoor Humidity 10. Current Indoor Humidity Arrow icon indicates direction temperature is trending. Arrow icon indicates direction humidity is trending. 27. Low Wind Speed 11.Clock Lowest speed from past 30 minutes or calibrated time interval. 12.Date 13. chill, daily, monthly, all time high & low records indoor/outdoor temperatures and humidity and high for wind speed. 28.Outdoor Sensor Signal Strength and LIVE Icons indicates historical data is being displayed. LIVE icon is shown while exiting function. BACK OF DISPLAY UNIT 29.Integrated Hang Hole 14. 12 to 24 Hour Weather Forecast Self-Calibrating Forecasting pulls data from the outdoor sensor to generate your personal forecast. For easy wall mounting 30. Plug-in for Power Adapter 31. Battery Compartment 15. Multi-Variable History Chart 32.A-B-C Switch 16. 33.CLEAR RECORD/RESET Button ID code that must match sensor’s A-B-C switch to ensure units synchronize. Displays the past 12 hours of data for pressure, temperature and humidity. Button Press to clear record currently being viewed. Press and HOLD for full reset to factory defaults. For displaying historical data. 17. + Button 18. For setup preferences. 34. Battery Compartment Cover Button 35.Power adapter For setup preferences. 5 SETUP 3-IN-1 Sensor Setup 1 Set the A-B-C Switch 1. Press the release lever and pull off the lower housing of the sensor. The A-B-C switch is located inside the battery compartment. It can be set to A, B or C. However, you must select the same letter choices for both the sensor and the display unit in order for the units to synchronize. 2. Remove the battery compartment cover. Take note of the A-B-C switch setting inside the battery compartment. 2 Install or Replace Batteries 3. Insert 4 x AA batteries into the battery compartment, as shown. Follow the polarity (+/-) diagram in the battery compartment. AcuRite recommends high quality alkaline or lithium batteries for the best product performance. Heavy duty or rechargeable batteries are not recommended. 4. Replace the battery cover and lower housing of the sensor. The 3-in-1 sensor requires lithium batteries in low temperature conditions. Cold temperatures can cause alkaline batteries to function improperly. Use lithium batteries in the 3-in-1 sensor for temperatures below -4ºF / -20ºC. ABC 1 A-B-C Switch set to match display unit 2 6 Display Unit Setup 1 Set the A-B-C Switch 2 Install or Replace Batteries L ocate the A-B-C switch inside the battery compartment. Set the A-B-C switch to A, B or C. You must select the same letter choices for both the display unit and the sensor in order for the units to synchronize. 1. Remove the battery compartment cover. 2. Insert 6 x AA alkaline batteries into the battery compartment, as shown. Follow the polarity (+/-) diagram in the battery compartment. 3. Replace the battery cover. 3 Plug Power Adapter into Electric Outlet 3 Plug in Power Adapter 1 2 Install Batteries 6 AA Batteries (optional) A-B-C Switch set to match sensor ABC IMPORTANT: Batteries are a backup power source to preserve records in the event of a power outage. Power adapter is recommended as the primary power source to enjoy the full functionality of this product. PLEASE DISPOSE OF OLD OR DEFECTIVE BATTERIES IN AN ENVIRONMENTALLY SAFE WAY AND IN ACCORDANCE WITH YOUR LOCAL LAWS AND REGULATIONS. BATTERY SAFETY: Clean the battery contacts and also those of the device prior to battery installation. Remove batteries from equipment which is not to be used for an extended period of time. Follow the polarity (+/-) diagram in the battery compartment. Promptly remove dead batteries from the device. Dispose of used batteries properly. Only batteries of the same or equivalent type as recommended are to be used. DO NOT incinerate used batteries. DO NOT dispose of batteries in fire, as batteries may explode or leak. DO NOT mix old and new batteries or types of batteries (alkaline/standard). DO NOT use rechargeable batteries. DO NOT recharge non-rechargeable batteries. DO NOT short-circuit the supply terminals. 7 Set the Time, Date & Units Press the “ ” button, located on the front of the display unit, to enter SET MODE. Once in set mode, the preference you are currently setting will blink on the display. To adjust the currently selected (flashing) item, press and release the “+“ or “--“ buttons (press and HOLD to fast adjust). To save your adjustments, press and release the “ ” button again to adjust the next preference. The preference set order is as follows: LANGUAGE (English or French) CLOCK HOUR CLOCK MINUTE CALENDAR MONTH CALENDAR DATE CALENDAR YEAR TEMPERATURE UNITS (ºF or ºC) WIND UNITS (MPH or KM/H) PRESSURE UNITS (inHg or hPa) You will automatically exit SET MODE if no buttons are pressed for 10 seconds. Enter setup mode at any time by pressing the “ ” button. Display Backlight Settings This weather station’s color display features three different lighting settings: High (100%) brightness, Medium (60%) brightness and Low (30%) brightness Using battery power alone, the backlight is available momentarily for 10 ” button. seconds by pressing the “ When display is powered with the power adapter, backlight remains on at 100% brightness. Press “ ” button once to dim to 60% brightness; press again to dim to 30%, press a 3rd time to enter “AUTO DIM” mode. AUTO DIM appears above time and date. AUTO DIM MODE: Automatically adjusts display brightness based on time of day & year. MARCH 11- NOVEMBER 4 6:00 a.m. - 9:00 p.m. = 100% brightness 9:01 p.m. - 5:59 a.m.= 30% brightness NOVEMBER 5 - MARCH 10 7:30 a.m. - 7:00 p.m.= 100% brightness 7:01 p.m. - 7:29 a.m. = 30% brightness 8 Placement for Maximum Accuracy AcuRite sensors are sensitive to surrounding environmental conditions. Proper placement of both the display unit and the sensor are critical to the accuracy and performance of this product. Display Unit Placement Place the display unit in a dry area free of dirt and dust. To ensure accurate temperature measurement, place out of direct sunlight and away from heat sources or vents. Display unit stands upright for tabletop use or is wall-mountable. 3-in-1 Sensor Placement The 3-in-1 sensor is designed to remain outdoors all year long. Choose an open location with no obstructions above or around the sensor for the most accurate measurements. Important Placement Guidelines Display unit and sensor must be within 330 feet (100 m) of each other. MAXIMIZE WIRELESS RANGE P lace units away from large metallic items, thick walls, metal surfaces, or other objects that may limit wireless communication. PREVENT WIRELESS INTERFERENCE P lace both units at least 3 feet (.9 m) away from electronic devices (TV, computer, microwave, radio, etc.). LOCATE AWAY FROM HEAT SOURCES Position sensor away from heaters, air conditioners, chimneys, exhaust vents, asphalt and concrete (surfaces that radiate heat). LOCATE AWAY FROM HUMIDITY SOURCES Avoid installing the sensor near pools, spas, or other bodies of water. Water sources may impact humidity accuracy. LOCATE AWAY FROM SPRINKLER HEADS DO NOT install the sensor where it will be sprayed by a sprinkler system. This may force water inside the sensor. LOCATE AWAY FROM WIND OBSTRUCTIONS DO NOT mount the sensor with obstructions around it. Consider a location that is a wide open area, with few structures around to ensure accurate wind speed measurement. Visit us online to view installation photos and video, or learn more about AcuRite technology: www.AcuRite.com/3in1 9 View Video 3-in-1 Sensor Installation Guidelines INSTALLATION HEIGHT Mount the sensor at least 5 feet (1.5 meters) off the ground (higher is better for wind measurement) in an open area. Secure to fence post, 2”x4” wood, 3/4” pole, etc. (not included) LEVEL INSTALLATION Use a bubble level (not included) to ensure level installation for accurate wind measurement. Top or Side Mount Top Mount 1. Insert the mounting base into the hole on the bottom of the sensor. Side Mount 2. Use one of the screws included in the hardware bag to fasten the lower sensor housing to the mounting base. 3. Fasten mounting base to a post or railing using 2 screws. Pole Mount 1. Insert pole (we recommend 3/4” schedule 40 PVC / 25.8 mm O.D., not included) or similar item all the way into the hole on the bottom of the sensor. Note: Mounting base is not used. 2. Use one of the screws included in the hardware bag to fasten the lower sensor housing securely to the pole. Pole Mount Do not install during electrical storms. Do not place pole or sensor near electrical lines or electrical equipment. The 3-in-1 sensor is now ready to use. Clear Data Collected During Installation When installation is complete, clear erroneous data from the display unit by pressing the “CLEAR RECORD/RESET” button for each category, on the back of the display inside the battery compartment. Basic Setup is Complete The 3-in-1 sensor will now synchronize with the display unit. It may take a few minutes for synchronization to complete. If both or one of the units appear to be functioning improperly, please refer to the troubleshooting section. 10 OPERATION Using The Weather Center Learning Mode Self-Calibrating Forecasting use a unique algorithm to analyze changes in pressure over a time period (called Learning Mode) to determine your altitude. After 14 days, the Learning Mode icon disappears from the display screen. At this point, the self-calibrated pressure is tuned in to your location and the unit is ready for superior weather prediction. Weather Forecast AcuRite’s patented Self-Calibrating Forecasting provides your personal forecast of weather conditions for the next 12 to 24 hours by collecting data from the sensor in your backyard. It generates a forecast with pinpoint accuracy - personalized for your exact location. STORMY & WINDY SNOW LIKELY (flashing=stormy) SNOW / RAIN MIX LIKELY RAIN LIKELY MOSTLY CLOUDY (flashing rain/snow) View the complete list of icons at www.AcuRite.com/acurite-icons Multi-Variable History Chart The multi-variable history chart lets you monitor the change in conditions over the past 12 hour time period (-12, -6, -3, -2, -1, 0). The chart automatically toggles between barometric pressure, outdoor temperature and outdoor humidity readings. Wind Speed The Wind Speed area of the display features the CURRENT speed, as well as the LOW, AVERAGE and PEAK wind speeds. PEAK, AVERAGE and LOW wind speeds are based on the past 30 minute time frame by default. The time frame can be adjusted during the calibration process (page 15). Barometric Pressure Subtle variations in barometric pressure greatly impact the weather. This weather center displays the current pressure with an arrow icon to indicate the direction the pressure is trending (FALLING, STEADY, or RISING). The multi-variable history chart lets you monitor the change in pressure over time. 11 Weather Select Features weather data groups shown at right. To change “Weather Select” category, press the “ “ button on the front of the display. When outdoor/ indoor temperature or humidity is chosen, the display cycles through the category’s records in this order: 24 Hour Change (data change since 24 hours ago) Today’s HIGH Today’s LOW Month’s HIGH Month’s LOW All Time HIGH All Time LOW When Wind Speed is chosen, the display cycles through the following records: Today’s HIGH Month’s HIGH All Time HIGH To clear a record, view one of the values listed above and press the “CLEAR RECORD/RESET” button, located on the back of the unit in the battery compartment. Dashes display to confirm the record has been cleared. REPLAY Historical Data This weather station stores historical data that can be viewed in Replay Mode. Data is recorded for the past 48 hours in 30 minute intervals (on the hour and half hour). Historical data is shown with a time and date stamp. Press the “ ” button to enter Replay Mode. The “ shown on the display below the time. “ icon and options will be REPLAY MODE OPTIONS: • Press the “ ” ( ) button to view 48-hour historical data. First, data is shown that was recorded 48 hours ago, and then the display cycles through the historical data in 30 minute intervals up to the current time. To exit Replay Mode and view live data, press the “ ” button • Press the “-” ( ) button to view historical data in 30 minute intervals previous to the current “LIVE” data. • While viewing historical data, press the “+” ( by a 30 minute interval. ) button to advance the data shown You will automatically exit Replay Mode if no buttons are pressed for 10 seconds. To manually exit Replay Mode at any time, press the “ ” button. The display may show or flash the “LIVE” icon to confirm you are exiting Replay Mode and returning to the current (“LIVE”) data view. 12 Troubleshooting Problem No outdoor sensor reception no bars Possible Solution • Relocate the display unit and/or the 3-in-1 sensor. The units must be within 330 ft (100 m) of each other. • Make sure both units are placed at least 3 feet (.9 m) away from electronics that may interfere with the wireless communication (such as TVs, microwaves, computers, etc). • Use standard alkaline batteries (or lithium batteries in sensor when temperature is below -4ºF/-20ºC). Do not use heavy duty or rechargeable batteries. NOTE: It may take up to 20 minutes for display unit and sensor to synchronize after batteries are replaced. • Synchronize the units: 1. Bring both the sensor and display unit indoors and remove at least one battery from each. 2. Make sure the A-B-C switch on both units match. 3. Reinstall the battery in the outdoor sensor. 4. Reinstall battery into the display unit. 5. Let the units sit within a couple feet of each other for about 20 minutes to gain a strong connection. Outdoor temperature is flashing or showing dashes Flashing of the outdoor temperature may be an indication of wireless interference: • Make sure the A-B-C switch in the battery compartments of both the display unit and sensor are switched to the same letter. You may choose A, B or C; but both units must match to sync up. Sometimes changing to a different channel can help. Inaccurate forecast • Weather Forecast icon predicts conditions for the next 12 to 24 hours, not current conditions. • Has Learning Mode icon disappeared from the display unit? Learning Mode must complete before forecast and pressure will be accurate. • Allow unit to run continuously for 33 days. Battery removal or resetting the display unit will restart Learning Mode. After 14 days, forecast should be fairly accurate, however Learning Mode calibrates for a total of 33 days. 13 Troubleshooting Problem Possible Solution Inaccurate temperature or humidity • Make sure both the display unit and 3-in-1 sensor are placed away from any heat sources or vents (see page 9). • Make sure both units are positioned away from moisture sources (see page 9). • Make sure 3-in-1 sensor is mounted at least 5 ft off of the ground. • Calibrate indoor and outdoor temperature and humidity (see page 15). Inaccurate wind readings • What is wind reading being compared to? Pro weather stations are typically mounted at 30 ft high or more. Make sure to compare data using a sensor positioned at the same mounting height. • Check location of the sensor. Ensure it’s mounted a minimum of 5 ft in the air with no obstructions around it (within several feet). •Ensure wind cups are spinning freely. If they hesitate or stop try lubricating with graphite powder or spray lubricant. Display screen not working • Check that the batteries are installed correctly. Batteries may need to be replaced. • Reset the display by pressing and HOLDING the “CLEAR RECORD/RESET” button, located on the back of the display unit in the battery compartment. Date and time will need to be entered after a reset. If your AcuRite product does not operate properly after trying the troubleshooting steps, visit www.AcuRite.com or call (877) 221-1252 for assistance. 14 Care & Maintenance Display Unit Care Clean with a soft, damp cloth. Do not use caustic cleaners or abrasives. Keep away from dust, dirt and moisture. Clean ventilation ports regularly with a gentle puff of air. 3-in-1 Sensor Care Clean the Sensor Clean with a soft, damp cloth. Do not use caustic cleaners or abrasives. Insect Prevention Insects may cause obstructions and interrupt data by nesting in or on the 3-in-1 sensor. To limit this problem, spray sensor housing with a household insect repellent. Note: Insect repellent should not be sprayed on the internal components or into the sensor. Consult insect repellent instructions prior to use. Snow & Freezing Weather The 3-in-1 sensor will not be damaged by freezing conditions. Clean the Wind Anemometer Remove foreign matter from the outside of the sensor for free movement of the wind anemometer. If needed, use a small amount of spray lubricant, clear silicone or graphite powder on the anemometer for improved movement. Note: Spray lubricant, clear silicone or graphite powder should not be sprayed on the internal components or into the sensor. Calibration The indoor / outdoor temperature and humidity readings, wind speed interval, and barometric pressure can be calibrated on the display unit to improve accuracy. Calibration can improve accuracy when sensor placement or environmental factors impact the data accuracy. 1. To access calibration mode, press AND HOLD the “-“, “ for at least 5 seconds. ” and “+“ buttons simultaneously 2. To adjust the currently selected (flashing) item, press and release the “-“ or “+“ buttons to calibrate the data value higher or lower from the actual reading. 3. To save your adjustments, press and release the “ ” button to adjust the next preference. The “ ” icon will remain illuminated next to calibrated values. The preference set order is as follows: OUTDOOR TEMPERATURE OUTDOOR HUMIDITY INDOOR TEMPERATURE INDOOR HUMIDITY WIND SPEED INTERVAL (30, 60, 90, 120, 150 or 180 minutes) BAROMETRIC PRESSURE (must be set to MANUAL mode to calibrate)* *To change from AUTO to MANUAL pressure mode and vice versa, press AND HOLD the “ ” button for at least 10 seconds. The display indicates the current pressure mode selected,“AUTO” or “MANUAL”. After 10 seconds of inactivity, the display will save the adjustments and exit calibration mode. Note: Calibration preferences will be erased if the display is reset or if batteries are removed and the power adapter is unplugged. 15 Specifications TEMPERATURE RANGE Outdoor: -40ºF to 158ºF; -40ºC to 70ºC Indoor: 32ºF to 122ºF; 0ºC to 50ºC HUMIDITY RANGE Outdoor: 1% to 99% Indoor: 1% to 99% WIND SPEED 0 to 99 mph; 0 to 159 kph WIRELESS RANGE 330 ft / 100 m depending on home construction materials OPERATING FREQUENCY 433 mhz POWER Display: 5 V, 100mA power adapter 6 x AA alkaline batteries Sensor: 4 x AA alkaline or lithium batteries Wind Speed: 18 second updates DATA REPORTING Outdoor temperature & humidity: 18 second updates Indoor temperature & humidity: 60 second updates FCC Information This device complies with part 15 of FCC rules. Operation is subject to the following two conditions: 1- This device may NOT cause harmful interference, and 2- This device must accept any interference received, including interference that may cause undesired operation. WARNING: Changes or modifications to this unit not expressly approved by the party responsible for compliance could void the user’s authority to operate the equipment. NOTE: This equipment has been tested and found to comply with the limits for a Class B digital device, pursuant to Part 15 of the FCC rules. These limits are designed to provide reasonable protection against harmful interference in a residential installation. This equipment generates, uses and can radiate radio frequency energy and, if not installed and used in accordance with the instructions, may cause harmful interference to radio communications. However, there is no guarantee that interference will not occur in a particular installation. If this equipment does cause harmful interference to radio or television reception, which can be determined by turning the equipment off and on, the user is encouraged to try to correct the interference by one or more of the following measures: • Reorient or relocate the receiving antenna. • Increase the separation between the equipment and the receiver. • Connect the equipment into an outlet on a circuit different from that to which the receiver is connected. • Consult the dealer or an experienced radio/TV technician for help. NOTE: The manufacturer is not responsible for any radio or TV interference caused by unauthorized modifications to this equipment. Such modifications could void the user authority to operate the equipment. This device complies with Industry Canada licence-exempt RSS standard(s). Operation is subject to the following two confitions: (1) This device may not cause interference, and (2) This device must accept any interference received, including interference that may cause undesired operation of the device. 16 Customer Support AcuRite customer support is committed to providing you with best-in-class service. For assistance, please have the model number of this product available and contact us in any of the following ways: (877) 221-1252 firstname.lastname@example.org 24/7 support at www.AcuRite.com ► Installation Videos ► Register your Product ► Instruction Manuals ► Support User Forum ► Replacement Parts ► Submit IMPORTANT Feedback & Ideas PRODUCT MUST BE REGISTERED TO RECEIVE WARRANTY SERVICE PRODUCT REGISTRATION Register online to receive 1 year warranty protection www.AcuRite.com ►Register a Product Limited One Year Warranty At AcuRite, we proudly uphold our commitment to quality technology. Chaney Instrument Co. warrants that all products it manufactures to be of good material and workmanship, and to be free of defects when properly installed and operated for a period of one year from the date of purchase. it manufactures which have received normal wear and tear, been damaged (including by acts of nature), tampered, abused, improperly installed, damaged in shipping, or repaired or altered by others than authorized representatives of Chaney. The above-described warranty is expressly in lieu of all other warranties, express or implied, and all other warranties are hereby expressly disclaimed, including without limitation the implied warranty of merchantability and the implied warranty of fitness for a particular purpose. Chaney expressly disclaims all liability for special, consequential or incidental damages, whether arising in tort or by contract from any breach of this warranty. Some states do not allow the exclusion or limitation of incidental or consequential damages, so the above limitation or exclusion may not apply to you. Chaney further disclaims all liability from personal injury relating to its products to the extent permitted by law. By acceptance of any of Chaney’s products, the purchaser assumes all liability for the consequences arising from their use or misuse. No person, firm or corporation is authorized to assume for Chaney any other liability in connection with the sale of its products. Furthermore, no person, firm or corporation is authorized to modify or waive the terms of this paragraph, and the preceding paragraph, unless done in writing and signed by a duly authorized agent of Chaney. This warranty gives you specific legal rights, and you may also have other rights which vary from state to state. We recommend that you visit us at www.AcuRite.com for the fastest way to register your product. However, product registration does not eliminate the need to retain your original proof of purchase in order to obtain warranty benefits. Chaney Instrument Co. warrants that all products it manufactures to be of good material and workmanship, and to be free of defects when properly installed and operated for a period of one year from the date of purchase. Remedy for breach of this warranty is limited to repair or replacement of the defective item(s). Any product which, under normal use and service, is proven to breach the warranty contained herein within ONE YEAR from date of sale will, upon examination by Chaney, and at its sole option, be repaired or replaced by Chaney. Transportation costs and charges for returned goods shall be paid for by the purchaser. Chaney hereby disclaims all responsibility for such transportation costs and charges. This warranty will not be breached, and Chaney will give no credit for products For in-warranty claims: Chaney Instrument Co. | 965 Wells St. | Lake Geneva, WI 53147 17 Weather Stations Temperature & Humidity Weather Alert Radio Kitchen Thermometers & Timers Clocks It’s more than accurate, it’s AcuRite offers an extensive assortment of precision instruments, designed to provide you with information you can depend on to Plan your day with confidence™. www.AcuRite.com Printed in China 01301CCDI INST 061014 ©Chaney Instrument Co. All rights reserved. AcuRite is a registered trademark of the Chaney Instrument Co., Lake Geneva, WI 53147. All other trademarks and copyrights are the property of their respective owners. AcuRite uses patented technology. Visit www.AcuRite.com/patents for details.
* Your assessment is very important for improving the work of artificial intelligence, which forms the content of this project