1. Set the Meter Dear ForaCare Premium V10 System Owner: Thank you for choosing ForaCare Premium V10 System Blood Glucose Monitoring System to take control of your diabetes. Front View of Meter ForaCare Premium V10 System will help you keep track of your blood glucose levels. PremiumVlO The Quick Start User Guide will assist you in setting up the system and testing your blood glucose levels. Please read it carefully before use. For complete information on the system's use and features, please refer to the Owner's Manual. BLOOD GLUCOSE Monitoring System Test Slot Display Screen M Button Press the set button to start Press4lto change values. Hold setting. Year flashes first. down 41to increase the number more quickly. Press the set Data Port button to move on to the next step. ra Test Strip Vial Test Strips Lancets Control Solution Please note: Use only ForaCare Premium V10 System test strips with ForaCare Premium V10 System meter. rs '-"8 '' 'u [e. Rear View of Meter [8 dEL 12h Uol I ~ Clear Cap (for obtaining blood from places other than your fingertip) Date flashes. Follow Lancing Device Battery Cover Quick Start User Guide Test Strip . EB ru '"""3 fU·U PM [t!1J • IIL2 L to make changes. The sequence of settings is: (1) the date (2) 12h (3)time (4) memory deletion (5) volume (6) language For self-testing. Read instructions before use. For In vitro diagnostic use. Distributed by Fora Care Fora Care Inc. 810 Lawrence Dr, Suite 104, Newbury Park, CA 91320 USA Products made in Taiwan Lancing device, lancets and clear cap may not be included. If your lancing device differs from the one shown here, please refer to the manufacturer's manual to ensure proper usage. Toll Free: 1-888-307-8188 (8:30am-5:00pm PST, Mon.-Fri.) 1-866-469-2632 (24 Hours, 7 daysjweek) 315-4 124100-005 www.foracare.comjusa 2. Set the Lancing Device The meter will automatically turn off when the setting is complete. For memory deletion, please refer to the Owner's Manual for details. Please Note: Your lancing device may differ from the one shown in here. 3. Test Your Blood Glucose 3. Test Your Blood Glucose 4. Read Your Result Your blood glucose result will display the following messages: Appears Twist and pull off the cap of the lancing device. Insert a lancet into the lancet holder and push down firmly until it is fully secured. m Tear off the security label Wash and dry your hands (Tamper Seal). When opening the thoroughly. Take a test strip out Gently squeeze around the punctured area to obtain enough While" • "is flashing on the display, hold and touch the vial for the first time, record the and then close the vial tightly. blood. absorbent hole of the test strip to Lo H, KETONE? When glucose < 20 mg/dL (1.1 mmoljL) > 600 mg/dL (33 .3 mmoljL) 2 240 mg/dL (13.3 mmoljL) the drop of blood until the date on the label. confirmation window is completely Time of day filled . Twist the protective disk off the lancet. Replace and twist the cap to the right to close the lock as shown in the picture until you hear the click. When the lock is properly closed lock indicator is in the extreme right position. fD Pull the cocking control back until it clicks. You will see a color change inside the release button when it is ready. ..,.. If it does not click, the device may have been cocked when the lancet was inserted. Fasting and before meal < 100 mg/dl (5.6 mmoljl ) 2 hours after meals < 140 mg/dl (7.8 mmoljl) Source: American Diabetes Association (2010). Clinical Practice Insert a test strip to turn on the Recommendations. Diabetes Care, 33 (Supplement 1): S1-S100. Take the pre-set lancing device. After the meter counts down to 0, Eject the used test strip. Discard monitor. The sequences of LCD Hold the lancing device firmly your test result will appear along the used lancet and test strip display are: "CHK" and against the puncture site. Press with the date and time and will be according to the Please work with your doctor to determine a target range that works the release button to obtain a automatically stored in the best for you. drop of blood. memory. recommendations of your healthcare provider. "c::Jil " -+flashing " • "with date and time. Holding the cap select on the depth of penetration by turning the adjustable tip in either direction so that the arrow on the cap points to the desired depth. Normal plasma glucose range for people without diabetes (mg/dL) For testing with control solution, please refer to the Owner's Manual. for detailed information.
* Your assessment is very important for improving the work of artificial intelligence, which forms the content of this project