PLV®·102b PORTABLE VOLUME VENTILATOR , £ RESPIRONICS PLV-102b When you need a portable volume ventilator capable of ventilating a wide range of patients, you need the PLV-102b. The PLV-102b uses a proven microprocessor-controlled, piston-driven ventilator platform to deliver volume ventilation in a range of settings. Its features address the needs of many patients who require volume ventilation and portability or transport in the sub-acute care, nursing home or home setting. FEATURES Portable· The PLV-102b is well suited for transport and portable applications due to its small size and light weight. The ventilator is capable of operating from any of three power sources: 120 volts AC, an internal 12 volt DC battery and an external 12 volt DC battery, all of which are automatically switched by the PLV-102b. Controls • The PLV-102b provides three modes: Control and Control-Assist, each of which can be used wit~ optional Sigh, and the SIMV mode. The sensitivity of the PLV-102b is adjustable from -6 to + 18 cm H20, which allows for use with PEEP up to 20 cm H20. The sensitivity is automatically and continuously corrected for barometriJ pressure changes. I Alarms and Displays· Alarms are provided for Low Pressure, High Pressure, Apnea, Inverse I:E ratio, Increase In11l1rlllflJ rum IlIlU runur OUUI UU iOUUOOI TIIOI 0 UI U UIOU Oill UlrrUi Dill UIUlIlI UUUUU IU UUUIUI III lruUUIOUllUUlIllY Large, backlit LeDs display Tidal Volume, Respiratory Rate, I:E Ratio and Inspiratory Flow Rate. POWER SOURCES FRONT PANEL CONTROLS Mode Selection: CONTROL CONTROL + SIGH ASSIST/CONTROL ASSIST/CONTROL + SIGH SIMV Tidal Volume: 0.05 to 0.20 +/- 0.02 liters 0.20 to 3.00 liters +/- 10% Sigh Volume: 150% of Set Tidal Volume Sigh Frequency: 1 Sigh Every 100 Tidal Breaths Breath Rate: 2 to 35 BPM +/- 0.5 36 to 40 BPM +/- 2 Inspiratory Flow Rate: 10 to 120 LPM +/- 10% Battery Test Switch: Test Internal/External Battery Main Power Switch: On, Off/Recharge DC Circuit Breaker: Push/Pull Actuation AC Circuit Breaker: Push/Pull Actuation Airway Pressure Limit: 5 to 100 cm H20 +/- 5 Sensitivity: +18 cm H20 +/- 1 to at least -6 cm H20 Low Pressure Alarm: 2 to 40 cm H20 +/- 2 41 to 50 cm H20 +/- 5 Manual Sigh: Push to Activate One Breath 30 Second Alarm Silence: Push to Activate FRONT PANEL VISUAL INDICATORS 120 VAC 50/60 Hz 12 VDC Internal Battery 12 VDC External Battery PHYSICAL DIMENSIONS Overall Dimensions: 9" H x 12.25" W x 12.' 5" D (22.9 cm x 31.1 cm x 1.1 cm) Weight: 28.2 lb. (12.8 kg) ALARMS Low Pressure: Visual and Continuo s Audible High Pressure: Intermittent Audible Apnea: Visual and Continuo s Audible Inverse I:E Ratio: Visual Warning Increase Inspiratory Flow: Visual Warning Low Internal Battery: Visual and Continuo s Audible Low External Battery: Visual and Continuo s Audible Reverse Battery Connection: Continuous Audible Power Failure: Continuous Audible Unit Malfunction: Intermittent Audible Power Source Change: Continuous Audible ORDERING INFORMATION Manual Sigh: Yellow Indicator Light Description Part Number Tidal Volume: Digital Display PLV-l02b Ventilator 1003111 Rate: Patient Breaths Per Minute: Digital Display I:E Ratio: Digital Display Inspiratory Flow Rate: Digital Display Increase Inspiratory Flow Rate: Red Indicator Light 120V AC Power: Green Indicator Light Internal Battery Source: Amber Indicator Light External Battery Source: Yellow Indicator Light Patient Airway Pressure: -10 to +100 cm H20 Assist/Spontaneous: Green Indicator Light 15-Second Delay (Low Pressure Alarm): Green Indicator Light 30-Second Alarm Silence: Yellow Indicator/Push Button Alarm Code Display (in Front Window): Digital Display !&1 Global Headquarters: Pittsburgh, PennsylvaniaUSA· 412-131-2100 Customer Service: 1-800-345-6443 RESPIRONICS@ Respironics Europe: +33-(0) 1-55-60- 9-80 www.resplronlcs.com Respironics Asia Pacific: +852-234-3 PLV-102b of U.S. patents ventilators are the subiscts #4,493,614, #4,617,637. #4.58 2-18 967, ~IIUUIlI~rp~IIDlligua AIIDlur~lgrrp!l~mlllU I!! r~QI!I~r~DIr!O~rrr!rl01r(~!plrUIU CAUTION: ©2000 U. S. federal Respironics. Inc. law restricts this device to sale by or on the order of a physici n.