Components Hand Control Functions 5 1 2 3 3 6 1 2 4 3 6 1 Secondary Panel 2 Lift Belts 3 Direction Levers 1 Up Button 2 Down Button 3 USB and Charge Port Important! The covers for the USB input and the charging input (3) must be closed for the hand control’s water protection (IPX5) to be valid. Charging the Lift Flashing Red - Charge lift immediately! 1 2 3 1 Status Indicator Light 2 Power On/Off 3 Raise Lift 4 5 4 Lower Lift 5 Emergency Lowering Status Indicator Lights Flashing Green On/Ready for Use Full Battery Constant Red Safety Catch Engaged Troubleshooting! Flashing Yellow On/Ready for Use Charge Battery Warning Constant Yellow Lift Charging Functions Blocked Flashing Red On/Ready for Use Low Battery Flashing Green/Yellow Service Warning Perform Inspection Constant Green Lift in Use Lift Loaded Flashing Red/Yellow Service Passed Battery Low QUICK REFERENCE GUIDE General Care Emergency Stop Activation Press down on red button to activate. Emergency De-Activation Turn red button clockwise. Important! The emergency lowering button must only be used in emergency situations. Suspending the lift Place the lift on a suitable surface and feed out belt by gently pulling out belt while at the same time pressing the “Down” button on either the hand control or secondary panel. Hook lift belt onto distance strap so that it is held firmly in position (easiest to hook from above at an angle as shown). Ensure the lift belt is firmly in place before use. 4 Telescoping Hanger Bar 5 Emergency Stop 6 Hand Control Secondary Panel Functions Using the lift 2 1 1 Connect charger plug to handset. 2 Connect plug to charger and insert plug into wall to charge. The charger must only be connected to an earthed socket. Note: If the emergency stop is activated, the lift will not charge! Important! The lift can only be used with a Human Care charger and must be disconnected to use the lift. Models: 55070 Human Care HC AB. All Rights Reserved. Room to room transfers Activate desired belt by turning directional lever. If lever faces towards the leg hook, the belt nearest to the leg hook is active. If it faces towards the hanger bar, the belt closest to the bar is active. Important! Do not turn the lever unless both belts are connected to their respective rail rollers. The lever must always point to the connected belt. Slings Attach head sling loops to carry bar first, then leg straps to leg hooks. Ensure the straps are secured beneath the safety pin. Charge lift every night. Wipe down with slightly dampened drying cloth. Disinfect with DuPont TM Virkon®. Never use strong detergents on lift. d water. Do not immerse in Activate emergency stop when not in use. Important! Do not use strap to transport! Lift must always be stored or moved in horizontal position. Rest on cart to move. Inspection Annual Inspection - The lift must be inspected at least once per year by a qualified technician certified by Human Care. Monthly Inspection - Check lift belts for damage or wear. Feed entire length to ensure proper functioning. Daily Inspection - Always inspect the lift for visible signs of damage before use. Ensure the lift is correctly mounted on the distance strap or rail roller. Always check slings and belts for signs of wear before use. Do not use if wear evident. Troubleshooting If the indicator light is not on, check the following: - Mains switch is on - Batteries do not require charging - Emergency stop button is not pushed in (deactivate by turning clockwise) If the indicator light is flashing, check the following: - Check the hand control button functions up / down when depressed (replace if necessary) - When pressing up, the lift is not already in its maximum upper position - When pressing down, the lift is not already in its maximum lower position If the indicator light is constant red: - The lift motor is overheated - allow to cool - The lift is overloaded - lower using emergency lowering button Proper lifting techniques Do not use the telescoping bar to move the patient and the lift. Do not hold onto the hooks while performing a lift. Once the patient is comfortably secured in sling, raise lift, hold sling and position/move patient. ROOMER S 220kg If the indicator light is constant yellow: - Charging in progress If the lift belt does not feed out properly: - Pull the belt upwards at the same time as you depress the Down button on the hand control / lift panel If the indicator changes to flashing red: - The batteries need charging, patient may be lifted a few more times 99588-Quick Reference Guide-Roomer S-Rev 02 (05 2017) Important! This overhead lift has been designed and tested for indoor use with one patient at a time. This product must not be immersed in water. Read all instructions for full product information! Important! This quick reference guide does not replace the user manual which can be downloaded at www.humancaregroup.com. Read all instructions before use!
* Your assessment is very important for improving the work of artificial intelligence, which forms the content of this project