April 12, 2017 Precision Medical, Inc. and Medical Fittings 300 Held
300 Held Drive Northampton, PA 18067 Phone: (610)-262-6090 Fax: (610)-262-6080 Website: www.precisionmedical.com April 12, 2017 Precision Medical, Inc. and Medical Fittings 300 Held Drive Northampton, PA 18067 Addendum To Precision Medical Flowmeters User Manual To Whom It May Concern, This letter is to supplement cleaning instructions for Precision Medical Air and Oxygen Flowmeters whether as single or with PTO to include flowmeters installed on wye blocks or tee branches. Precision Medical Air and Oxygen Flowmeters are not single patient/disposable items. This includes: 1MFA series 3MFA series 4MFA series 6MFA series 7MFA series 8MFA series 19MFA series Flowmeters can be externally cleaned/disinfected using hospital approved wipes with quaternary cleaning, sodium hypochlorite, or hydrogen peroxide. Alcohol based wipes should not be used to clean/disinfect oxygen flow meters and accessories due to the risk of fire. We recommend following the directions of the manufacturer of the chosen cleaner. The flowmeters should be wiped down between patient use. Precision Medical, Inc.
* Your assessment is very important for improving the work of artificial intelligence, which forms the content of this project