Instruction Manual Wireless Thermometer model 00411/00609SBDIA/00611A3 CONTENTS Unpacking Instructions............ 2 Package Contents................... 2 Product Registration................ 2 Features & Benefits.................. 3 Setup..................................... 4 Install or Replace Batteries...... 4 Placement Guidelines.............. 5 Using the Thermometer............ 6 Troubleshooting...................... 7 Care & Maintenance............... 8 Specifications.......................... 8 FCC Information..................... 8 Customer Support................... 9 Warranty............................... 9 Questions? Contact Customer Support at (877) 221-1252 or visit www.AcuRite.com SAVE THIS MANUAL FOR FUTURE REFERENCE. Congratulations on your new AcuRite product. To ensure the best possible product performance, please read this manual in its entirety and retain it for future reference. Unpacking Instructions Remove the protective film that is applied to the LCD screen prior to using this product. Locate the tab and peel off to remove. Package Contents 1. Display unit with tabletop stand 2. Outdoor sensor 3. Instruction Manual IMPORTANT PRODUCT MUST BE REGISTERED TO RECEIVE WARRANTY SERVICE PRODUCT REGISTRATION Register online to receive 1 year warranty protection www.AcuRite.com ►Register a Product 2 Features & Benefits 1 14 2 15 3 13 12 4 5 6 11 10 7 DISPLAY UNIT 8 16 9 9. Current Indoor Humidity Arrow icon indicates the direction the humidity is trending 1. Current Outdoor Temperature Arrow icon indicates the direction the temperature is trending 10. High Indoor Temperature and Humidity Record 2. High Outdoor Temperature Record 11. Display Low Battery Indicator 3. Low Outdoor Temperature Record 12. Outdoor Sensor Low Battery Indicator 4. High Outdoor Humidity Record 5. Current Outdoor Humidity Arrow icon indicates the direction the humidity is trending. 13. Outdoor Sensor Signal Strength 6. Low Outdoor Humidity Record 14.Integrated Hanger for easy placement OUTDOOR SENSOR 7. Current Indoor Temperature Arrow icon indicates the direction the temperature is trending 15.Wireless Signal Indicator Flashes when data is being sent to the display unit 8. Low Indoor Temperature and Humidity Record 16.Battery Compartment Cover 3 SETUP Main Unit Detachable Stand Included with the main unit is a detachable stand for tabletop placement. Install or Replace Batteries AcuRite recommends high quality alkaline batteries for the best product performance. Heavy duty or rechargeable batteries are not recommended. The outdoor sensor requires lithium batteries in low temperature conditions. Cold temperatures can cause alkaline batteries to function improperly. Use lithium batteries in the outdoor sensor for temperatures below -4ºF / -20ºC. Outdoor Sensor 1. S lide off the battery compartment cover. 2. Insert 2 x AA batteries into the battery compartment, as shown. Follow the polarity (+/-) diagram in the battery compartment. 3. R eplace the battery cover. Display Unit 1. S lide off the battery compartment cover. 2. Insert 2 x AA batteries into the battery compartment, as shown. Follow the polarity (+/-) diagram in the battery compartment. 3. R eplace the battery cover. Select Degrees Fahrenheit or Celsius To select between degrees Fahrenheit (ºF) or Celsius (ºC) temperature units, press the “ºF/ºC” button located on the back of display. PLEASE DISPOSE OF OLD OR DEFECTIVE BATTERIES IN AN ENVIRONMENTALLY SAFE WAY AND IN ACCORDANCE WITH YOUR LOCAL LAWS AND REGULATIONS. BATTERY SAFETY: Clean the battery contacts and also those of the device prior to battery installation. Remove batteries from equipment which is not to be used for an extended period of time. Follow the polarity (+/-) diagram in the battery compartment. Promptly remove dead batteries from the device. Dispose of used batteries properly. Only batteries of the same or equivalent type as recommended are to be used. DO NOT incinerate used batteries. DO NOT dispose of batteries in fire, as batteries may explode or leak. DO NOT mix old and new batteries or types of batteries (alkaline/standard). DO NOT use rechargeable batteries. DO NOT recharge non-rechargeable batteries. DO NOT short-circuit the supply terminals. 4 Placement for Maximum Accuracy AcuRite sensors are sensitive to surrounding environmental conditions. Proper placement of both the display unit and outdoor sensor are critical to the accuracy and performance of this product. Display Unit Placement Place the display unit in a dry area free of dirt and dust. Display unit stands upright for tabletop use or is wall-mountable. Outdoor Sensor Placement Sensor must be placed outside to observe outdoor conditions. Sensor is water resistant and is designed for general outdoor use, however, to extend its life place the sensor in an area protected from direct weather elements. Hang the sensor using the integrated hang holes or hanger, or by using string (not included) to hang it from a suitable location, like a well covered tree branch. The best location is 4 to 8 feet above the ground with permanent shade and plenty of fresh air to circulate around the sensor. Important Placement Guidelines •To ensure accurate temperature measurement, place units out of direct sunlight and away from heat sources or vents. •Display unit and outdoor sensor must be within 165 ft (50 m) of each other. •To maximize wireless range, place units away from large metallic items, thick walls, metal surfaces, or other objects that may limit wireless communication. •To prevent wireless interference, place both units at least 3 ft (.9 m) away from electronic devices (TV, computer, microwave, radio, etc.) Setup is Complete The sensor will now synchronize with the display unit. It may take a few minutes for synchronization to complete. Please refer to the troubleshooting section of this manual if anything appears to be functioning improperly. (165 feet maximum) (50 meters) 5 OPERATION Using the Thermometer High/Low Records High an low records reflect the minimum and maximum temperature recorded since the unit was powered on, since the batteries were changed, or since it was manually reset (whichever was most recent). To view the LOW temperature record, press and release the “ ” icon appears on the display next to the low records. ” button. The “ To view the HIGH temperature record, press and release the “ “ ” icon appears on the display next to the high records. ” button. The To reset the LOW records, press AND HOLD the “ ” button while viewing the low records. Dashes will display to confirm all low values have been cleared. To reset the HIGH records, press AND HOLD the “ ” button while viewing the high records. Dashes will display to confirm all high values have been cleared. 6 Troubleshooting No outdoor sensor reception no bars If the wireless signal indicator shows no bars: •R elocate the display unit and/or the outdoor sensor. The units must be within 165 ft (50 m) of each other. •M ake sure both units are placed at least 3 feet (.9 m) away from electronics that may interfere with the wireless communication (such as TVs, microwaves, computers, etc). •U se standard alkaline batteries (or lithium batteries in sensor when temperature is below -4ºF/-20ºC). Do not use heavy duty or rechargeable batteries. NOTE: It may take up to 20 minutes for display unit and sensor to synchronize after batteries are replaced. Outdoor t emperature is flashing or showing dashes Flashing of the outdoor temperature is generally an indication of wireless interference. 1. B ring both the sensor and display unit indoors and remove at least one battery from each. 2. Reinstall the battery in the outdoor sensor. 3. Reinstall the battery in the display unit. 4. L et the units sit within a couple feet of each other for about 20 minutes to gain a strong connection. Inaccurate temperature •M ake sure both the display unit and sensor are placed out of direct sunlight and away from any heat sources or vents (see page 5). Display screen not working •C heck that the batteries are installed correctly. Batteries may need to be replaced. If your AcuRite product does not operate properly after trying the troubleshooting steps, visit www.AcuRite.com or call (877) 221-1252 for assistance. 7 Care & Maintenance Display Unit Care Clean with a soft, damp cloth. Do not use caustic cleaners or abrasives. Keep away from dust, dirt and moisture. Clean ventilation ports regularly with a gentle puff of air. Outdoor Sensor Care Clean with a soft damp cloth. Do not use caustic cleaners or abrasives. Specifications TEMPERATURE RANGE Outdoor: -40ºF to 158ºF; -40ºC to 70ºC Indoor: 32ºF to 122ºF; 0ºC to 50ºC HUMIDITY RANGE 1% to 99% RH WIRELESS RANGE 165 ft / 50 m depending on home construction materials WIRELESS FREQUENCY 433 MHz POWER Display: 2 x AA alkaline batteries Sensor: 2 x AA alkaline or lithium batteries DATA REPORTING Outdoor data: 31 second updates Indoor data: 60 second updates FCC Information This device complies with part 15 of FCC rules. Operation is subject to the following two conditions: 1- This device may NOT cause harmful interference, and 2- This device must accept any interference received, including interference that may cause undesired operation. WARNING: Changes or modifications to this unit not expressly approved by the party responsible for compliance could void the user’s authority to operate the equipment. NOTE: This equipment has been tested and found to comply with the limits for a Class B digital device, pursuant to Part 15 of the FCC rules. These limits are designed to provide reasonable protection against harmful interference in a residential installation. This equipment generates, uses and can radiate radio frequency energy and, if not installed and used in accordance with the instructions, may cause harmful interference to radio communications. However, there is no guarantee that interference will not occur in a particular installation. If this equipment does cause harmful interference to radio or television reception, which can be determined by turning the equipment off and on, the user is encouraged to try to correct the interference by one or more of the following measures: • Reorient or relocate the receiving antenna. • Increase the separation between the equipment and the receiver. • Connect the equipment into an outlet on a circuit different from that to which the receiver is connected. • Consult the dealer or an experienced radio/TV technician for help. NOTE: The manufacturer is not responsible for any radio or TV interference caused by unauthorized modifications to this equipment. Such modifications could void the user authority to operate the equipment. This device complies with Industry Canada licence-exempt RSS standard(s). Operation is subject to the following two conditions: (1) This device may not cause interference, and (2) This device must accept any interference received, including interference that may cause undesired operation of the device. 8 Customer Support AcuRite customer support is committed to providing you with best-inclass service. For assistance, please have the model number of this product available and contact us in any of the following ways: (877) 221-1252 firstname.lastname@example.org 24/7 support at www.AcuRite.com ► Installation Videos ► Register your Product ► Instruction Manuals ► Support User Forum ► Replacement Parts ► Submit IMPORTANT PRODUCT MUST BE REGISTERED TO RECEIVE WARRANTY SERVICE PRODUCT REGISTRATION Register online to receive 1 year warranty protection ►Register a Product www.AcuRite.com Limited One Year Warranty At AcuRite, we proudly uphold our commitment to quality technology. Chaney Instrument Co. warrants that all products it manufactures to be of good material and workmanship, and to be free of defects when properly installed and operated for a period of one year from the date of purchase. Feedback & Ideas warranty will not be breached, and Chaney will give no credit for products it manufactures which have received normal wear and tear, been damaged (including by acts of nature), tampered, abused, improperly installed, damaged in shipping, or repaired or altered by others than authorized representatives of Chaney. The above-described warranty is expressly in lieu of all other warranties, express or implied, and all other warranties are hereby expressly disclaimed, including without limitation the implied warranty of merchantability and the implied warranty of fitness for a particular purpose. Chaney expressly disclaims We recommend that you visit us at liability for special, consequential or incidental damages, www.AcuRite.com for the fastest way to register all whether arising in tort or by contract from any breach of this your product. However, product registration warranty. Some states do not allow the exclusion or limitation of incidental or consequential damages, so the above limitation does not eliminate the need to retain your or exclusion may not apply to you. Chaney further disclaims all original proof of purchase in order to obtain liability from personal injury relating to its products to the extent warranty benefits. permitted by law. By acceptance of any of Chaney’s products, Chaney Instrument Co. warrants that all products it manufactures the purchaser assumes all liability for the consequences arising from their use or misuse. No person, firm or corporation is to be of good material and workmanship, and to be free of authorized to assume for Chaney any other liability in connection defects when properly installed and operated for a period of with the sale of its products. Furthermore, no person, firm or one year from the date of purchase. Remedy for breach of this corporation is authorized to modify or waive the terms of this warranty is limited to repair or replacement of the defective paragraph, and the preceding paragraph, unless done in writing item(s). Any product which, under normal use and service, is and signed by a duly authorized agent of Chaney. This warranty proven to breach the warranty contained herein within ONE gives you specific legal rights, and you may also have other YEAR from date of sale will, upon examination by Chaney, rights which vary from state to state. and at its sole option, be repaired or replaced by Chaney. Transportation costs and charges for returned goods shall For in-warranty claims:Chaney Instrument Co. be paid for by the purchaser. Chaney hereby disclaims all 965 Wells St., Lake Geneva, WI 53147 responsibility for such transportation costs and charges. This 9 10 11 Weather Stations Temperature & Humidity Weather Alert Radio Kitchen Thermometers & Timers Clocks It’s more than accurate, it’s AcuRite offers an extensive assortment of precision instruments, designed to provide you with information you can depend on to Plan your day with confidence™. www.AcuRite.com Printed in China 00611CASBA2 INST 061214 © Chaney Instrument Co. All rights reserved. AcuRite is a registered trademark of the Chaney Instrument Co., Lake Geneva, WI 53147. All other trademarks and copyrights are the property of their respective owners. AcuRite uses patented technology. Visit www.AcuRite.com/patents for details.