PCI Express® X16 Gen3 to MiniSAS HD Adapter for External Cable Quick Start Guide 1 Introduction Use this document for quick installation and setup. Teledyne LeCroy’s PCI Express X16 Gen3 to MiniSAS HD Adapter card provides and simple means for test and debug of PCIe® device/end-points using miniSAS HD connectors. It supports data rates of 2.5 GT/s, 5 GT/s and 8 GT/s. Several test points and headers provide easy access to check status of bus signals. The PCI Express X16 Gen3 to MiniSAS HD Adapter card works with miniSAS HD cables up to 0.5 meter in length as defined in the PCI Express External Cabling Specification Revision 3.0, Version 0.6. If SAS compatible cables are used, you must route the PCIe side bands and reference clock to your DUT from the host/root complex. PCI Express X16 Gen3 to MiniSAS HD Adapter 2 Components The adapter package includes the following components: • PCI Express X16 Gen3 to MiniSAS HD Adapter • Differential Clock cable • Quick Start Guide (this document) Inspect the received shipping container for any damage. Unpack the container and account for each of the system components listed on the accompanying packing list. Visually inspect each component for absence of damage. In the event of damage, notify the shipper and Teledyne LeCroy. Retain all shipping materials for shipper’s inspection. 3 Connections . PCI Express X16 Gen3 to MiniSAS HD Adapter: Interconnection Overview J16: I2C Interface See Table 2 below. Test Points: MiniSAS HD test points. See Table 5 below. J42: See PCIe Bus Side band signals see Table 3 below. J24: Port 3 CH[12-15] J24: Port 2 CH[8-11] J6: PCIe RefClk to DUT* miniSAS J3/J4: PCIe RefClk to DUT. MMCX Differential Output.* HD Cables J22: Port 1 SW1: PCIe Link Width Selection. See settings Table 1 below. CH[4-7] J22: Port 0 CH[0-3] J8: PCIe Bus Reserved signals. See Table 4 below. PCIe Card Slot * Note: For J6 and J3/J4: PCIe Reference Clock is not forwarded though the miniSAS HD cable. Outputs are provided in the adapter for debug purposes. Access to I2C Bus The I2C bus is connected from the PCIe Bus card slot to the MiniSAS HD connector J22: Port 0 and J16. J16 provides access to I2C/SMBus signals from the PCIe slot for other I2C analysis equipment. See Table 2 below, for the I2C connections on J16. 4 Switch Configuration: SW1 Table 1: PCIe Link Width Configurations SW1-1 SW1-2 SW1-3 SW1-4 PCIe Bus Link Width ON OFF OFF OFF x1 OFF ON OFF OFF x4 (default setting) OFF OFF ON OFF x8 OFF OFF OFF ON x16 Note 1: This switch sets the PRNST# bus pins so the card can be appropriately identified by the root complex. Note 2: Other switch configurations not valid. Note 3: Factory settings for all SW1 are: SW1-1 OFF, SW1-2 ON, SW1-3 OFF, SW1-4 OFF. 5 Connector Signals Connector signals are provided primarily as test points but they can be driven for debug purposes as long as the supplied signals meet the PCIe electrical specification. Table 2: J16 Connections PIN 1 2 3 4 DESCRIPTION DGND I2C_SDA_EXT I2C_SCL_EXT VCC (+3.3V to +5.0V) Note: VCC required only if I2C signals are not pulled up on the I2C analysis equipment. Table 3: J42 Connections PIN 1 2 3 4 DESCRIPTION PERST# IN WAKE# IN CLKREQ# IN DGND Table 4: J8 Connections PIN 1 2 3 4 5 6 DESCRIPTION RSVDA19 RSVDA32 RSVDA33 RSVDA50 RSVDB30 RSVDB82 6 Test Points Table 5: Mini SAS HD Test Points and Functions Test Point Number Function TP24 DGND TP53 P2_CBLPRSNT# TP55 P3_CINT# TP57 P3_CADDR TP28 P3_CMICLK TP29 P3_CMIDAT TP26 P2_CMIDAT TP27 P2_CMICLK TP54 P2_CADDR TP25 DGND TP52 P2_CINT# TP50 P1_CADDR TP49 P1_CINT# TP51 P1_CBLPRSNT# TP31 P1_CMIDAT TP30 P1_CMICLK TP47 P0_CADDR TP46 P0_CINT# 7 Environmental Conditions • Temperature: Operating 32° F to 122° F (0° C to 50° C) • Temperature: Non-Operating 14° F to 176° F (-10° C to 80° C) • Humidity: Operating 10% to 90% RH (non-condensing) Teledyne LeCroy Customer Support Online Download Periodically check the Teledyne LeCroy Protocol Solutions Group web site for software updates and other support related to this product. Software updates are available to users with a current Maintenance Agreement. Mail: Web: E-mail: Tel: Tel: Fax: Trademarks and Servicemarks Changes Teledyne LeCroy is a trademark of Teledyne LeCroy. All other trademarks are property of their respective companies. 3385 Scott Blvd., Santa Clara, CA 95054-3115 teledynelecroy.com/Support/SoftwareDownload/ email@example.com (800) 909-7112 (USA and Canada) (408) 653-1260 (worldwide) (408) 727-6622 (worldwide) Product specifications are subject to change without notice. Teledyne LeCroy reserves the right to revise the information in this document without notice or penalty. © 2015 Teledyne LeCroy, Inc. All rights reserved. Part Number: 926475-00 Rev A This document may be printed and reproduced without additional permission, but all copies should contain this copyright notice.
* Your assessment is very important for improving the work of artificial intelligence, which forms the content of this project