PR-TRENDvent-eng-0610-03.pdf 1 21.01.2011 09:53:14 Dimensions (w x h x l) 330 mm x 110 mm x 280 mm Mass 4,2 kg Power Supply 100 V - 250 V AC, 50 - 60 Hz; 24 V DC / 5 A Internal Battery Lithium-Ion-Rechageable, 24 V; 2.3 Ah Average Power Consumption. 52 W Noise Level < 30 dB(A) overall < 25 dB(A) at 10 hPa Parameters PCV/APCV PSV CPAP S ST T IPAP 4 - 40 hPa 4 - 40 hPa 2 - 20 hPa 4 - 40 hPa 4 - 40 hPa 4 - 40 hPa PEEP 0 - 20 hPa 0 - 20 hPa 2 - 20 hPa 2 - 20 hPa 2 - 20 hPa Frequency 4 - 50 bpm 4 - 50 bpm; Off 4 - 50 bpm 4 - 50 bpm Insp.Time 0,3 - 8,0 s EPAP TRENDvent Technical Data Accessories The functional bag was developed especially for the ventilator TRENDvent. Through the bag window, the functions of the device can be controlled. For the accessories an enlargeable compartment with separat space for the humidifier is integrated. The wide belts care for a good wearing comfort as backpack or sling bag. Many attachment features at the bag allow an easy assembly at wheel chair. The external rechargable battery is intended as autonomous power supply for all HOFFRICHTER ventilators including TRENDvent. It was especially developed for homecare use, use in hospitals and effort in mobile use. The rechargable battery enables an additonal using time of 12 hours (in standard adjustments). 0,3 - 8,0 s Apnoea Limit Off; 3 - 60 s; Off; 3-60 s Auto [as alarm] 1-5 1-5 1-5 1-5 1 - 5; Auto 1 - 5; Auto Off; 1 - 5; Auto Ramp 1-5 Insp.Trigger Off; 1 - 5; Auto 1 - 5; Auto 3 - 60 s; Auto Exp.Trigger 10 - 90%; Auto 10 - 90%; Auto 10 - 90%; Auto Max. Insp.Time 1 - 10 s 1 - 10 s 1 - 10 s Vt. min. 0.05 - 2 l; Off 0.05 - 2 l; Off 0.05 - 2 l; Off 0.05 - 2 l; Off 0.05 - 2 l; Off Add. Pressure 3 - 10 hPa 3 - 10 hPa 3 - 10 hPa 3 - 10 hPa 3 - 10 hPa The remote alarm box is a wire alarm system signalling alarms of the TRENDvent for home use. The alarm is generated by the ventilator which is not within the operator’s immediate hearing range. There is no difference between alarm type and alarm priority in alarm signalling. The FiO2-Sensor-Set measures the oxygen at the air outlet of the ventilator which shows the actual value in the display of the device. Alarm Parameters Max. Volume 0.2 - 2.5 l; Off 0.2 - 2.5 l; Off 0.2 - 2.5 l; Off 0.2 - 2.5 l; Off 0.2 - 2.5 l; Off 0.2 - 2.5 l; Off Min.Volume 0.01- 2 l; Off 0.01- 2 l; Off 0.01- 2 l; Off 0.01- 2 l; Off 0.01- 2 l; Off 0.01- 2 l; Off 10 - 120 bpm; 10 - 120 bpm; 10 - 120 bpm; 10 - 120 bpm; 10 - 120 bpm; Off Off Off Off Off Max. Frequency 10 - 120 bpm; Off Max. Oxygen 30 - 100 %; Off 30 - 100 %; Off 30 - 100 %; Off 30 - 100 %; Off 30 - 100 %; Off 30 - 100 %; Off Min. Oxygen 18 - 90 %; Off Pres. Difference 1 - 10 hPa Leakage 18 - 90 %; Off 18 - 90 %; Off 18 - 90 %; Off 18 - 90 %; Off 18 - 90 %; Off 1 - 10 hPa 1 - 10 hPa 1 - 10 hPa 1 - 10 hPa 1 - 10 hPa AquaTREND uni, the optional therapy air humidifier is able to be adjusted quickly with only one click. Easy cleaning and a large filling volume (300 ml) increases the user friendliness. An ingenious air-guidance system prevents water from getting into the therapy device. 20 - 200 l/min; 20 - 200; l/min; 20 - 200; l/min; 20 - 200; l/min; 20 - 200; l/min; 20 - 200; l/min; Off Off Off Off Off Off Specifications and Performance Max. Stable Pressure Limit 60 hPa Max. Working Pressure 40 hPa Min. Stable Pressure Limit 0 hPa Min. Working Pressure 0 hPa Max. Flow 250 l/min Technical Requirements for Accessories Oxygen Inlet: Bacterial Filter: Type of Connection Port Quick Coupling Connections 22 / 15 mm Cone (according to EN 1281-1) Manufacturer: Pressure < 1000 hPa Resistance < 2.3 hPa at 60 l/min Flow < 15 l/min CE-Mark required! HOFFRICHTER GmbH Mettenheimer Straße 12/14 19061 Schwerin Germany Phone: +49 385 39925 0 Fax: +49 385 39925-25 E-Mail: firstname.lastname@example.org www.hoffrichter.de Technical specifications are subject to technical modifications. PR-TRENDvent-eng-0610-03 Sales: For more noblesse PR-TRENDvent-eng-0610-03.pdf 2 21.01.2011 09:53:16 TRENDvent • Usage of Leakage- and Valve Tubing Systems With an easy switching, the system can be used with a leakage circuit or a valve circuit. PCV Pressure Controlled Ventilation • The ventilation archetype is pressure- and time-controlled. Airway Pressure IPAP controlled inspiration controlled inspiration controlled inspiration PEEP The wide, big handle guarantees an easy transport of the device. PCV - Pressure Controlled Ventilation • If the trigger is active ((A)PCV), the patient can request additionnal controlled inspiration breathes. • Invasive and non invasive ventilation for adults and children • Leakage- and valve ventilation • Clear menu structure • Up to three ventilation presettings IPAP Airway Pressure • Pressure modes with optional volume guarantee controlled inspiration assisted controlled inspiration assisted controlled inspiration spontaneous breath trigger spontaneous breath trigger All connections for the tubes are located next to each other on the frontside of the device. controlled inspiration A PCV - Assisted Pressure Controlled Ventilation • Extreme low noise level PSV Pressure Supported Ventilation • In this mode, normally every breath is triggered by the patient. The exception is the activation of the background frequency, caused by missing spontaneuos breathing of the patient. If the device is operated with a single line patient circuit with an external expiration valve, the ventilation can be done invasively, e.g. via a tracheostoma and also non invasively via a mask. Controlled Inspiration Airway Pressure The well-arranged user interface is very easy to handle. High comfort and maximum safety are the results in clinical and home use. The optional humidifier AquaTREND uni can be clicked on easyly. PEEP • Battery and mains operation TRENDvent was developed for the pressure controlled respiration therapy and ventilation of adults and children. Six different ventilation modes are available for an individual therapy with a minimum breathing volume of 50 ml. The operating concept impresses by the first contact. As soon as the panel is touched, the useable areas and the display are illuminated. The parameters are adjusted with the Control Pad. The cleaning of the surface is as easy as can be. The operating panel is made of one single piece of glas without any gaps. The function of the panel can be locked for more safety. IPAP PEEP Spontaneous Breath Spontaneous Breath Trigger Trigger PSV - Pressure Supported Ventilation At the backside of the device, the connections for the power supply (external supply), data transmission, a COM-connector, a connector for the remote alarm box and an USB-plug for the link between device and personal computer, are located.
* Your assessment is very important for improving the work of artificial intelligence, which forms the content of this project