About AL8000NFC Breathalyzer The Fuel Cell Sensor Technology, AL8000NFC is intended to measure blood alcohol concentration of human body. It is easy to operation and handy, compact size of breath alcohol tester and showing high accuracy. Also it is simply connectable with android application by NFC tag function. So test data can be memorized on application and shared by SMS, E-mail and social networking services. About ALCOTAG Android application ALCOTAG is Android application to communicate with breath alcohol tester AL8000NFC by NFC (Near Field communication) capability. It is consist of measurement mode, data saving & sharing as well as checkable user’s drink habit & trend based on accumulated test data. In case of “TEST WITH CAMERA” mode can memorize person’s photo who actually tested by taking a picture while blowing. So it will be useful not only for who want to save test result on the application but also for who want to check subject’s identity. *Please download ALCOTAG App from the Google play store of your mobile device or please contact local distributor How to use AL8000NFC breathalyzer 1. Press the power button. 2. Warming up until you can see “0” sign. 3. Blowing through the mouthpiece into the unit for 4 seconds. 4. The result is displayed for 10 seconds and turn off automatically * If you keep press and hold the power switch, you can see “C” and “A” on the LCD C: The number of test after calibration A: The number of alcohol detected How to use AL8000NFC breathalyzer with ALCOTAG application ALCOTAG have 2 operating mode. Test with camera mode is taking picture of subject’s face while he or her blowing so it can be verifiable who tested actually. Normal test mode is simply loading your test result to your mobile device from your AL800NFC breathalyzer by NFC tagging. *Before using ALCOTAG app, please make sure that; -your mobile device must be installed Android operating system. -checking NFC chip installed area of you mobile device before tagging it. 1. Test with camera Turn on the breathalyzer and Tag breathalyzer to the Camera will be turned Tag breathalyzer with Check your result on the make sure the READY STATUS mobile device back-to- on and picture will be a result to the mobile mobile device (display “0”) is shown back taken automatically in device back-to-back 5s while blowing 2. Normal test Turn on the breathalyzer Tag breathalyzer with Check your result on the and make sure the READY a result to the mobile mobile device STATUS (displaying “0”) device back-to-back is shown Blow Components Permanent mouthpiece If you push the hook aside like above, you can use permanent mouthpiece. Contents Breathalyzer, Mouthpieces(x5), AAA Batteries(x2), Pouch and User’s manual Cautions a. Wait at least 20 minutes after eating or drinking before testing. Alcohol remaining in the mouth, or even excess saliva, may interfere with testing. This includes alcohol-based mouthwash or similar products. b. Avoid testing in high wind or enclosed spaces. Poor ventilation may also lengthen the time required for warm up between tests. c. Do not blow smoke, saliva or other contaminants into the mouthpiece, as the sensor may become damaged. d. Do NOT use Breathalyzer as a tool to drink and drive. e. Keep out of extreme temperatures and out of the reach of children. Specifications Indication of B.A.C 0.00 to 4.00 ‰ or 0.00 to 2.00 mg/ℓ or 0.00 to 0.40 %BAC by LCD Display Warm up time 3 seconds ~ 4 minutes Response time within 10 seconds at 0.100 %BAC Sensor Power supply weight dimensions(mm) Fuel Cell Sensor Two 1.5V "AAA"size alkaline batteries 66g (Including Batteries) 100(height) X 44(width) X 15(thickness) Ambient conditions Operation : 5 to 40 ℃ Accuracy ± 5% at 0.050 %BAC Storage : 0 to 50 ℃ Warning display of AL8000NFC breathalyzer Flo If you see “Flo” in the display, the breath sample was not strong enough for measurement. Please power on the unit again and make sure the breath is strong and steady during sampling. bAt CAL If you see “bAt” in the display, the battery power is low. Please replace the batteries. If you tested over 1,000 times, “CAL” will show in the display 3 seconds then warming up for test. Unit needs recalibration service. In a “TEST WITH CAMERA” mode, “Pt” will appear on the breathalyzer after Pt ready status tag, it means breathalyzer is communicated with App. under the camera working mode and ready to accept blow sampling. So blow steadily 4 seconds with staring camera of the mobile device Warranty The manufacturer warrants the product to be free from defects in workmanship or material (excluding calibration) under normal use for six (6) months from the date of purchase. The manufacturer’s obligation under this warranty is limited to replacing, adjusting or repairing the unit if returned along with proof of purchase. This warranty is void if the unit has been tampered with or maliciously damaged. Recalibration & Service For recalibration of breathalyzer and technical support, please contact local distributor or authorized service agent.
* Your assessment is very important for improving the work of artificial intelligence, which forms the content of this project