OWNER’S MANUAL MEDICAL MONITOR Please read this manual carefully before operating your set and retain it for future reference. 27HJ710S www.lg.com Copyright ⓒ 2017 LG Electronics Inc. All Rights Reserved. 2 ENGLISH Please note that this information is for proper use and safety of the equipment. The following symbols may indicate a hazardous situation in which, if not heeded, may result in serious injury or even death to the user or others, or damage to the equipment. WARNING CAUTION Indicates warning and safety instructions. If not adhered to, it could result in death or serious injury to the user or others. Indicates a hazardous situation which, if not heeded, may result in minor or moderate injury to the user or others, or damage to the equipment. For users in the United States •• United State federal law restricts this equipment to be used by or on the order of a physician. For users in other countries •• This equipment is to be used by or on the order of a licensed person under the related laws for each country. WARNING •• To avoid the risk of electric shock, this equipment must only be connected to a supply mains with protective earth. •• Do not modify this equipment without authorization of the manufacturer. Intended use •• This Medical Monitor is intended to provide color video displays and images from medical equipment which include laparoscopy and endoscopy systems for surgery and various medical imaging systems. EC REP Authorized representative in the European community. LG Electronics European Shared Service Center B.V. Krijgsman 1, 1186 DM Amstelveen, The Netherlands Tel: +31-20-456-3132 LG Electronics Inc. 77, Sanho-daero, Gumi-si, Gyeongsangbuk-do, 39381, Republic of Korea Tel: +82-1544-8777 3 SAFETY INFORMATION ENGLISH Safety Standard Medical Device Classification Classification by protection type against Electric Shock Classification according to the degree of protection against ingress of water or particulate matter Mode of operation Environment of Use Class Ⅰ equipment Front: IP35 Except for Front: IP32 Continuous Operation This equipment is not suitable for use in the presence of flammable anesthetic or oxygen. Regulations Safety and Electromagnetic Compatibility Information This equipment has been tested and found to comply with the limits for medical devices in IEC 60601-1-2. These limits are designed to provide reasonable protection against harmful interference in a typical medical installation. This equipment generates, uses and can radiate radio frequency energy. If not installed and used in accordance with the instructions, it may cause harmful interference to other devices in the vicinity. However, there is no guarantee that interference will not occur in a particular installation. If this equipment does cause harmful interference to other devices, which can be determined by turning the equipment off and on, the user is encouraged to try to correct the interference by one or more of the following measures. •• Reorient or relocate the equipment. •• Increase the separation between the equipment. •• Connect the equipment into an outlet on a circuit different from that to which the other devices are connected. •• Contact manufacturer or authorized agent for help. Radio Frequency compliance FCC (For USA) FCC NOTICE This device complies with part 15 of the FCC Rules. Operation is subject to the following two conditions: (1) This device may not cause harmful interference, and (2) this device must accept any interference received, including interference that may cause undesired operation. 4 ENGLISH FCC CAUTION Changes or modifications not expressly approved by the party responsible for compliance could void the user’s authority to operate the equipment. This transmitter must not be co-located or operated in conjunction with any other antenna or transmitter. FCC WARNING This equipment may generate or use radio frequency energy changes or modifications to this equipment may cause harmful interference unless the modifications are expressly approved in the instruction manual. The user could lose the authority to operate this equipment if an unauthorized change or modification is made. Note: This equipment has been tested and found to comply with the limits for a Class A digital device, pursuant to part 15 of the FCC Rules. These limits are designed to provide reasonable protection against harmful interference when the equipment is operated in a commercial environment. This equipment generates, uses, and can radiate radio frequency energy and, if not installed and used in accordance of this equipment in a residential area is likely to cause harmful interference in which cause the user will be required to correct the interference at his own expense. For Europe DECLARATION OF CONFORMITY The complete Declaration of Conformity may be requested through the following postal address: LG Electronics European Shared Service Center B.V., Krijgsman 1, 1186 DM Amstelveen, The Netherlands or can be requested at our dedicated DoC website: http://www.lg.com/global/support/cedoc/cedoc# Electro-Magnetic Compatibility Information Electro-Magnetic Emissions This EUT is intended for use in the electromagnetic environment specified below. The customer or the user of the EUT should assure that it is used in such an environment. Immunity Test Compliance RF Emissions CISPR 11 Group 1 RF Emissions CISPR 11 Harmonic emissions IEC 61000-3-2 Voltage fluctuations/ Flicker emissions IEC 61000-3-3 Class A A Complies Electromagnetic Environment – Guidance The EUT uses RF energy only for its internal function. Therefore, its RF emissions are very low and are not likely to cause any interference in nearby electronic equipment. The EUT is suitable for use in all establishments, including domestic establishments and those directly connected to the public lowvoltage power supply network that supplies buildings used for domestic purposes. 5 Electro-Magnetic Immunity Immunity Test Electrostatic discharge (ESD) IEC 61000-4-2 IEC 60601-1-2 Test Level ± 8 kV contact ± 2 kV, ± 4 kV, ± 8 kV, ± 15 kV air Compliance Level ± 8 kV contact ± 15 kV air Electromagnetic Environment – Guidance Floors should be wood, concrete or ceramic tile. If floors are covered with synthetic material, the relative humidity should be at least 30%. Electrical fast transient/ ± 2 kV burst 100 kHz repetition IEC 61000-4-4 frequency ± 2 kV 100 kHz repetition frequency Mains power quality should be that of a typical commercial or hospital environment. Surge Line-to-line IEC 61000-4-5 ± 0,5 kV, ± 1 kV ± 1 kV Mains power quality should be that of a typical commercial or hospital environment. Surge Line-to-ground IEC 61000-4-5 Voltage dips IEC 61000-4-11 ± 0,5 kV, ± 1 kV, ± 2 kV 0 % UT; 0,5 cycle At 0°, 45°, 90°, 135°, 180°, 225°, 270° and 315° 0 % UT; 1 cycle and 70 % UT; 25/30 cycles Single phase: at 0° 0 % UT; 250/300 cycle ± 2 kV Voltage interruptions IEC 61000-4-11 RATED power frequency 30 A/m magnetic fields (50/60Hz) IEC 61000-4-8 0 % UT; 0,5 cycle At 0°, 45°, 90°, 135°, 180°, 225°, 270° and 315° 0 % UT; 1 cycle and 70 % UT; 25/30 cycles Single phase: at 0° 0 % UT; 250/300 cycle Mains power quality should be that of a typical commercial or hospital environment. If the user of the EUT image intensifier requires continued operation during power mains interruptions, it is recommended that the EUT image intensifier be powered from an uninterruptible power supply or a battery. 30 A/m Power frequency magnetic fields should be at levels characteristic of a typical location in a typical commercial or hospital environment. NOTE •• UT is the A.C mains voltage prior to application of the test level. ENGLISH This EUT is intended for use in the electromagnetic environment specified below. The customer or the user of the EUT should assure that it is used in such an environment. 6 ENGLISH Immunity Test IEC 60601-1-2 Test Level Conducted disturbances induced by RF fields IEC 61000-4-6 3V 0,15 MHz – 80 MHz 6 V in ISM bands between 0,15 MHz and 80 MHz 80 % AM at 1 kHz Radiated RF EM 3 V/m fields 80 MHz – 2,7 GHz IEC 61000-4-3 80 % AM at 1 kHz Compliance Level 3V 0,15 MHz – 80 MHz 6 V in ISM bands between 0,15 MHz and 80 MHz 80 % AM at 1 kHz 3 V/m 80 MHz – 2,7 GHz 80 % AM at 1 kHz Electromagnetic Environment – Guidance Portable and mobile RF communications equipment should be used no closer to any part of the EUT, including cables, than the recommended separation distance calculated from the equation applicable to the frequency of the transmitter. Recommended separation distance: 80 MHz to 800 MHz 800 MHz to 2.7 GHz Where P is the maximum output power rating of the transmitter in watts(W) according to the transmitter manufacturer and d is the recommended separation distance in meters(M). Field strengths from fixed RF transmitters as determined by an electromagnetic site survey, should be less than the compliance level in each frequency range. Interference may occur in the vicinity of equipment marked with the following symbol: NOTE •• At 80 MHz and 800 MHz, the higher frequency range applies. •• These guidelines may not apply in all situations. Electromagnetic propagation is affected by absorption and reflection from structures, objects and people. 1 Field strengths from fixed transmitters, such as base stations for radio (cellular/cordless) telephones and land mobile radios, amateur radio, AM and FM radio broadcast and TV broadcast cannot be predicted theoretically with accuracy. To assess the electromagnetic environment due to fixed RF transmitters, an electromagnetic site survey should be considered. If the measured field strength in the location in which the EUT is used exceeds the applicable RF compliance level above, the EUT should be observed to verify normal operation. If abnormal performance is observed, additional measures may be necessary, such as re-orienting or relocating the EUT. 2 Over the frequency range 150 kHz to 80 MHz, field strengths should be less than [V1] V/m. 7 Test specifications for ENCLOSURE PORT IMMUNITY to RF wireless communications equipment Band (MHz) 385 Maximum Distance power (m) (W) IMMUNITY TEST LEVEL (V/m) Service Modulation 380 –390 TETRA 400 Pulse modulation 18 Hz 1,8 0,3 27 430 – 470 GMRS 460, FRS 460 FM ± 5 kHz deviation 1 kHz sine 2 0,3 28 704 – 787 LTE Band 13, 17 Pulse modulation 217 Hz 0,2 0,3 9 800 – 960 GSM 800/900, TETRA 800, iDEN 820, CDMA 850, LTE Band 5 Pulse modulation 18 Hz 2 0,3 28 1 700 – 1 990 GSM 1800; CDMA 1900; GSM 1900; DECT; LTE Band 1, 3, 4, 25; UMTS Pulse modulation 217 Hz 2 0,3 28 2 450 2 400 – 2 570 Bluetooth, WLAN, 802.11 b/g/n, RFID 2450, LTE Band 7 Pulse modulation 217 Hz 2 0,3 28 5 240 5 500 5 785 5 100 – 5 800 WLAN 802.11 a/n Pulse modulation 217 Hz 0,2 0,3 9 450 710 745 780 810 870 930 1 720 1 845 1 970 NOTE •• If necessary to achieve the IMMUNITY TEST LEVEL, the distance between the transmitting antenna and the ME EQUIPMENT or ME SYSTEM may be reduced to 1 m. The 1 m test distance is permitted by IEC 61000-4-3. a) For some services, only the uplink frequencies are included. b) The carrier shall be modulated using a 50 % duty cycle square wave signal. c) As an alternative to FM modulation, 50 % pulse modulation at 18 Hz may be used because while it does not represent actual modulation, it would be worst case. ENGLISH Test frequency (MHz) 8 Symbols ENGLISH Symbols Descriptions Refer to instruction manual/ booklet Alternate current Direct current Off (power: disconnect from the main switch) On (power: connect from the main switch) Warning Caution E486403 UL classified mark of medical equipment with respect to electrical shock, fire and mechanical hazards only in accordance with ANSI/AAMI ES60601-1 (2005) + AMD 1 (2012), CAN/ CSA-C22.2 No. 60601-1 (2014) Manufacturer Date of manufacture SN Serial number Non-ionizing radiation WEEE: Waste Electrical and Electronic Equipment EC REP Authorized representative in the European community. CE Marking 9 Humidity limitation Pressure limitation IPN N 1 For the customers in the U.S.A. Caution Federal law (United States of America) restricts this device to sale by or on the order of a licensed healthcare practitioner. 2 Ingress of protection Warning WARNING: To avoid the risk of electric shock, this equipment must only be connected to a supply mains with protective earth. Connection Do not connect the monitor with anything other than specified. Otherwise, it may result in fire or electric shock. To avoid the risk of electric shock, this monitor must only be connected to supply mains with protective earth. Handling Always be sure to keep checking the condition of the system and the patient to ensure they are normal during the use of the monitor. If any problem is found, take appropriate measures, such as stopping the operation of the monitor, as required. Never disassemble or modify the monitor as it may result in fire or electric shock. Also, since the monitor incorporates parts that may cause electric shocks and other hazardous parts, touching them may cause death or serious injury. Do not hit or drop the monitor. The monitor may be damaged if it receives a strong jolt, which may result in fire or electric shock if the monitor is used without being repaired. The operator must not touch connectors of the monitor and the patient simultaneously. The product has lower breaking capacity type. So do not install at the building power system prospective short-circuit current exceeding 35 A. ENGLISH Temperature limit 10 Caution ENGLISH Do not install the ME Equipment in a location without easy disconnect accessibility. Environment of Use and Storage Do not install the monitor in a location with the conditions listed below. Otherwise, it may result in failure or malfunction, cause fire or injury. •• Close to facilities where water is used. •• Where it will be exposed to direct sunlight. •• Close to air-conditioner or ventilation equipment. •• Close to heat source such as a heater. •• Prone to vibration •• Unsecure location place. •• Dusty environment. •• Saline or sulfurous environment. •• High temperature or humidity. •• Freezing or condensation. Do not place the storage case in a location with the conditions listed below. •• Where the cable of the monitor unit will be strongly pulled when the monitor is put into the case, otherwise, the cable may be damaged, resulting in fire or electric shock. •• Where someone might get their foot caught in the cable of the monitor. Handling For safety reasons, be sure to turn off the power when monitor is not used. This monitor is contraindicated for pregnant woman. MANUFACTURER will make available on request circuit diagrams, component part lists, descriptions, calibration instructions, or other information that will assist SERVICE PERSONNEL to repair those parts of ME EQUIPMENT. Location of Cables Make sure all cables are located so that they cannot be stepped on, tripped over, or otherwise subjected to damage or stress. If the monitor is defective, do not disassemble the monitor randomly. Maintenance of the monitor should be done by a manufacturer. 11 SAFETY PRECAUTIONS WARNING CAUTION Failure to follow the instructions may result in serious injury or even death. Failure to follow the instructions may result in minor injury to people or damage to the product. Precautions When Installing the Product WARNING •• Do not place the product close to heat sources such as radiators, fires, stoves, etc. •• Be careful when disinfecting the product, as some disinfectants may cause fire. •• Do not install the product in a damp or dusty area. •• If the product emits smoke, strange odors or sounds, disconnect the power cord and contact the manufacturer. •• Do not leave the power or signal cable, etc. in walkways. •• Do not install the product onto a wall close to sources of oil or oil mist. •• If the product has been dropped or its case is damaged, turn off the power and unplug the power cord from the power outlet. CAUTION •• Ensure that the product is installed at least 100 mm away from the wall for good ventilation to prevent deformation of the product or a fire due to the increase in the internal temperature. •• Keep the product away from direct sunlight. •• Do not install the product close to hot objects, such as lights. •• Make sure the air vents are not blocked by a tablecloth or curtain. •• Do not install the product near magnetic field generating equipment, such as transformers or high-voltage power lines. Installing the product in locations that do not meet the general conditions may seriously damage the product’s picture quality, life cycle and appearance. Please check with our service engineer before installing in such locations. Locations that do not meet the general conditions: places where the product is not intended for use, such as those where fine dust or oil mist is generated, places where chemicals are used, places where the temperature is too high or low, places where the humidity is too high, or places where the product is likely to be used for a long period of time (e.g. airports or train stations). ENGLISH The Safety Precautions are intended to prevent unexpected danger or harm by helping the user to use the product safely and for its intended purpose. 12 Precautions for the AC/DC Adaptor and Power ENGLISH WARNING •• If water or a foreign substance enters the product (power cord or AC/DC adaptor), disconnect the power cord immediately and contact the manufacturer. •• Always connect the power cord to a grounded outlet. •• Do not touch the power plug or AC/DC adaptor with wet hands. If the pins of the plug are wet or dusty, clean and dry them before use. •• Insert the power plug securely so that it does not come loose. Do not use the product if the power outlet is loose. •• Make sure the power cord is fully inserted into the AC/DC adaptor. •• Do not insert any conductive objects into the device connection end of the power cord, while the power cord is inserted into the wall outlet. Avoid touching the power plug immediately after pulling from the wall outlet. •• Be sure to use the power cords and AC/DC adaptors provided or ones approved by LG Electronics, Inc. •• Use the product at the rated voltage only. •• Disconnect the power cable when not using the product for long periods. •• Always pull by the plug when unplugging the power cord. Do not bend the power cord with excessive force. •• Take care not to step on or place heavy objects (electronic appliances, clothing, etc.) on the power cord or AC/DC adaptor. Be careful not to damage the power cord or AC/DC adaptor with sharp objects. •• Never disassemble, repair or modify the power cord or AC/DC adaptor. •• To turn off the main power, remove the power cord, which should be positioned so it is easy to access for operation. •• If the product is connected to an AC wall outlet, it will not be disconnected from the AC power source even if you turn off the switch. CAUTION •• Do not remove the power cord when the product is in use. •• Be sure to keep the outlet, AC/DC adaptor and pins of the power plug clean from dust, etc. •• Do not turn the monitor on or off by plugging in or unplugging the power plug from the power outlet. (Do not use the power plug as a switch.) •• Keep the power cord away from heating devices. Precautions When Transporting the Product WARNING •• Before moving the product, disconnect the power cord and all other connected cables. •• When carrying the product, make sure the screen faces forward and hold it firmly in both hands. 13 CAUTION product after it has fallen may result in electric shock or fire. Contact the manufacturer. Precautions When Using the Product WARNING •• Do not disassemble, repair or modify the product by yourself. If the product needs to be checked, reset or repaired, contact the manufacturer. •• Do not allow water to enter the product and keep the product dry at all times. •• If there is a gas leak, do not touch the power outlet. Open the windows for ventilation. •• Do not impact the front or sides of the screen with a hard object, such as a metal object. Do not scratch the screen. •• If liquid or a foreign object falls into the product, switch the product off and unplug it from the wall outlet. Contact the manufacturer. •• Stop using the product if no image appears on the screen or no sound is heard. Switch the product off immediately. Unplug the product from the power outlet and contact the manufacturer. •• Do not use the product in any environment with excessively high temperatures or humidity. •• Do not use high-voltage electrical products near the product. This may result in product malfunction due to electric shock. (e.g. electrical mosquito swatter) CAUTION •• Do not use or store the product near inflammables. •• Take a break to protect your health and vision when using the product for long periods of time. •• Do not push hard against the screen surface or scratch it with your hands or sharp objects, such as a nail, pencil or pen. •• Keep the product clean. •• Refer to the Owner's Manual and set the product to the correct resolution and frequency. Otherwise, you may experience blurred vision. •• Do not leave the product where it can get wet. Do not place anything containing liquid on top of the product, such as a flower vase. •• Because the LCD requires high technology consisting of hundreds of pixels, pixelated spots (red, green, white or black) may appear on the screen during use. This is normal for an LCD screen and not an error, nor is it related to the monitor's performance. •• Given the nature of the viewing angle, the brightness and color of the LCD may vary between the left and right or top and bottom, depending on the viewing position. This is normal for an LCD screen and not an error, nor is it related to the monitor's performance. ENGLISH •• Do not discard the delivery box with the original packaging. Put the product in the box when carrying. •• When moving or unpacking the product, make sure that two or more people safely lift it because it is heavy. Using the 14 Precautions for Connecting to Other Medical Devices ENGLISH Take the following precautions before using this product or connecting it to other medical devices. •• Stop using this device for medical purposes if you experience interruptions or discomfort in executing medical activities. •• Quick movements, shaking and the focus of the video displayed on the monitor, its distance from the user, the user's point of view, the user's physical fitness and other factors may cause discomfort (eye fatigue, dizziness, vomiting, nausea, etc.). •• Before using the product, check that the image of the connected device displays properly on the screen of this product. Precautions for Use with Electric Scalpels, etc. •• When this product is used with electric scalpels, etc., the image may appear unstable, distorted or abnormal due to the strong electromagnetic waves or voltage emitted by the device. This is not a malfunction. •• If you are using this device with equipment that emits strong radio waves or voltage, install this device in a direction that minimizes radio wave interference. Recommendation for Using Multiple Units •• Electronic products may fail unexpectedly. If using the monitor for medical, emergency or any other critical use, we strongly recommend that you use multiple monitors or have a spare monitor ready. Precautions When Cleaning the Product The medical LCD monitor's front protective panel has been specially processed to reflect less light. Using solvents such as thinner or benzene, acidic, alkaline or corrosive detergents, chemical cleaning fibers, etc. may damage the finish material on the surface. •• Unplug the power cord before cleaning. •• Never spray water directly onto the monitor or use thinner or benzene solvents, acidic, alkaline or corrosive detergents, or cleaning or sterilizing chemical cleaning fibers, etc. as they may damage the product. •• To clean the product, unplug the power cord and wipe it gently with a soft cloth. Do not spray with water or wipe with a wet cloth. When cleaning the product or the screen, do not use cleansers, automobile or industrial thinners, abrasives or wax, benzene, alcohol, etc., as this may damage the product. •• To clean the front frame, spray water onto a soft cloth two to four times and wipe in one direction only. •• Do not rub the product with a stained cloth using more force than necessary. This may scratch the surface of the product. •• Do not allow the product to remain in contact with rubber or vinyl resin for extended periods. The surface finish may deteriorate or the coating may peel off. •• If you do not clean the product, dust will accumulate inside and may create a fire hazard or malfunction. 15 Image Sticking Precautions products have the same issue. Damage resulting from long-term static images is not covered by product warranty. LICENSE Each model has different licenses. Visit www.lg.com for more information on the license. The terms HDMI and HDMI High-Definition Multimedia Interface, and the HDMI logo are trademarks or registered trademarks of HDMI Licensing LLC in the United States and other countries. VESA, VESA logo, DisplayPort compliance logo and DisplayPort compliance logo for dual-mode source devices are all registered trademarks of the Video Electronics Standards Association. The SuperSpeed USB Trident logo is a registered trademark of USB Implementers Forum, Inc. ENGLISH •• Displaying a still image for a prolonged period may damage the screen, resulting in image sticking. Most third-party 16 PRODUCT SPECIFICATIONS ENGLISH The product specifications are subject to change without prior notice for product improvements. “ ~ ” refers to alternating current (AC), “ LCD Screen Resolution ” refers to direct current (DC). Type TFT (Thin Film Transistor) LCD (Liquid Crystal Display) Screen Pixel Pitch 0.1554 mm x 0.1554 mm Maximum resolution Recommended resolution Video signal Input Connectors Output Connectors Power Source 3840 x 2160 @ 60 Hz Horizontal frequency 30 kHz to 135 kHz Vertical frequency 56 Hz to 61 Hz SDI IN, DVI-D IN, HDMI IN, DP (DisplayPort) IN, USB UP ( SDI OUT, DVI-D OUT, DP OUT Power Rating 19 V 6.32 A ), USB ( Power consumption AC-DC Adaptor Environmental Conditions Max 120 W Off Mode: ≤ 0.3 W DA-120D19 type, manufactured by Asian Power Devices Inc. (APD) Input: 100-240 V~ 50-60 Hz, 1.8-0.7 A Output: 19 V 6.32 A Class I Operating Conditions Temperature 0 °C to 40 °C Storing Conditions Dimensions (mm) Weight (without packaging) (kg) Humidity Pressure Temperature 0 % to 80 % 700 hPa to 1060 hPa -20 °C to 60 °C Humidity Pressure 0 % to 85 % 500 hPa to 1060 hPa Monitor Size (Width x Height x Depth) 654.4 x 410.9 x 58 7.7 ), RS-232C 17 Prevention of electric shock Operating mode Class Ⅰ ENGLISH Classification according to the degree of protection against ingress of water or particulate Medical Specifications matter Level of safety in places where flammable anesthetics are mixed with air, oxygen or nitrogen oxides Front: IP35 Except for Front: IP32 Not suitable for use in places where flammable anesthetics are mixed with air, oxygen or nitrogen oxides. Continuous IP35 IP35 18 TABLE OF CONTENTS ENGLISH ASSEMBLY AND PREPARING-------- 19 -- Product Composition-------------------------- 19 -- Supported Drivers and Software-------------- 20 -- Product and LED Control buttons------------ 21 LED Control button Functions Connectors -- Installing the Monitor------------------------- 25 Cable connection and organize Installing on the Wall SETTINGS----------------------------- 28 -- Activate Main Menu--------------------------- 28 QUICK SETTINGS INPUT PICTURE GENERAL TROUBLESHOOTING----------------- 38 PRODUCT SPECIFICATIONS--------- 40 -- Preset Modes---------------------------------- 40 DisplayPort HDMI DVI -- Extension Timing ----------------------------- 42 -- Power indicator-------------------------------- 42 EXTERNAL CONTROLLER SETUP--- 43 19 ASSEMBLY AND PREPARING ENGLISH Product Composition Please check whether all the components are included in the box before using the product. If there are any missing components, contact the retailer where you purchased the product. Note that the product and related components may look different from those shown here. CD (Software/Owner's Manual)/ Regulatory Manual/Cards Power Cord DisplayPort Cable HDMI Cable AC/DC Adaptor CAUTION •• Always use genuine LG components to ensure safety and product performance. •• The product warranty will not cover damage or injury caused by the use of unauthorized components. •• It is recommend that use the supplied components. •• If you use generic cables not certified by LG, the screen may not display or there may be image noises. •• Need to use the authorized components about the below accessories. Unauthorized components may be cause of the damage and malfunction of the product. Component Standard HDMI Cable Support HDMI 2.0 version, UL, Impedance 100ohm DisplayPort Cable Support DISPLAYPORT 1.2a version, UL, Impedance 100ohm Power Cord US – Approved Medical grade regulation Others – Approved country safety regulation The AC/DC adaptors and etc. except the upper components need to be used only supplied by manufacturer. 20 NOTE ENGLISH •• The components may look different from those illustrated here. •• Without prior notice, all product information and specifications contained in this manual are subject to change to improve the performance of the product. •• To purchase optional accessories, visit an electronics store or an online shopping site, or contact the retailer from which you purchased the product. •• The power cord provided may differ depending upon the region. Supported Drivers and Software Check the drivers and software supported by your product and refer to the manuals on the CD enclosed in the product package. Installation Priority 27HJ710S Monitor Driver Drivers and Software Recommended O True Color Pro Optional O •• Required and Recommended: You can download and install the latest version from the enclosed CD or from the LGE website (www.lg.com). •• Optional: You can download and install the latest version from the LGE website (www.lg.com). 21 Product and LED Control buttons ENGLISH 1 2 3 4 5 6 7 8 9 22 LED Control button Functions ENGLISH 1 Input Selects the input mode. 2 PBP/PIP Displays the screens of two input modes on one monitor. 3 Hotkey 1, 2 Opens the Hotkey Settings menu. 4 Brightness Adjusts the screen brightness. 5 OK Selects menus or options and confirms. 6 Contrast Adjusts the screen contrast. 7 Menu Displays LED Control buttons on the front panel and opens the menu mode. 8 Power The green indicator illuminates if the power is on. The arrow indicates the position of the indicator power switch. Power switch Turns the power on/off. 9 NOTE •• The Power button is located at the bottom right on the front of the monitor. •• If the Control Key LED is turned off, press the Control Key [Menu] button to turn the Control button LED on. When the Control Key LED is turned on, you can control the Control Key functions. 23 Connectors ENGLISH DC-IN (19 V ) RS-232C DC-IN (19 V DC-IN (19 V 1 1 2 3 4 5 ) RS-232C DC-IN (19 V ) 2 RS-232C SDI IN RS-232C ) SDI IN SDI OUT SDI IN SDI OUT SDI IN SDI OUT 3 DVI-D IN SDI OUT DVI-D OUT DVI-D IN DVI-D OUT DVI-D IN DVI-D IN HDMI IN HDMI IN DVI-D OUT DVI-D OUT 4 DP IN DP IN DP OUT HDMI IN DP OUT SERVICE ONLY 5V HDMIDPININ 5 SERVICE ONLY 5V 0.9 A 0.9 A DP IN DP OUT DP OUT SERVICE ONLY 6 5V SERVICE ONLY 0.9 A 7 5V 0.9 A 8 DC-IN (19 V ) Connector -- Connect the AC/DC adaptor. RS-232C Connector -- Connect to an external device through the RS-232C connector and control the monitor. SDI IN / SDI OUT Connector -- Receives or transmits serial digital component signals. DVI-D IN / DVI-D OUT Connector -- Receives or transmits digital video signals. HDMI IN Connector -- Receives digital video signals. -- DVI to HDMI/DP (DisplayPort) to HDMI cables may cause compatibility issues. Make sure to use certified cables that bear the HDMI logo. If you do not use a certified HDMI cable, the screen may not display, or a connection error may occur. ►►Recommended HDMI cable types -- High-speed HDMI®/TM cables -- High-speed HDMI®/TM Ethernet cables 24 6 ENGLISH 7 8 DP IN / DP OUT Connector -- Receives or transmits digital video signals. -- There may be no video output depending on the DP (DisplayPort) version of your PC. -- A cable with DisplayPort 1.2 specifications is recommended when using a Mini DP to DP (Mini DisplayPort to DisplayPort) cable. SERVICE ONLY Connector -- The USB port is used for service only. 5V 0.9 A (USB Connector) -- Used for HW Calibrator (Optional Accessory) connection. -- A keyboard, mouse or USB device can be connected. CAUTION Precautions for Using USB Storage Devices •• A USB storage device installed using an automatic recognition program or its own driver may not be recognizable. •• Some USB devices may not be supported or may not work properly. •• It is recommended that you use a USB hub or hard disk drive with power supplied. (If insufficient power is supplied, the USB storage device may not be detected properly) 5V 0.9 A / (USB Connector) -- Connect your accessory to the USB input port. -- To use USB 3.0, connect a USB 3.0 Type A-B cable to your PC. NOTE •• All signal OUT connectors (SDI, DVI, DP) output signals when the monitor's power switch is on. Signals are not output when the power switch is off. •• This monitor supports the *Plug and Play feature. *Plug and Play: A feature that allows you to add a device to your computer without having to reconfigure anything or install any manual drivers. 25 Installing the Monitor 3 After tidying the cables, attach the back door and use the product. ENGLISH Cable connection and organize Before connecting the connectors, remove the back door as shown below. 1 Remove the back door by lifting it in the direction of the arrow. The back door is held in place with a magnet. CAUTION •• When the back door is attached to the monitor, the monitor meets water resistance standards. Do not use the monitor with the back door detached, as the water resistance capability is not guaranteed without the door. 2 After installing the cables, tidy them up using the cable holders. 26 Installing on the Wall ENGLISH Install the monitor at least 100 mm away from the wall on each side of the monitor to ensure sufficient ventilation. Detailed installation instructions can be obtained from your local dealer. Please refer to the manual to install and set up a tilting wall mounting bracket. Wall Mount (mm) Standard screw Required screws Wall Mount Plate (optional) 100 x 100 M4 x L10 4 RW120 200 x 100 M4 x L10 4 RW240 100 mm NOTE 100 mm 100 mm 100 mm •• The screws (M4 x L10) can be found fastened in the wall mount screw holes on the back of the monitor. •• Use the screws specified by VESA standards. •• The wall mount kit includes an installation manual and the necessary parts. •• The wall mount plate is an optional item. You can To install the monitor to a wall, attach a wall mount plate (optional) to the back of the monitor. Make sure that the wall mount plate (optional) is securely fixed to the monitor and to the wall. 1 Using screws longer than the standard length may damage the inside of the product. 2 A non-VESA standard screw may damage the product and cause the monitor to fall. LG Electronics is not liable for any accidents relating to the use of non-standard screws. obtain additional accessories from your local dealer. •• The length of screws may differ depending on the wall mount. Be sure to use the proper length. •• For more information, please refer to the manual for the wall mounting bracket. 27 Precautions for Connecting the Power Cord CAUTION ENGLISH •• Disconnect the power cord first. Then move or install the monitor. There is risk of electric shock. •• Installing the monitor on the ceiling or on a slanted wall may result in the monitor falling off, which could lead to injury. Use an authorized LG wall mount and contact the local dealer or qualified personnel. •• Applying excessive force when tightening screws may damage the monitor. Such damage is not covered by the product warranty. •• To prevent injury, this apparatus must be securely attached to the wall in accordance with the installation instructions. (This only pertains to Australia and New Zealand.) •• Use the wall mounting bracket and screws that confirm to VESA standards. Damage caused by the use or misuse of inappropriate components is not covered by the product warranty. •• When measured from the back of the monitor, the length of each installed screw must be 8 mm or less. Wall mount plate Back of the monitor Wall mount plate Back of the monitor Screw specifications : M4 x L10 Max. 8 mm 100-240 V ~ •• Make sure to use the power cord that is provided in the product package. Connect the cord to a grounded power outlet. •• If you need another power cord, please contact your local dealer or the nearest retail store. 28 SETTINGS ENGLISH Activate Main Menu 1 Press the [Menu] button to activate the LED Control button. Pressing the [Menu] button while the LED Control button is activated brings up the OSD menu. 2 Use the [ Brightness 3 Press the [Menu] or [ ] left/right and [ Contrast ] up/down LED Control buttons to navigate. Brightness] LED Control button to exit the OSD menu. CAUTION ••The actual On Screen Displays (OSDs) that appear on your monitor may differ from those shown in this manual. QUICK SETTINGS Quick Settings Brightness Quick Settings Input 100 > Contrast 70 > Hot key Settings > User Preset > Picture General : Exit / : Move / : Ok 29 The following settings are available. Description Adjusts the brightness of the screen. NOTE •• You can turn the Brightness Stabilization to On/ Off by pressing the button in the Brightness menu. •• When the Brightness Stabilization is On, Brightness menus becomes disabled. •• When the Brightness Stabilization is On, SMART ENERGY SAVING and DFC menus become disabled. Contrast Adjusts the color contrast of the screen. Hot key Settings Assigns hotkeys for picture settings. Once configured, the assigned menu is activated when a hotkey is pressed. (PIP Size, Mono, Color Temp, Gamma, Black Stabilizer, Screen Zoom, Off) Hotkey 1 Assigns a function to Hotkey 1. Hotkey 2 Assigns a function to Hotkey 2. User name can be configured using the on-screen keyboard. User Name Assigns a user name to USER 1-10. Load User Settings Loads settings stored in USER 1-10. Save User Settings Saves the current settings to USER 1-10. Default User Settings Loads the default settings. User Preset ENGLISH Quick Settings Brightness 30 INPUT ENGLISH <Default screen> <Main screen and sub screen> Input Input Quick Settings Quick Settings DisplayPort MAIN DisplayPort Input Input Main Input List Picture DisplayPort > Aspect Ratio Full Wide > PBP / PIP General DVI-D > PIP Size Small > Main/Sub Screen Change : Exit Picture / Input List > Aspect ratio > PBP / PIP General > PIP Size Small > Main/Sub Screen Change : Move / : Ok : Exit / : Move / : Ok The following settings are available. Input Main Input List Input List Aspect Ratio Description Selects the input mode. Adjusts the aspect ratio of the screen. (Full Wide, Original, 1:1) NOTE •• The display may look the same for Full Wide, Original, and 1:1 options at the recommended resolution (3840 x 2160). Main Full Wide Displays the video to fit the PBP / PIP screen, regardless of the video signal input. Original Displays the video in the aspect ratio of the video signal input on the PBP / PIP screen. Sub Full Wide Displays the video to fit the PBP / PIP screen, regardless of the video signal input. Original Displays the video in the aspect ratio of the video signal input on the PBP / PIP screen. 31 Input Description Displays the screens of two input modes on one monitor. PIP Size Adjusts the PIP size. (Small, Medium, Large) Main/Sub Screen Change Toggles between the main screen and the sub screen in PBP / PIP mode. NOTE •• When there is no Sub Screen signal, PIP Size and Main/Sub Screen Change are disabled. ENGLISH PBP / PIP 32 PICTURE ENGLISH Picture Picture Mode Quick Settings Input Custom > Picture Adjust > Color Adjust > Picture Reset > Picture General : Exit / : Move / : Ok The following settings are available. Picture Picture Mode Description Custom Mono sRGB EBU Allows the user to adjust each element. The color mode of the main menu can be adjusted. Monochrome (black and white) color mode. A standard RGB color mode for the monitor and printer. A standard TV PAL color mode for broadcasts. REC709 A standard HDTV color mode for broadcasts. SMPTE-C DICOM A standard TV NTSC color mode for broadcasts. A mode optimized for viewing medical images. You can brighten the screen in the OSD menu. Adjusts to the last calibrated screen. Adjusts to the previously calibrated screen. Calibration 1 Calibration 2 NOTE •• Brightness Stabilization function can be enabled in Custom mode. •• If the Picture Mode is changed in the DisplayPort (DP) input, the screen may flicker or the resolution of your PC screen may be affected. •• Calibration 2: This menu is enabled if you install the TRUE COLOR PRO program and perform a calibration. 33 Picture Description Adjusts the brightness of the screen. NOTE Brightness •• You can turn the Brightness Stabilization to On/ Off by pressing the button in the Brightness menu. •• When the Brightness Stabilization is On, Brightness menus becomes disabled. •• When the Brightness Stabilization is On, SMART ENERGY SAVING and DFC menus become disabled. Adjusts the color contrast of the screen. Adjusts the sharpness of the screen. Maintains the brightness configured for medical environments. On Adjusts the brightness automatically. Off Deactivates the function, and allows the user to configure the brightness. SUPER RESOLUTION+ High Select this option for crystal clear images. Best for high-definition video or games. Middle The optimized picture quality is displayed when a user wants images between low and high modes for comfortable viewing. Best for UCC or SD video. Low The optimized picture quality is displayed when a user wants smooth and natural images. Best for still images with less movement. Off Select this option for the normal user experience. Disables SUPER RESOLUTION+. Black Level Sets the offset level. (HDMI only) •• Offset: A reference for video signals. Offset is the darkest color that the monitor can display. High Keeps the current contrast ratio of the screen. Low Lowers the black levels and raises the white levels from the current contrast ratio of the screen. Contrast Sharpness Brightness Stabilization ENGLISH Picture Adjust 34 Picture ENGLISH Picture Adjust Description HDMI ULTRA HD Deep Provides sharper images when connected to a device supporting Color ULTRA HD Deep Color. NOTE ••If the connected device does not support ULTRA HD Deep Color, the feature may not work properly. ••If you experience a problem, set HDMI ULTRA HD Deep Color to Off. Formats supported at 4K@ 60 Hz1) Deep color / chroma sampling Frame rate Resolution (Hz) 8-bit 10-bit YCbCr 4:2:0 59.94 YCbCr 4:2:2 3840 x 2160 YCbCr 4:4:4 60.00 RGB 4:4:4 1) Supported when HDMI ULTRA HD Deep Color is set to ON. DFC On Off Response Time Black Stabilizer Adjusts the brightness automatically according to the screen. Disables the DFC feature. Sets the response time for displayed pictures based on the speed of the screen. For a normal environment, Normal is recommended. For a fastmoving picture, Fast is recommended. Setting the response time to Fast may cause image sticking. Fast Sets the response time to fast. Normal Sets the response time to normal. Slow Sets the response time to slow. Off Does not use the response time improvement feature. Objects can be seen more clearly in dark screen picture by adjusting the black level. Increasing the Black Stabilizer value brightens the low gray levels on the screen. (You can easily distinguish objects on a dark game screen.) Reducing the Black Stabilizer value darkens the low gray levels and increases the dynamic contrast on the screen. 35 Picture Description Uniformity Adjusts the uniformity of screen brightness. NOTE ••Enabling Uniformity may reduce the overall brightness of the Color Adjust Gamma Color Temp image. On Enables the Uniformity feature. Off Disables the Uniformity feature. Custom gamma settings: Gamma 1.8, Gamma 2.0, Gamma 2.2, Gamma 2.4, Gamma 2.6, DICOM Gamma Curve. The higher the gamma value, the darker the image becomes. Likewise, the lower the gamma value, the lighter the image becomes. Adjusts the DICOM Gamma. (Custom, 6500K, 8500K, 9300K, Manual) Custom The user can customize the red, green and blue color. 6500K Sets the picture color to the color temperature of a reddish 6500K. 8500K Sets the picture color to the color temperatures of 8500K, setting to a color between red and blue. 9300K Sets the picture color to the color temperature of a bluish 9300K. Manual Adjusts the color temperature in 500K increments. (Note that 9300K is supported rather than 9500K.) Red Green Blue Six Color Picture Reset You can customize the picture color using red, green and blue. Fulfills your color requirements by adjusting the hue and saturation of the six colors (red, green, blue, cyan, magenta and yellow). The settings are then saved. Hue Adjusts the tone of the screen colors. Saturation Adjusts the saturation of the screen colors. The lower the value, the less saturated and brighter the colors become. The higher the value, the more saturated and darker the colors become. Returns the colors to the default settings. ENGLISH Picture Adjust 36 GENERAL ENGLISH General Language Quick Settings English > SMART ENERGY SAVING Off > HW Calibration On > RS-232C Input > LED Control Button 20Sec Time Out > DVI Power Supply Picture Off > Hot key Settings > User Preset General > Auto Screen Off On > DisplayPort 1.2 Enable > OSD Lock Off > Reset : Exit > / : Move / : Ok The following settings are available. General Description Language Sets the menu screen to the desired language. SMART ENERGY SAVING Conserve energy by using luminance compensation algorithm. High Saves energy using the high-efficiency SMART ENERGY SAVING feature. Low Saves energy using the low-efficiency SMART ENERGY SAVING feature. Off Disables the SMART ENERGY SAVING feature. RS-232C and HW Calibration cannot be used together simultaneously. HW Calibration RS-232C LED Control Button On Enables the HW Calibration feature. Off Disables the HW Calibration feature. RS-232C and HW Calibration cannot be used together simultaneously. Serial Port Enables or disables the RS-232C feature. Set ID Configures the Set ID. (Configuration range: 1-10) Adjusts how long the control button LEDs remain lit. (Always On, 20Sec Time Out, 10Sec Time Out, 5Sec Time Out) 37 General Hot key Settings User Preset Auto Screen Off DisplayPort 1.2 Description Supplies power to devices connected to the DVI input port as a dongle without power. On Activates the DVI Power Supply feature. Off Deactivates the DVI Power Supply feature. Assigns hotkeys for picture settings. Once configured, the assigned menu is activated when a hotkey is pressed. (PIP Size, Mono, Color Temp, Gamma, Black Stabilizer, Screen Zoom, Off) Hotkey 1 Assigns a function to Hotkey 1. Hotkey 2 Assigns a function to Hotkey 2. User name can be configured using the on-screen keyboard. User Name Assigns a user name to USER 1-10. Load User Settings Loads settings stored in USER 1-10. Save User Settings Saves the current settings in USER 1-10. Default User Settings Loads the default settings. The screen turns off automatically when there are no monitor signals for a set period of time. On Enables the Auto Screen Off feature. Off Disables the Auto Screen Off feature. Enables or disables DisplayPort 1.2. NOTE •• Be sure to configure this option according to the DisplayPort version supported by your graphics card. Set this option to Off if your graphics card is not compatible. •• When DisplayPort 1.2 is disabled, you cannot use the 10-bit output of your graphics card. OSD Lock This feature disables the configuration and adjustment of menus. On Enables the OSD lock feature. Off Disables the OSD lock feature. NOTE •• All features other than the Quick Settings menu, the Input List, Aspect Ratio and PBP/ PIP features in the Input menu and the OSD Lock feature in the General menu are disabled. Reset Do you want to reset your settings? Yes No Restores the default settings. Cancels the reset. ENGLISH DVI Power Supply 38 TROUBLESHOOTING ENGLISH Nothing is displayed on the screen. Is the monitor's power cord plugged in? •• Check if the power cord is correctly plugged into the power outlet. Is the power indicator on? •• Check your power cable connection and turn the power switch on. Is the power indicator displaying as green? •• Check if the input setting is correct when the PC and the product are connected. (MENU > Quick Settings > Input) Is the "Out of Range" message •• This occurs when the signals transferred from your PC (graphics card) displayed? are outside the horizontal or vertical frequency range of your monitor. Please see the "Product Specification" section of this manual to set the appropriate frequency. Is the 'No Signal' message displayed? •• This is displayed when the signal cable between the PC and the monitor is missing or disconnected. Check the cable and reconnect it. The screen may not be displayed properly when configuring DisplayPort (DisplayPort) 1.2 in a Mac product. The screen retains an image. Does image sticking occur even when the monitor is turned off? ••DisplayPort 1.2 may not be supported depending on the Mac product. Please contact the manufacturer. •• Displaying a still image for a prolonged period may damage the screen, resulting in image sticking. •• To extend the lifetime of the monitor, use a screensaver. The screen display is unstable and shaky. Images displayed on the monitor leave shadow trails. Did you select the appropriate resolution? •• If the selected resolution is HDMI 1080i 60/50 Hz (interlaced), the screen may be flickering. Change the resolution to 1080P or the recommended resolution. NOTE ••Vertical Frequency: In order to display an image, the screen must be refreshed dozens of times per second, like a fluorescent lamp. The number of times the screen is refreshed per second is called vertical frequency, or refresh rate, and is represented by Hz. ••Horizontal Frequency: The time it takes to display one horizontal line is called the horizontal cycle. If 1 is divided by the horizontal interval, the result is the number of horizontal lines displayed per second. This is called horizontal frequency and is represented by kHz. 39 NOTE graphics card to the recommended (optimal) resolution in Control Panel > Display > Settings in Windows. (The settings may differ depending on your operating system) •• Not setting your graphics card to the recommended (optimal) resolution may result in blurred text, a dimmed screen, a truncated display area or misalignment of the display. •• The setting method may differ depending on the computer and operating system. Some resolutions may not be available depending on your graphics card's performance. If this is the case, contact the manufacturer of your computer or graphics card for help. •• Some graphics cards may not support the 3840 x 2160 resolution. If the resolution cannot be displayed, contact the manufacturer of your graphics card. The display color is abnormal. Does the display appear discolored (16 colors)? Does the display color appear unstable or monochrome? Are there spots on the screen? •• Set the number of colors to 24-bits (True Color) or above: Control Panel > Display > Settings > Color Quality in Windows (the settings may differ depending on your operating system). •• Check the connection status of the signal cable and connect it properly or reinsert your PC's graphics card. •• When using the monitor, pixelated spots (red, green, blue, white or black) may appear on the screen. This is normal for an LCD screen. It is not an error, nor is it related to the monitor's performance. The 'Unknown Monitor' message appears when the monitor is connected. Did you install the monitor driver? •• Install the enclosed monitor driver provided with the monitor or go to the LG Electronics homepage (http://www.lg.com) to download and install the monitor driver. •• Check your graphics card user manual to see if the Plug and Play function is supported. ENGLISH •• Check if your graphics card's resolution or frequency is within the range supported by your monitor and set your 40 PRODUCT SPECIFICATIONS ENGLISH The product specifications are subject to change without prior notice for product improvements. Preset Modes DisplayPort Preset Modes Horizontal Frequency (kHz) Vertical Frequency (Hz) 640 x 480 31.469 59.94 -/- 800 x 600 37.879 60.317 +/+ 1024 x 768 48.363 60 -/- 1152 x 864 54.347 60.05 +/+ Polarity (H/V) 1280 x 720 45 60 +/+ 1280 x 1024 63.981 60.02 +/+ 1600 x 900 60 60 +/+ 1920 x 1080 67.5 60 +/- 2560 x 1440 88.79 59.95 +/- 3840 x 2160 66.66 30 +/- 3840 x 2160 133.32 60 +/- Remarks 41 HDMI Horizontal Frequency (kHz) Vertical Frequency (Hz) 640 x 480 31.469 59.94 -/- 800 x 600 37.879 60.317 +/+ 1024 x 768 48.363 60 -/- 1152 x 864 54.347 60.05 +/+ 1280 x 720 45 60 +/+ 1280 x 1024 63.981 60.02 +/+ 1600 x 900 60 60 +/+ 1920 x 1080 67.5 60 +/- 2560 x 1440 88.79 59.95 +/- 3840 x 2160 67.5 30 +/- 3840 x 2160 135 60 +/- Preset Modes Horizontal Frequency (kHz) Vertical Frequency (Hz) Polarity (H/V) 640 x 480 31.469 59.94 -/- 800 x 600 37.879 60.317 +/+ 1024 x 768 48.363 60 -/- 1152 x 864 54.347 60.05 +/+ Polarity (H/V) Remarks When the HDMI ULTRA HD Deep Color is set to On DVI 1280 x 720 45 60 +/+ 1280 x 1024 63.981 60.02 +/+ 1600 x 900 1920 x 1080 60 67.5 60 60 +/+ +/- Remarks ENGLISH Preset Modes 42 Extension Timing ENGLISH Vertical dimensions Vertical Frequency (Hz) DVI HDMI DP SDI 480i 480p 576p 576i 720p 720p 1080i 1080p 1080i 1080p 1080p 1080p 2160p 2160p 2160p 2160p 2160p 59.94/60 59.94/60 50 50 59.94/60 50 59.94/60 59.94/60 50 50 23.98/24 29.97/30 23.98/24 25 29.97/30 50 59.94/60 O O O O O O O O - O O O O O O O O O O O O O O O O O O - O O O O O O O O O - Power indicator Mode LED Color On Mode Green 43 EXTERNAL CONTROLLER SETUP Use an RS-232C cable to remotely control the monitor (see Figure 1). RS-232C (PC) (Monitor) Figure 1: RS-232 Connection Diagram RXD TXD DTR GND DSR RTS CTS 2 3 4 5 6 7 8 PC 2 3 4 5 6 7 8 RXD TXD DTR GND DSR RTS CTS Monitor * There are no connections to Pin 1 and Pin 9. 5 9 1 6 ENGLISH The actual product may differ from the picture shown. Connect the RS-232C (serial connector) of the PC to the RS-232C IN connector at the back of the projector. Purchase a cable to connect the RS-232C connectors, as the cable is not provided as an accessory. 44 Set ID Function ENGLISH This function allows you to assign a unique ID to the monitor to control it remotely from your PC. For "Actual Data Structure (Base 16 • Base 10)", see page 47 1 Press the [Menu] button. 2 Use the , , , buttons to select General. Then press the OK button. 3 Use the , , , buttons to select Set ID. Then press the OK button. 4 Select a Set ID to assign. Then press the OK button. The Set ID can be a value from 1 to 10. 5 Press the EXIT button to exit (pressing the [Menu] button hides the OSD). Communication Parameters •• Baud rate: 9600 bps (UART) •• Data length: 8 bits •• Parity bit: None •• Stop bit: 1 bit •• Communication code: ASCII code •• Crossed (reverse) cable used. 45 Command Reference List Command2 01. Power 02. Screen Mute 03. Input select (Main) 04. Input select (Sub) 05. Aspect Ratio (Main) 06. Aspect Ratio (Sub) k k k k k k a d b y c o Data (Hexadecimal) 00 - 01 00 - 01 00 - 03 00 - 03 00 - 02 00 - 01 07. PBP/PIP k n 00 - 05 08. PIP Size 09. Main/Sub Screen Change 10. Picture Mode 11. Brightness 12. Contrast 13. Sharpness 14. Brightness Stabilization k m d k k k m p a x h g k b 00 - 02 01 00 - 08 00 - 64 00 - 64 00 - 64 00 - 01 15. SUPER RESOLUTION+ m c 00 - 03 16. Black Level 17. HDMI Ultra HD Deep Color 18. DFC 19. Response Time 20. Black Stabilizer 21. Uniformity 22. Gamma 23. Color Temp 24. Red Gain 25. Green Gain 26. Blue Gain 27. Language 28. SMART ENERGY SAVING 29. LED Control Button 30. DVI Power Supply 31. Auto Screen Off 32. DisplayPort 1.2 33. OSD Lock 34. Reset m m m m m m m k j j j f m m m m m k f d e f g h i j u w y z i k l m n o m k 00 - 01 00 - 01 00 - 01 00 - 03 00 - 64 00 - 01 00 - 05 00 - 04 00 - 64 00 - 64 00 - 64 00 - 10 00 - 02 00 - 03 00 - 01 00 - 01 00 - 01 00 - 01 00 - 01 ENGLISH Command1 Title 46 Transmission/Reception Protocol ENGLISH Transmission [Command1][Command2][ ][Set ID][ ][Data][Cr] [Command 1]: j, k, m, x [Command 2]: This command is used to control the monitor. [Set ID]: Used to identify the monitor being controlled. [Set ID] can be assigned to each monitor under GENERAL in the Settings Menu. A value from 1 to 10 can be assigned. By selecting '0' for the [Set ID] value in the protocol format, you can control all connected monitors. *The value is displayed as base 10 on the OSD menu and used as base 16 (0x00 - 0x63) in the transmission/reception protocol for remote control. [Data]: Transmits a setting value (Data) required for the command described previously. (base 16) When the Data ‘FF’ is sent, the setting value corresponding to the specific command is read (Data read mode). [Cr]: Carriage return, which is ‘0x0D’ in ASCII code. [ ]: Space, which is ‘0x20’ in ASCII code. OK Acknowledgment [Command2][ ][Set ID][ ][OK][Data][x] When the Data has been successfully received, the monitor sends an ACK response signal in the above format. The Data showing the current state is received in the Data read mode. The Data from the PC is simply returned in the Data write mode. 47 Error Acknowledgment When the set receives an abnormal piece of Data for an unsupported function or there is a communication error, it returns ACK in the format above. Data 00 : Illegal code Actual Data Structure (Base 16 • Base 10) •• See the table below when inserting a base-16 value in [Data]. •• The channel setup command (ma) uses a 2-byte base-16 value ([Data]) for channel number input. 00: Step 0 01: Step 1 (Set ID 1) ... 0A: Step 10 (Set ID 10) ... 0F: Step 15 (Set ID 15) 10: Step 16 (Set ID 16) ... 32: Step 50 (Set ID 50) 33: Step 51 (Set ID 51) ... 63: Step 99 (Set ID 99) ... C7: Step 199 C8: Step 200 ... * Commands may work differently depending on the model and signal. FE: Step 254 FF: Step 255 ... 01 00: Step 256 ... 27 0E: Step 9998 27 0F: Step 9999 ... ENGLISH [Command2][ ][Set ID][ ][NG][Data][x] 48 01. Power (Command: k a) ENGLISH ▶▶ This controls the monitor power on/off. Transmission [k][a][ ][Set ID][ ][Data][Cr] Data 00: Power Off 01: Power On Ack [a][ ][Set ID][ ][OK/NG][Data][x] 02. Screen Mute (Command: k d) ▶▶ This controls the monitor display power on/off. Transmission [k][d][ ][Set ID][ ][Data][Cr] Data 00: Screen Mute Off 01: Screen Mute On Ack [c][ ][Set ID][ ][OK/NG][Data][x] 03. Input select (Main) (Command: k b) ▶▶ This controls the input mode of the main screen. Transmission [k][b][ ][Set ID][ ][Data][Cr] Data 00: SDI 02: HDMI 01: DVI 03: DisplayPort Ack [b][ ][Set ID][ ][OK/NG][Data][x] 04. Input select (Sub) (Command: k y) ▶▶ This controls the input mode of the sub screen. Transmission [k][y][ ][Set ID][ ][Data][Cr] Data 00: SDI 02: HDMI Ack [y][ ][Set ID][ ][OK/NG][Data][x] 01: DVI 03: DisplayPort 49 05. Aspect Ratio (Main) (Command: k c) Transmission [k][c][ ][Set ID][ ][Data][Cr] Data 00: Full Wide 02: 1:1 01: Original Ack [c][ ][Set ID][ ][OK/NG][Data][x] 06. Aspect Ratio (Sub) (Command: k o) ▶▶ This adjusts the aspect ratio of the sub screen. Transmission [k][o][ ][Set ID][ ][Data][Cr] Data 00: Full Wide 01: Original Ack [o][ ][Set ID][ ][OK/NG][Data][x] 07. PBP/PIP (Command: k n) ▶▶ This controls the PBP/PIP mode. Transmission [k][n][ ][Set ID][ ][Data][Cr] Data 00: Full Screen 02: PIP 1 04: PIP 3 01: 5:5 03: PIP 2 05: PIP 4 Ack [n][ ][Set ID][ ][OK/NG][Data][x] 08. PIP Size (Command: k p) ▶▶ This adjusts the PIP size. Transmission [k][p][ ][Set ID][ ][Data][Cr] Data 00: Small 02: Large Ack [p][ ][Set ID][ ][OK/NG][Data][x] 01: Medium ENGLISH ▶▶ This adjusts the aspect ratio of the main screen. 50 09. Main/Sub Screen Change (Command: m a) ENGLISH ▶▶ This controls the main/sub screen change in PBP mode. Transmission [m][a][ ][Set ID][ ][Data][Cr] Data 01: Main/Sub Screen exchange Ack [a][ ][Set ID][ ][OK/NG][Data][x] 10. Picture Mode (Command: d x) ▶▶ This controls the picture mode. Transmission [d][x][ ][Set ID][ ][Data][Cr] Data 00: Custom 02: sRGB 01: Mono 03: EBU 04: REC709 05: SMPTE-C 06: DICOM 07: Calibration 1 08: Calibration 2 Ack [x][ ][Set ID][ ][OK/NG][Data][x] 11. Brightness (Command: k h) ▶▶ This adjusts the screen brightness. Transmission [k][h][ ][Set ID][ ][Data][Cr] Data Min: 00 - Max: 64 Ack [h][ ][Set ID][ ][OK/NG][Data][x] 12. Contrast (Command: k g) ▶▶ This adjusts the screen contrast. Transmission [k][g][ ][Set ID][ ][Data][Cr] Data Min: 0 0 - Max: 64 Ack [g][ ][Set ID][ ][OK/NG][Data][x] 51 13. Sharpness (Command: k k) Transmission [k][k][ ][Set ID][ ][Data][Cr] Data Min: 00 - Max: 64 Ack [k][ ][Set ID][ ][OK/NG][Data][x] 14. Brightness Stabilization (Command: m b) ▶▶ This adjusts the Brightness Stabilization feature. Transmission [m][b][ ][Set ID][ ][Data][Cr] Data 00: Off 01: On Ack [b][ ][Set ID][ ][OK/NG][Data][x] 15. SUPER RESOLUTION+ (Command: m c) ▶▶ This controls the SUPER RESOLUTION+ feature. Transmission [m][c][ ][Set ID][ ][Data][Cr] Data 00: High 02: Low 01: Middle 03: Off Ack [c][ ][Set ID][ ][OK/NG][Data][x] 16. Black Level (Command: m d) ▶▶ This controls the offset level (for HDMI only). Transmission [m][d][ ][Set ID][ ][Data][Cr] Data 00: High 01: Low Ack [d][ ][Set ID][ ][OK/NG][Data][x] 17. HDMI ULTRA HD Deep Color (Command: m e) ▶▶ This controls the Ultra Deep Color feature. (HDMI only) Transmission [m][e][ ][Set ID][ ][Data][Cr] Data 00: On 01: Off Ack [e][ ][Set ID][ ][OK/NG][Data][x] ENGLISH ▶▶ Adjusts the sharpness of the screen. 52 18. DFC (Command: m f) ENGLISH ▶▶ This controls the DFC feature. Transmission [m][f][ ][Set ID][ ][Data][Cr] Data 00: On 01: Off Ack [f][ ][Set ID][ ][OK/NG][Data][x] 19. Response Time (Command: m g) ▶▶ This controls the response time. Transmission [m][g][ ][Set ID][ ][Data][Cr] Data 00: Fast 02: Slow 01: Normal 03: Off Ack [g][ ][Set ID][ ][OK/NG][Data][x] 20. Black Stabilizer (Command: m h) ▶▶ This controls the Black Stabilizer feature. Transmission [m][h][ ][Set ID][ ][Data][Cr] Data Min: 0 0 - Max: 64 Ack [h][ ][Set ID][ ][OK/NG][Data][x] 21. Uniformity (Command: m i) ▶▶ This controls the Uniformity Calibration feature. Transmission [m][i][ ][Set ID][ ][Data][Cr] Data 00: On 01: Off Ack [i][ ][Set ID][ ][OK/NG][Data][x] 22. Gamma (Command: m j) ▶▶ This adjusts the Gamma settings. Transmission [m][j][ ][Set ID][ ][Data][Cr] Data 00: Gamma 1.8 02: Gamma 2.2 04: Gamma 2.6 Ack [j][ ][Set ID][ ][OK/NG][Data][x] 01: Gamma 2.0 03: Gamma 2.4 05: DICOM Gamma Curve 53 23. Color Temp (Command: k u) Transmission [k][u][ ][Set ID][ ][Data][Cr] Data 00: Custom 02: 8500K 04: Manual 01: 6500K 03: 9300K Ack [u][ ][Set ID][ ][OK/NG][Data][x] 24. Red Gain (Command: j w) ▶▶ This adjusts the red gain. Transmission [j][w][ ][Set ID][ ][Data][Cr] Data Min: 00 - Max: 64 Ack [w][ ][Set ID][ ][OK/NG][Data][x] 25. Green Gain (Command: j y) ▶▶ This adjusts the green gain. Transmission [j][y][ ][Set ID][ ][Data][Cr] Data Min: 00 - Max: 64 Ack [y][ ][Set ID][ ][OK/NG][Data][x] 26. Blue Gain (Command: j z) ▶▶ This adjusts the blue gain. Transmission [j][z][ ][Set ID][ ][Data][Cr] Data Min: 00 - Max: 64 Ack [z][ ][Set ID][ ][OK/NG][Data][x] 27. Language (Command: f i) ▶▶ This adjusts the language of the OSD. Transmission [f][i][ ][Set ID][ ][Data][Cr] English - Korean (17 languages) Ack [i][ ][Set ID][ ][OK/NG][Data][x] ENGLISH ▶▶ This adjusts the screen color temperature. 54 28. SMART ENERGY SAVING (Command: m k) ENGLISH ▶▶ This adjusts the SMART ENERGY SAVING feature. Transmission [m][k][ ][Set ID][ ][Data][Cr] Data 00: High 02: Off 01: Low Ack [k][ ][Set ID][ ][OK/NG][Data][x] 29. LED Control Button (Command: m l) ▶▶ Adjusts how long the control button LEDs remain lit. Transmission [m][l][ ][Set ID][ ][Data][Cr] Data 00: Always On 02: 10sec time out 01: 20sec time out 03: 5sec time out Ack [l][ ][Set ID][ ][OK/NG][Data][x] 30. DVI Power Supply (Command: m m) ▶▶ This controls the DVI Power Supply feature. Transmission [m][m][ ][Set ID][ ][Data][Cr] Data 00: On 01: Off Ack [m][ ][Set ID][ ][OK/NG][Data][x] 31. Auto Screen Off (Command: m n) ▶▶ This adjusts the time after which the screen turns off automatically when there are no monitor signals. Transmission [m][n][ ][Set ID][ ][Data][Cr] Data 00: On 01: Off Ack [n][ ][Set ID][ ][OK/NG][Data][x] 32. DisplayPort 1.2 (Command: m o) ▶▶ This enables or disables DisplayPort 1.2. Transmission [m][o][ ][Set ID][ ][Data][Cr] Data 00: Enable 01: Disable Ack [o][ ][Set ID][ ][OK/NG][Data][x] 55 33. OSD Lock (Command: k m) Transmission [k][m][ ][Set ID][ ][Data][Cr] Data 00: Off 01: On Ack [m][ ][Set ID][ ][OK/NG][Data][x] 34. Reset (Command: f k) ▶▶ This controls the reset operation. Transmission [f][k][ ][Set ID][ ][Data][Cr] Data 00: Picture Reset 01: Factory Reset Ack [k][ ][Set ID][ ][OK/NG][Data][x] ENGLISH ▶▶ This controls the OSD Lock feature. To obtain the source code under GPL, LGPL, MPL, and other open source licenses, that is contained in this product, please visit http://opensource.lge.com. In addition to the source code, all referred license terms, warranty disclaimers and copyright notices are available for download. LG Electronics will also provide open source code to you on CD-ROM for a charge covering the cost of performing such distribution (such as the cost of media, shipping, and handling) upon email request to email@example.com. This offer is valid for three (3) years from the date on which you purchased the product. Read the Owner's Manual (CD) carefully and keep it at hand. Please note that the label attached to the product provides information for product support. Model Serial No. WARNING: This equipment is compliant with Class A of CISPR 32. In a residential environment this equipment may cause radio interference.