Thinking ahead. Focused on life. AdvErL Evo Tips and Treatment Settings Irradiation Target (Indications) Tip Type Healthy Enamel PPS Energy Level Setting Spray Water (Hz) (mJ) On/Off 10 160 - 210 On 20 160 - 170 On 20 90 - 140 On 10 - 20 70 - 110 On 25 70 - 80 On 10 - 25 50 - 70 On 10 - 20 30 - 100 On 20 - 25 30 - 80 On 20 40 - 100 On 25 40 - 80 On 20 40 - 100 On 25 40 - 80 On Occlusal Surface Healthy Enamel Cavo Surface C400F C600F C800F Hard Healthy Dentin Caries Bone C400F C600F C800F S600T Calculus PS400T(S) Perio PS600T(S) Granulation Soft Tissue Incision Excision PS800T(S) S600T Begin treatment on the lowest energy setting and gradually increase to a higher level while monitoring the effectiveness. ! WARNING • A direct, reflected or scattered laser beam can cause permanent blindness. All individuals in the laser use area must wear laser safety glasses. • Use with care. Cooling air from the contact tip can cause subcutaneous emphysema or air embolism. Please refer to the full operations manual for complete instructions on how to operate and maintain this unit. Handpiece Attachment, Hanger, Tips & Tip Stand Cleaning/Sterilization Cleaning of Tips Autoclaving (Sterilization) 1. After using a tip, immerse the fiber part of the tip in tap water and emit the laser for 3 to 5 seconds. Use an enzymatic detergent (such as CIDEZYME® by Johnson & Johnson) to clean off blood and other contaminants. 1. Put the handpiece attachments, hanger and tips in sterilization pouches or the tip stand for autoclaving. 2. Place them inside the autoclave chamber. 2. Immerse the tip in a chemical disinfectant for the recommended time as described in the manufacturer’s instructions. Use one of the disinfectants listed below at the concentration specified for medical instruments cleaning. 3. Set autoclave cycle to the following parameters: • Temperature: 135 °C (275 °F) • Chlorhexidine Gluconate (Such as Hibiten ) • Time: 10 to 15 minutes • Ethanol for disinfection (70% to 80% by volume) • Dry time: 0 min (dry naturally) •D o not use ultrasonic cleaning methods; this could break the tip or cause its connector to fall off. • Do not use the drying stage if the autoclave has one. The temperature might be too high. ® •C ontact tips will wear out and must be replaced periodically. Replace damaged tips and those that cannot be adequately cleaned. 3. Wash tips thoroughly with tap water after using the cleaning solution. 4. Wipe the contact tip with cotton. Take care not to pull out the optical fiber. Cleaning of Handpiece Attachment Wipe entire outer surface of handpiece attachment with a soft cloth dampened with ethanol for disinfection (70% to 80% by volume). Three O-Rings Large Medium Small 4. After the autoclave cycle is complete, let them cool. Make sure all 3 o-rings of the tips are intact and are not damaged in any way. Pay special attention to the smallest o-ring. If this is missing or damaged, water could seep into the handpiece and damage it or cause the laser to lose power. It also might harm the drum lens. Cleaning of Hanger Wipe entire hanger outer surface with a soft cloth dampened with ethanol for disinfection (70% to 80% by volume). Maintenance Grease Handpiece End Drum Lens Cleaning (Handpiece End) Grease the handpiece end each day before use. 1. Rotate the applicator to apply grease and then remove it from its case. Disconnect the handpiece attachment and clean the lens on the end of the handpiece with the lens cleaner provided. Dampen the end of the lens cleaner with ethanol for disinfection (70% to 80% by volume) and lightly wipe the lens. Aiming Beam Good 2. Use a piece of gauze to wipe off excess grease. * Do not wipe off too much grease; leave enough to lubricate the handpiece attachment. 3. Insert the applicator into the handpiece as far as it will go and rotate it; then remove it. When finished, return the applicator back to its case. Hazy Ball Lens Cleaning (R-Attachment) Clean the ball lens on the end of the R-Attachment after each patient. Dampen the end of the lens cleaner with ethanol for disinfection (70% to 80% by volume) and lightly wipe the lens. Spray Water Bag Replacement (Sterile Water for Spray) 1-877-JMORITA (566-7482) www.morita.com/usa If air gets in the lines when the bag is replaced, depress the foot switch halfway down for 90 seconds to activate the water pump only and force the air out. L-1000 05/14 Before use, check the level of the water in the spray bag.
* Your assessment is very important for improving the work of artificial intelligence, which forms the content of this project