Efficient enlighten engineering 2010 BC Infant Radiant Warmer Features: ? Micro Controller based Electronics ? Controls: Servo Skin/Manual and Apgar ? Pre-warm: Use the equipment with 30% Heater Output in Manual Mode. ? No need of calibration required for temperature probe. ? High Intensity Quartz Heating element for effective Radiation (Ceramic heater Optional) ? Halogen Observation Lamp ? The Heater Module rotates 90° to both sides to facilitate X-Ray Procedures. ? The Parabolic Reflector is designed to send uniform heat to the bed surface. ? Memory Backup to restore set temperature and previous mode, in case of power failure. ? Optimum Bed dimensions ensure optimum work space for the nursing staff. ? Comprehensive Alarms and safety cut-offs. ? Programmable Mute time. ? Alert (Alarm) when sensor is detached from the patient's skin (optional) TECHNICAL SPECIFICATIONS ? Instant Warming and Uniform heating. Controller Electronics Mode of operation Skin mode control range Manual mode timer : : : : Apgar timer : Percentage of heater output Memory back up : : Programmable mute time Safety heater cut off : : Over temperature heater cut off : Temperature sensor (probe) Accuracy of skin sensor Accuracy of skin probe Interchangeability Calibration of skin sensor (probe) : : Heater element : Power rating Maximum heater wattage Maximum power requirement Examination light X-ray facility Mattress tilt IV pole Pump mounting pole Castor : : : : : : : : : Monitor shelf : A15898 ISO 9001:2000 & ISO 13485:2003 nice NeÖtech Chennai : : High speed micro controller Servo skin, manual & apgar o o 30.0 C to 38.0 C 1 min to 15 mins (selectable) 15 mins to 30 mins (optional) Alerts at 1, 5, 10, 15 & 20 mins (selectable) 10% to 100% in 10% increments Equipment restores the previous settings User can select the mute time If the skin sensor is removed from the baby's skin o If the skin temperature is >39 C in any skin/manual mode Thermistor sensor ± 0.1oC ±0.2 C No need of calibration after Changing new probe Quartz Infrared heater Ceramic heater (optional) ~230V AC, 50Hz, 700 W 600 Watts 700 Watts 12V, 50W Halogen lamp Underneath the bed ± 30 degrees on both sides Maximum load 2.2 Kg Maximum load 2.2 Kg 4” castors 4 Nos. (2 with brake & 2 without brake 2 Nos. Fixed o Height adjustable heater module Cabinet Operating noise level : : : Available only in nice 2010 Available only in nice 2010BC <50 dBA Controller Alarms: Check patient : Skin temperature high Skin temperature low Skin temperature probe fail Over temperature Heater fail System fail Power fail : : : : : : : If the skin sensor is removed from the baby's skin o > 1.0 C from the set temperature o < 1.0 C from the set temperature Skin Probe short or open o If skin temperature is >39 C Heater defect Electronic component defect AC power interruption Physical characteristics Equipment Weight Mattress Mattress from floor Maximum load Side panels Maximum load for monitor shelf : 85 cms(L) x70 cms(W)x185 cms(H) : 70 Kgs (nice 2010BC) : 73 cms (L) x 53 cms (W) x 5 cms (H) : 95 cms : 10 Kgs : Aluminum hinged acrylic panels : 4.5 Kgs Heater module swivel : Environmental Operating temperature Storage temperature Operating humidity Storage humidity Altitude Quality/Test Approval Quality Electrical & product safety ±90° 0 o : 15.0 C to 35.0 C 0 o : -10.0 C to 60.0 C : 15% to 90% RH, non condensing : 15% to 90% RH, non condensing : Sea level to 1-9 miles, 3 Km : : : ISO 9001:2000 & ISO 13485:2003 IEC 60601-1, IEC 60601-1-2 IEC 60601-1-4, IEC 60601-2-21 74-00-004, Rev 00 nice 2010 BC Infant Warmer with Bed & Storage Cabinet, nice 2010 B Infant Warmer with Bed & Storage Tray, nice 2010 Mobile Free Standing Infant Warmer with Height Adjustment.
* Your assessment is very important for improving the work of artificial intelligence, which forms the content of this project