Color LCD monitor Important Please read this “Instructions for Use”, and “Installation Manual” (separate volume) carefully to familiarize yourself with safe and effective usage. Please retain this manual for future reference. ••For monitor adjustment and settings, refer to the Installation Manual. ••The latest “Instructions for Use” is available for download from our web site : http://www.eizoglobal.com English Instructions for Use SAFETY SYMBOLS This manual and this product use the safety symbols below. They denote critical information. Please read them carefully. WARNING CAUTION Failure to abide by the information in a WARNING may result in serious injury and can be life threatening. Failure to abide by the information in a CAUTION may result in moderate injury and/or property or product damage. Indicates a warning or caution. For example, Indicates a prohibited action. For example, Indicates a mandatory action. For example, indicates an “electrical shock” hazard. means “Do not disassemble”. means “Ground the unit”. This product has been adjusted specifically for use in the region to which it was originally shipped. If operated outside this region, the product may not perform as stated in the specifications. No part of this manual may be reproduced, stored in a retrieval system, or transmitted, in any form or by any means, electronic, mechanical, or otherwise, without the prior written permission of EIZO Corporation. EIZO Corporation is under no obligation to hold any submitted material or information confidential unless prior arrangements are made pursuant to EIZO Corporation’s receipt of said information. Although every effort has been made to ensure that this manual provides up-to-date information, please note that EIZO monitor specifications are subject to change without notice. 2 PRECAUTIONS •• This product has been adjusted specifically for use in the region to which it was originally shipped. If the product is used outside the region, it may not operate as specified in the specifications. •• To personal safety and proper maintenance, please read carefully this section and the caution statements on the monitor. Location of the Caution Statements WARNING AVERTISSEMENT WARNUNG Symbols on the unit Symbol This symbol indicates Main Power Switch: Press to turn the monitor’s main power off. Main Power Switch: Press to turn the monitor’s main power on. Power button: Press to turn the monitor’s power on or off. Alternating current Alerting electrical hazard CAUTION: Refer to “SAFETY SYMBOLS” (page 2). WEEE marking: Product must be disposed of separately; materials may be recycled. CE marking: EU conformity mark in accordance with the provisions of Council Directive 93/42/EEC and 2011/65/EU. Manufacturer Date of manufacture Caution: Federal law (USA) restricts this device to sale by or on the order of a licensed healthcare practitioner. PRECAUTIONS 3 English ●●IMPORTANT WARNING If the unit begins to emit smoke, smells like something is burning, or makes strange noises, disconnect all power connections immediately and contact your EIZO representative for advice. Attempting to use a malfunctioning unit may result in fire, electric shock, or equipment damage. Do not disassemble or modify the unit. Opening the cabinet or modifying the unit may result in fire, electric shock, or burn. Refer all servicing to qualified service personnel. Do not attempt to service this product yourself as opening or removing covers may result in fire, electric shock, or equipment damage. Keep small objects or liquids away from the unit. Small objects accidentally falling through the ventilation slots into the cabinet or spillage into the cabinet may result in fire, electric shock, or equipment damage. If an object or liquid falls/spills into the cabinet, unplug the unit immediately. Have the unit checked by a qualified service engineer before using it again. Place the unit at a sturdy and stable place. A unit placed on an inadequate surface may fall and result in injury or equipment damage. If the unit falls, disconnect the power immediately and ask your local EIZO representative for advice. Do not continue using a damaged unit. Using a damaged unit may result in fire or electric shock. Use the unit in an appropriate location. Otherwise, fire, electric shock, or equipment damage may result. •• •• •• •• •• •• •• •• •• •• Do not place outdoors. Do not place in any form of transportation (ships, aircraft, trains, automobiles, etc.). Do not place in dusty or humid environments. Do not place in locations where water may be splashed on the screen (bathrooms, kitchens, etc.) Do not place in locations where steam comes in direct contact with the screen. Do not place near heat generating devices or humidifiers. Do not place in locations where the product is subject to direct sunlight. Do not place in environments with inflammable gas. Do not place in environments with corrosive gases (such as sulfur dioxide, hydrogen sulfide, nitrogen dioxide, chlorine, ammonia, and ozone) Do not place in environments with dust, components that accelerate corrosion in the atmosphere (such as sodium chloride and sulfur), conductive metals, and so on To avoid danger of suffocation, keep the plastic packing bags away from babies and children. Use the enclosed power cord and connect to the standard power outlet in your country. Be sure to use within the rated voltage of the power cord. Otherwise, fire or electric shock may result. Power supply: 100-240Vac 50/60Hz To disconnect the power cord, grasp the plug firmly and pull. Tugging on the cord may damage and result in fire or electric shock. The equipment must be connected to a grounded main outlet. Failure to do so may result in fire or electric shock. Use the correct voltage. •• The unit is designed for use with a specific voltage only. Connection to another voltage than specified in this “Instructions for Use” may cause fire, electric shock, or equipment damage. Power supply: 100-240Vac 50/60Hz •• Do not overload your power circuit, as this may result in fire or electric shock. 4 PRECAUTIONS WARNING English Handle the power cord with care. •• Do not place the cord underneath the unit or other heavy objects. •• Do not pull on or tie the cord. If the power cord becomes damaged, stop using it. Using of a damaged cord may result in fire or electric shock. The operator should not touch the patient while touching the product. This product has not been designed to be touched by patients. Never touch the plug and power cord if it begins to thunder. Touching them may result in electric shock. When attaching an arm stand, please refer to the user’s manual of the arm stand and install the unit securely. Otherwise, the unit may become detached, resulting in injury and/or equipment damage. Before installation, make sure that desks, walls, or any other installation surface has adequate mechanical strength. If the unit falls, please ask your local EIZO representative for advice. Do not continue using a damaged unit. Using a damaged unit may result in fire or electric shock. When reattaching the tilt stand, please use the same screws and tighten them securely. Do not touch a damaged LCD panel directly with bare hands. Liquid crystal is poisonous. If any part of your skin comes in direct contact with the panel, wash thoroughly. If liquid crystal enters your eyes or mouth, immediately flush with large amounts of water and seek medical attention. PRECAUTIONS 5 CAUTION Handle with care when carrying the unit. Disconnect the power cord and cables when moving the unit. Moving the unit with the power cord or cables attached is dangerous and may result in injury. Carry or place the unit according to the correct specified methods. •• When carrying, grasp and firmly hold the unit as shown in the illustration below. •• Monitors of size 30 inches and above are heavy. When unpacking and/or carrying the monitor, ensure at least two people are utilized. Dropping the unit may result in injury or equipment damage. Do not block the ventilation slots on the cabinet. •• Do not place any objects on the ventilation slots. •• Do not install the unit in a place with poor ventilation or inadequate space. •• Do not use the unit laid down or upside down. Blocking the ventilation slots prevents proper airflow and may result in fire, electric shock, or equipment damage. Do not touch the plug with wet hands. Doing so may result in electrical shock. Use an easily accessible power outlet. This is to facilitate disconnecting the power in case of a problem. Periodically clean the area around the power plug and the ventilation slot of the monitor. Dust, water, or oil on the plug may result in fire. Unplug the unit before cleaning it. Cleaning the unit while it is plugged into a power outlet may result in electric shock. If you plan to leave the unit unused for an extended period of time, disconnect the power cord from the wall socket after turning off the power switch for the safety and the power conservation. 6 PRECAUTIONS Notice for this monitor This product is intended to be used in displaying and viewing digital images, including standard and multiframe digital mammography, for review, analysis and diagnosis by trained medical practitioners. It is specially designed for breast tomosynthesis applications. Attention •• Mammographic images with lossy compression must not be reviewed for primary image interpretations. Mammographic images may only be interpreted using an FDA cleared display that meets technical specifications reviewed and accepted by FDA. •• This product may not be covered by warranty for uses other than those described in this manual. •• The specifications noted in this manual are only applicable when the following are used: -- Power cords provided with the product -- Signal cables specified by us •• Only use optional products manufactured or specified by us with this product. Precautions for Use •• Otherwise, parts (such as the LCD panel) may deteriorate in the long-term. Periodically check that they are operating normally. •• When the screen image is changed after displaying the same image for extended periods of time, an afterimage may appear. Use the screen saver or power save function to avoid displaying the same image for extended periods of time. •• If the monitor displays continuously over a long period of time, dark smudges or burn-in may appear. To maximize the life of the monitor, we recommend the monitor be turned off periodically. •• An afterimage may appear even after a short time period has elapsed depending on the displayed image. If this occurs, changing the image or leaving the power off for a few hours may solve the problem. •• The backlight of the LCD panel has a fixed lifetime. When the screen becomes dark or begins to flicker, please contact your local EIZO representative. •• The screen may have defective pixels or a small number of light dots on the screen. This is due to the characteristics of the panel itself, and is not a malfunction of the product. •• Do not press on the panel or edge of the frame strongly, as this may result in display malfunctions, such as interference patterns, etc. If pressure is continually applied to the panel, it may deteriorate or damage your panel. (If the pressure marks remain on the panel, leave the monitor with a black or white screen. The symptom may disappear.) •• Do not scratch or press on the panel with any sharp objects, as this may result in damage to the panel. Do not attempt to brush with tissues as this may scratch the panel. •• Depending on the environment, the value measured by the built-in illuminance sensor may differ from the value shown on a stand-alone illuminometer. •• When the monitor is cold and brought into a room or the room temperature goes up quickly, dew condensation may occur on the interior and exterior surfaces of the monitor. In that case, do not turn the monitor on. Instead wait until the dew condensation disappears, otherwise it may cause some damage to the monitor. Notice for this monitor 7 English Intended use To Use the Monitor for a Long Time ●●Maintenance •• The display quality of monitors is affected by the quality level of input signals and the degradation of the product. Perform visual checks and periodic constancy tests to comply with medical standards / guidelines according to your application, and carry out calibration as necessary. Use of the RadiCS monitor quality control software enables you to perform high-level quality control that meets medical standards / guidelines. •• It takes about 15 minutes for the performance of electrical parts to stabilize. Please wait 15 minutes or more after the monitor power has been turned on or the monitor has recovered from the power saving mode, and then adjust the monitor. •• We recommend that monitors be set to the recommended level or lower to reduce changes in luminosity caused by long-term use and maintain stable brightness. •• To adjust measurement results of the integrated calibration sensor (Integrated Front Sensor) to those of an EIZO external sensor (UX1 or UX2 sensor) that is sold separately, perform correlation between the Integrated Front Sensor and the external sensor using RadiCS / RadiCS LE. Periodical correlation allows you to maintain the measurement accuracy of the Integrated Front Sensor at a level equivalent to that of the external sensor. ●●Cleaning Periodic cleaning is recommended to keep the monitor looking new and to prolong its operation lifetime. Gently wipe off any dirt on the cabinet or panel surface with a soft cloth soaked in a small amount of water or one of the chemicals listed below. Chemicals that may be used for cleaning Material name Ethanol Isopropyl alcohol Chlorhexidine Benzalkonium chloride Alkyldiaminoethylglycine Glutaral Ethanol Isopropyl alcohol Hibitane Welpas Tego 51 Sterihyde Product name Attention •• Do not use chemicals on a frequent basis. Chemicals such as alcohol and antiseptic solution may cause gloss variation, tarnishing, and fading of the cabinet or panel, and also quality deterioration of the image. •• Never use any thinner, benzene, wax, and abrasive cleaner, which may damage the cabinet or panel. •• Do not let chemicals come into direct contact with the monitor. Note •• The optional ScreenCleaner is recommended for cleaning the cabinet and panel surface. To use the monitor comfortably •• Staring at the monitor for a long time tires your eyes. Take a 10-minute rest every hour. •• Look at the screen from a proper distance and from a proper angle. 8 Notice for this monitor CONTENTS English PRECAUTIONS....................................................... 3 ●●IMPORTANT.................................................... 3 Notice for this monitor.......................................... 7 Intended use.............................................................. 7 Precautions for Use.................................................. 7 To Use the Monitor for a Long Time........................ 8 ●●Maintenance..................................................... 8 ●●Cleaning........................................................... 8 To use the monitor comfortably.............................. 8 CONTENTS.............................................................. 9 Chapter 1 Introduction...................................... 10 1-1. Features.........................................................10 1-2. Package Contents.........................................11 1-3. Controls and Functions...............................13 ●●EIZO LCD Utility Disk......................................12 Chapter 2 2-1. Installation / Connection................. 14 Before Installing the Product......................14 ●● Installation Requirements...............................14 2-2. Connecting Cables.......................................15 2-3. Turning On the Power...................................17 2-4. Adjusting the Screen Height and Angle.....17 Chapter 3 No-Picture Problem......................... 18 Chapter 4 Specifications.................................. 19 4-1. Specifications List........................................19 4-2. Compatible Resolutions............................. 20 4-3. Optional Accessories.................................. 20 Appendix............................................................... 21 FCC Declaration of Conformity..............................21 Medical Standard.................................................... 22 EMC Information..................................................... 23 CONTENTS 9 Chapter 1 Introduction Thank you very much for choosing an EIZO color LCD monitor. 1-1. Features ●●Hybrid monochrome and color display This color monitor can display images at a high level of brightness of up to 1100 cd / m2, which is close to the brightness of a monochrome monitor. Therefore, this product can display digital color images, including those used for medical purposes such as breast MRI or CT images, ultrasonic images, and pathological images as well as monochrome images for breast tomosynthesis or mammographies that require the monitor to have a high level of display performance. Also, when the Hybrid Gamma PXL function is enabled, this product automatically differentiates between monochrome and color parts of the same image at a pixel level, and displays them respectively in optimal gradations. ●●Quality control •• This monitor has a built-in calibration sensor (Integrated Front Sensor). This sensor enables the monitor to perform calibration (SelfCalibration) and Grayscale Check independently. •• Using RadiCS LE that is attached to the monitor, you can manage history related to the monitor, and the SelfCalibration target and execution schedule. •• The RadiCS monitor quality control software enables you to perform quality control that meets medical standards / guidelines. 10 Chapter 1 Introduction ●●Simple wiring In addition to a DisplayPort input terminal, an output terminal is also provided. English •• From the output terminal ( ), a signal can be output to a different monitor. ●●Monitor operation from the mouse and keyboard Using the RadiCS / RadiCS LE monitor quality control software, you can perform the following monitor operations using the mouse and keyboard: •• Switching CAL Switch modes •• Switching input signals •• Function that assigns any CAL Switch mode to a part of the screen and displays an image (Pointand-Focus) •• Entering power saving mode (Backlight Saver) 1-2. Package Contents Check that all of the following items are contained in the package. If any of these are missing or damaged, contact your dealer or local EIZO representative listed on the attached sheet. Note •• It is recommended that the box and packing materials be stored so that they can be used to move or transport this product. •• Monitor •• Power cord •• Digital signal cable: PP300 x 1 DisplayPort - DisplayPort •• USB cable: UU300 x 1 •• EIZO LCD Utility Disk (CD-ROM) •• Instructions for Use •• Digital signal cable: DD300DL x 1 DVI - DVI (dual link) Chapter 1 Introduction 11 ●●EIZO LCD Utility Disk The CD-ROM contains the following items. Refer to "Readme.txt" on the disk for software startup procedures or file reference procedures. •• Readme.txt file •• RadiCS LE monitor quality control software (for Windows) •• User’s Manual Monitor Installation Manual RadiCS LE User's Manual •• Outline dimensions RadiCS LE RadiCS LE enables you to perform the following quality control and monitor operations. For more information about the software or setup procedures, refer to RadiCS LE User's Manual. Quality control •• Executing calibration •• Displaying test results in a list and creating a test report •• Setting the SelfCalibration target and execution schedule Monitor operations •• Switching CAL Switch modes •• Switching input signals •• Function that assigns any CAL Switch mode to a part of the screen and displays an image (Pointand-Focus) •• Entering power saving mode (Backlight Saver) Attention •• The specifications of RadiCS LE are subject to change without notice. The latest version of RadiCS LE is available for download from our web site: http://www.eizoglobal.com To use RadiCS LE For information on how to install and use RadiCS LE, refer to RadiCS LE User's Manual (on the CD-ROM). When using RadiCS LE, connect the monitor to your PC using the supplied USB cable. For more information about how to connect the monitor, see “2-2. Connecting Cables” (page 15). Note •• In an environment where it is difficult to connect the USB cable, enabling DDC communication allows you to RadiCS LE without having to use the USB cable. For information on how to configure DDC communication, refer to the Installation Manual (on the CD-ROM). In view of communication speed and operation stability, USB connection is recommended. 12 Chapter 1 Introduction 1-3. Controls and Functions English 1 2 Approx. 30° 12 6 13 14 Setting menu*1 15 3 4 5 6 7 8 9 10 11 1. Integrated Front Sensor This sensor is used to perform calibration and Grayscale Check. (Movable) 2. Ambient Light Sensor This sensor measures environmental illumination. Environmental illuminance measurement is performed using the RadiCS / RadiCS LE quality control software. 3. Presence Sensor This sensor detects movements made by a person in front of the monitor. 4. Operation switches 5. 6. 7. 8. 9. 10. 11. 12. 13. 14. 15. *1 Displays the operation guide. Set menus according to the operation guide. Turns the power on or off. The switch indicator is lit when you turn the power on. The indicator color differs depending on the monitor's operation status. Green: Monitor in operation, Orange: Power saving mode, Off: Main power / power off Connect it to a USB device. To set up a daisy-chain connection, connect the cable to USB downstream port the USB upstream port of another monitor. Connect this port to the PC when you use software that needs a USB connection or USB upstream port connect a USB device (peripheral device that supports USB) to the USB downstream port. DVI-D connector Connect it to the PC. DisplayPort input Connect it to the PC. To set up a daisy-chain connection from another monitor, connector connect the cable to the DisplayPort output connector of that monitor. DisplayPort output To set up a daisy-chain connection, connect the cable to the DisplayPort input connector connector of another monitor. Power connector Connects the power cord. Stand The height and angle can be adjusted. Main power switch Turns the main power on or off. : On, : Off, Security lock slot Complies with Kensington’s MicroSaver security system. Cable holder Holds the monitor cables. For information on how to use, refer to the Installation Manual (on the CD-ROM). switch Chapter 1 Introduction 13 Chapter 2 Installation / Connection 2-1. Before Installing the Product Carefully read “PRECAUTIONS” (page 3) and always follow the instructions. If you place this product on a lacquer-coated desk, the color may adhere to the bottom of the stand due to the composition of the rubber. Check the desk surface before use. ●●Installation Requirements When installing the monitor in a rack, ensure that there is adequate space around the sides, back and top of the monitor. Attention •• Position the monitor so that there is no light to interfere with the screen. 14 Chapter 2 Installation / Connection 2-2. Connecting Cables •• Check that the monitor and the PC are powered off. •• When replacing the current monitor with this monitor, refer to “4-2. Compatible Resolutions” (page 20) to change the PC settings for resolution and vertical scan frequency to those that are available for this monitor, before connecting the PC. 1. Turn the monitor screen 90° clockwise. The monitor is installed in landscape orientation before shipment. Attention •• Before turning the monitor screen, raise the monitor to its highest position. 2. Connect signal cables. Check the shapes of the connectors, and connect the cables. After connecting the DVI cable, tighten the fasteners to secure the connector. Attention •• The monitor has two types of DisplayPort connectors: input and output. When connecting the monitor to a PC, connect the cable to the input connector. •• When connecting to multiple PCs, switch the input signal. For details, refer to the Installation Manual (on the CD-ROM). Signal cable: DD300DL Signal cable: PP300 or Chapter 2 Installation / Connection 15 English Attention Setting up a daisy-chain connection Output the signal input to the input connector to another monitor. Otherwise, input the signal from the output connector of another monitor. Attention •• Visit the EIZO website for information about monitors and graphic boards that can be used for the daisychain connection: http://www.eizoglobal.com •• To set up a daisy-chain connection, you need to select “Signal Format” - “DisplayPort” on the Administrator Settings menu, and set “Version” to “1.2”. For details, refer to the Installation Manual (on the CD-ROM). •• Remove the cap before connecting the signal cable. Signal cable: PP300 Signal cable: PP300 3. Plug the power cord into a power outlet and the power connector on the monitor. Insert the power cord fully into the monitor. 4. When you use RadiCS / RadiCS LE or connect a USB device (peripheral device that supports USB) to the monitor, connect the USB cable to the monitor's USB upstream port and the PC. Connecting to the Monitor 16 Chapter 2 Installation / Connection Connecting to a Computer 2-3. Turning On the Power English 1. Touch to turn on the power to the monitor. The power switch indicator of the monitor lights up green. If the indicator does not light up, see “Chapter 3 No-Picture Problem” (page 18). Note •• When the power of the monitor is not on, touching any button except causes to flash. 2. Turn on the PC. The screen image appears. If an image does not appear, refer to “Chapter 3 No-Picture Problem” (page 18) for additional advice. Attention •• For the maximum power saving, it is recommended that the Power button be turned off. When not using the monitor, you can turn off the main power supply or disconnect the power plug so that the power is cut completely. Note •• In order to maximize the monitor's lifespan by impeding brightness degradation and to reduce power consumption, carry out the following: -- Use the power saving function of the PC or monitor. -- Turn off the monitor after using it. 2-4. Adjusting the Screen Height and Angle Hold left and right edge of the monitor with both hands, and adjust the screen height, tilt and swivel of the screen to the best condition for working. Attention •• After the adjustment is finished, make sure that the cables are correctly connected. Chapter 2 Installation / Connection 17 Chapter 3 No-Picture Problem Problem 1. No picture •• Power switch indicator does not light up •• Power switch indicator lights up: Green •• Power switch indicator lights up: Orange •• Power switch indicator blinks: Orange, Green 2. The message below appears. 18 Possible cause and remedy •• •• •• •• •• Check whether the power cord is connected properly. Turn the main power switch on. Touch . Turn off the main power, and then turn it on again. Increase "Brightness", "Contrast", or "Gain" in the Setting menu. For details, refer to the Installation Manual (on the CD-ROM). •• Turn off the main power, and then turn it on again. •• Switch the input signal. For details, refer to the Installation Manual (on the CD-ROM). •• Move the mouse or press any key on the keyboard. •• Check whether the PC is turned on. •• If the Presence Sensor is set to "On", the monitor may be in power saving mode. Try moving closer to the monitor. •• Check whether the signal cable is connected properly. When inputting the DisplayPort signal, connect to . is used for output when a daisy-chain connection is set up. •• Turn off the main power, and then turn it on again. •• Connect via the signal cable specified by EIZO. Turn off the main power, and then turn it on again. This message appears when the signal is not input correctly even though the monitor is functioning properly. •• This message appears when no signal is input. Example: •• The message shown left may appear, because some PCs do not output the signal immediately after power-on. •• Check whether the PC is turned on. •• Check whether the signal cable is connected properly. •• Switch the input signal. For details, refer to the Installation Manual (on the CD-ROM). •• When inputting the DisplayPort signal, try switching the DisplayPort version. For details, refer to the Installation Manual (on the CD-ROM). •• Turn off the main power, and then turn it on again. •• The message indicates that the input signal is out of the specified frequency range. Example: •• Check whether the PC is configured to meet the resolution and vertical scan frequency requirements of the monitor (see “4-2. Compatible Resolutions” (page 20)). •• Reboot the PC. •• Select an appropriate setting using the graphics board’s utility. For more information, refer to the User’s Manual of the graphics board. Chapter 3 No-Picture Problem Chapter 4 Specifications English 4-1. Specifications List Type LCD Panel Type Backlight Size RX560: Anti-Glare RX560-AR: Anti-Reflection Color (IPS) LED 54.1 cm (21.3 inch) Resolution (H x V) 2048 × 2560 Display Size (H x V) 337.9 mm × 422.4 mm Pixel Pitch 0.165 mm Display Colors 10-bit (DisplayPort): 1073.74 million colors (Max.) 8-bit (DVI): 16.77 million colors Video Signals Viewing Angles (H / V, typical) Recommended Brightness Response Time (typical) 178˚ / 178˚ Input Terminals DisplayPort × 1, DVI-D (dual link) × 1 Output Terminal DisplayPort × 1 Horizontal scanning frequency Vertical scan frequency 31 kHz - 135 kHz Frame Synchronization mode Dot clock 500 cd/m2 12 ms (black -> white -> black) DisplayPort: 59 Hz - 61 Hz (720×400 : 69 Hz - 71 Hz, 2560×2048 : 23 Hz - 51 Hz) 59 Hz - 61 Hz DVI: (720×400 : 69 Hz - 71 Hz, 2560×2048 : 24 Hz - 51 Hz) 23.5 Hz - 25.5 Hz, 47.0 Hz - 51.0 Hz USB Port DisplayPort: 25 MHz - 290 MHz DVI : 25 MHz - 165 MHz 165 MHz - 290 MHz (Dual link) Upstream port × 1, downstream port × 2 Standard USB Specification Revision 2.0 Power Input Maximum Power Consumption Power Save Mode Standby Mode Dimensions ( W × H × D ) 100 - 240 VAC ±10 %, 50 / 60 Hz 0.90 A - 0.40 A 87 W or less Physical Specifications Operating Environmental Requirements 1.0 W or less*1 1.0 W or less*2 354.5 mm × 476.0 mm - 566.0 mm × 200.0 mm (Tilt: 0˚) 354.5 mm × 504.7 mm - 594.7 mm × 264.1 mm (Tilt: 30˚) Dimensions ( W × H × D ) 354.5 mm × 452.0 mm × 78.0 mm (Without Stand) Net Weight Approx. 8.1 kg Net Weight Approx. 5.3 kg (Without Stand) Height Adjustment Range 90 mm (Tilt: 0˚) Tilt Up 30˚, down 5˚ Swivel 70˚ Rotation 90° (rotate counterclockwise from a portrait orientation) Temperature 0 ˚C - 35 ˚C (32 ˚F - 95 ˚F) Humidity 20 % - 80 % R.H. (no condensation) Air Pressure 540 hPa - 1060 hPa Chapter 4 Specifications 19 Temperature -20 ˚C - 60 ˚C (-4 ˚F - 140 ˚F) Transportation / Storage Humidity 10 % - 90 % R.H. (no condensation) Environmental Air Pressure 200 hPa - 1060 hPa Requirements *1 When DVI input is used, the USB upstream port is not connected, “Auto Input Detection”: “Off”, “Power Save”: “High”, “DP Power Save”: “On”, “DisplayPort” - “Version”: “1.1”, “DDC”: “Off”, and no external load is connected *2 When the USB upstream port is not connected, “DP Power Save”: “On”, “DisplayPort” - “Version”: “1.1”, “DDC”: “Off”, no external load is connected 4-2. Compatible Resolutions The monitor supports the following resolutions. √: Supported Resolution (H x V) Vertical scan frequency 720 × 400 640 × 480 800 × 600 1024 × 768 1280 × 1024 1600 × 1200 2560 × 2048 2048 × 2560 2560 × 2048 2048 × 2560 2560 × 2048 2048 × 2560 70 Hz 60 Hz 60 Hz 60 Hz 60 Hz 60 Hz 50 Hz 50 Hz 48 Hz 48 Hz 25 Hz 25 Hz DisplayPort Portrait Landscape √ √ √ √ √ √ √*1 √*2 - √ √ √ √ √ √ √*1 √*2 - DVI Portrait Landscape √ √ √ √ √ √ √*3 √ √ √ √ √ √ √ √*3 √ - *1 Only when the “DisplayPort version” is “1.1”. *2 Only when the “DisplayPort version” is “1.2”. *3 A dual link signal only. 4-3. Optional Accessories The following accessories are available separately. For the latest information about the optional accessories and information about the latest compatible graphics board, refer to our web site. http://www.eizoglobal.com Calibration Kit Network QC Management Software Cleaning Kit Comfort Light for Reading Rooms Panel protector 20 Chapter 4 Specifications EIZO "RadiCS UX2" Ver. 4.6.0 or later EIZO "RadiCS Version Up Kit" Ver. 4.6.0 or later EIZO "RadiNET Pro" Ver. 4.6.0 or later EIZO "RadiNET Pro Lite" Ver. 4.6.0 or later EIZO “ScreenCleaner” EIZO “RadiLight” RP-918 Appendix English FCC Declaration of Conformity For U.S.A., Canada Only FCC Declaration of Conformity We, the Responsible Party EIZO Inc. 5710 Warland Drive, Cypress, CA 90630 Phone: (562) 431-5011 declare that the product Trade name: EIZO Model: RadiForce RX560 is in conformity with Part 15 of the FCC Rules. Operation of this product is subject to the following two conditions: (1) this device may not cause harmful interference, and (2) this device must accept any interference received, including interference that may cause undesired operation. This equipment has been tested and found to comply with the limits for a Class B digital device, pursuant to Part 15 of the FCC Rules. These limits are designed to provide reasonable protection against harmful interference in a residential installation. This equipment generates, uses, and can radiate radio frequency energy and, if not installed and used in accordance with the instructions, may cause harmful interference to radio communications. However, there is no guarantee that interference will not occur in a particular installation. If this equipment does cause harmful interference to radio or television reception, which can be determined by turning the equipment off and on, the user is encouraged to try to correct the interference by one or more of the following measures. * Reorient or relocate the receiving antenna. * Increase the separation between the equipment and receiver. * Connect the equipment into an outlet on a circuit different from that to which the receiver is connected. * Consult the dealer or an experienced radio/TV technician for help. Changes or modifications not expressly approved by the party responsible for compliance could void the user’s authority to operate the equipment. Note Use the attached specified cable below or EIZO signal cable with this monitor so as to keep interference within the limits of a Class B digital device. - AC Cord - Shielded Signal Cable (enclosed) Canadian Notice This Class B information technology equipment complies with Canadian ICES-003. Cet équipement informatique de classe B est conforme à la norme NMB-003 du Canada. Appendix 21 Medical Standard •• It shall be assured that the final system is in compliance to IEC60601-1-1 requirement. •• Power supplied equipment can emit electromagnetic waves, that could influence, limit or result in malfunction of the monitor. Install the equipment in a controlled environment, where such effects are avoided. Classification of Equipment ------ 22 Type of protection against electric shock : Class I EMC class: EN60601-1-2:2015 Group 1 Class B Classification of medical device (MDD 93/42/EEC): Class I Mode of operation : Continuous IP Class : IPX0 Appendix EMC Information Environments of Intended Use The RadiForce series is intended to be used in Professional healthcare facility environments such as clinics and hospitals. The following environments are not suitable for the RadiForce series to be used: •• Home healthcare environments •• In the vicinity of high-frequency surgical equipments such as electrosurgical knives •• In the vicinity of short-wave therapy equipments •• RF shielded room of the medical equipment systems for MRI •• In shielded location Special environments •• Installed in vehicles including ambulances. •• Other special environments WARNING The RadiForce series requires special precautions regarding EMC and need to be installed. You need to carefully read EMC Information and the "PRECAUTIONS" section in this document, and observe the following instructions when installing and operating the product. Be sure to use the cables attached to the product, or cables specified by EIZO. Use of cables other than those specified or provided by EIZO of this equipment could result in increased electromagnetic emissions or decreased electromagnetic immunity of this equipment and improper operation. Cable length: 3 m max. The RadiForce series should not be used adjacent to or stacked with other equipment. If adjacent or stacked use is necessary, the equipment or system should be observed to verify normal operation in the configuration in which it will be used. When using a portable RF communication equipment, keep it 30 cm (12 inches) or more away from any part, including cables, of the RadiForce series. Otherwise, degradation of the performance of this equipment could result. Anyone who connects additional equipment to the signal input part or signal output parts, configuring a medical system, is responsible that the system complies with the requirements of IEC/EN60601-1-2. Technical Descriptions Electromagnetic emissions The RadiForce series is intended for use in the electromagnetic environment specified below. The customer or the user of the RadiForce series should assure that it is used in such an environment. Emission test Compliance RF emissions CISPR11 / EN55011 Group 1 RF emissions CISPR11 / EN55011 Harmonic emissions IEC / EN61000-3-2 Voltage fluctuations / flicker emissions IEC / EN61000-3-3 Class B Class D Electromagnetic environment - Guidance The RadiForce series uses RF energy only for its internal function. Therefore, its RF emission are very low and are not likely to cause any interference in nearby electronic equipment. The RadiForce series is suitable for use in all establishments, including domestic establishments and those directly connected to the public low-voltage power supply network that supplies buildings used for domestic purposes. Complies Appendix 23 English The RadiForce series has a performance that appropriately displays images. Electromagnetic immunity The RadiForce series has been tested at the following compliance levels according to the testing requirements for professional healthcare facility environments defined in IEC / EN60601-1-2. Customers and users of the RadiForce series must ensure that the RadiForce series is used in the following environments: Immunity test Electrostatic discharge (ESD) IEC / EN61000-4-2 Electrical fast transients / bursts IEC / EN61000-4-4 Surges IEC / EN61000-4-5 Voltage dips, short interruptions and voltage variations on power supply input lines IEC / EN61000-4-11 Power frequency magnetic fields IEC / EN61000-4-8 24 Appendix Test level for professional healthcare facility environments Compliance level ±8 kV contact discharge ±15 kV air discharge ±8 kV contact discharge ±15 kV air discharge ±2 kV power lines ±1 kV input / output lines ±2 kV power lines ±1 kV input / output lines ±1 kV line to line ±2 kV line to ground 0 % UT (100 % dip in UT) 0.5 cycles and 1 cycle 70 % UT (30 % dip in UT) 25 cycles 0 % UT (100 % dip in UT) 5 sec ±1 kV line to line ±2 kV line to ground 0 % UT (100 % dip in UT) 0.5 cycles and 1 cycle 70 % UT (30 % dip in UT) 25 cycles 0 % UT (100 % dip in UT) 5 sec 30 A/m (50 / 60 Hz) 30 A/m Electromagnetic environment - Guidance Floors should be wood, concrete or ceramic tile. If floors are covered with synthetic material, the relative humidity should be at least 30%. Mains power quality should be that of a typical commercial or hospital environment. Mains power quality should be that of a typical commercial or hospital environment. Mains power quality should be that of a typical commercial or hospital environment. If the user of the RadiForce series requires continued operation during power mains interruptions, it is recommended that the RadiForce series be powered from an uninterruptible power supply or a battery. Power frequency magnetic fields should be at levels characteristic of a typical location in a typical commercial or hospital environment. The product should be kept at least 15 cm away from the source of power frequency magnetic fields during use. Electromagnetic immunity Immunity test Test level for professional healthcare facility environments Compliance level Electromagnetic environment - Guidance Portable and mobile RF communications equipment should be used no closer to any part of the RadiForce series, including cables, than the recommended separation distance calculated from the equation applicable to the frequency of the transmitter. Conducted disturbances induced by RF fields IEC / EN61000-4-6 Radiated RF fields IEC / EN61000-4-3 Recommended separation distance 3 Vrms 3 Vrms 150 kHz - 80 MHz 6 Vrms ISM bands between 150 kHz and 80 MHz 6 Vrms 3 V/m 80 MHz - 2.7 GHz 3 V/m d = 1.2√P d = 1.2√P, 80 MHz - 800 MHz d = 2.3√P, 800 MHz - 2.7 GHz Where “P” is the maximum output power rating of the transmitter in watts (W) according to the transmitter manufacturer and “d” is the recommended separation distance in meters (m). Field strengths from fixed RF transmitters, as determined by an electromagnetic site surveya), should be less than the compliance level in each frequency rangeb). Interference may occur in the vicinity of equipment marked with the following symbol. Note 1 Note 2 Note 3 UT is the a.c. mains voltage prior to application of the test level. At 80 MHz and 800 MHz, the higher frequency range applies. Guidelines regarding conducted disturbances induced by RF fields or radiated RF fields may not apply in all situations. Electromagnetic propagation is affected by absorption and reflection from structures, objects and people. Note 4 The ISM bands between 150 kHz and 80 MHz are 6.765 MHz to 6.795 MHz, 13.553 MHz to 13.567 MHz, 26.957 MHz to 27.283 MHz, and 40.66 MHz to 40.70 MHz. a) Field strengths from fixed transmitters, such as base stations for radio (cellular/cordless) telephones and land mobile radios, amateur radio, AM and FM radio broadcast and TV broadcast cannot be predicted theoretically with accuracy. To assess the electromagnetic environment due to fixed RF transmitters, an electromagnetic site survey should be considered. If the measured field strength in the location in which the RadiForce series is used exceeds the applicable RF compliance level above, the RadiForce series should be observed to verify normal operation. If abnormal performance is observed, additional measures may be necessary, such as reorienting or relocating the RadiForce series. b) Over the frequency range 150 kHz to 80 MHz, field strengths should be less than 3 V/m. Appendix 25 English The RadiForce series has been tested at the following compliance levels according to the testing requirements for professional healthcare facility environments defined in IEC / EN60601-1-2. Customers and users of the RadiForce series must ensure that the RadiForce series is used in the following environments: Recommended separation distances between portable or mobile RF communication equipment and the RadiForce Series The RadiForce series is intended for use in an electromagnetic environment in which radiated RF disturbances are controlled. The customer or the user of the RadiForce series can help prevent electromagnetic interference by maintaining a minimum distance between portable and mobile RF communications equipment (transmitters) and the RadiForce series. Immunity to proximity fields from following RF wireless communication equipments has been confirmed: Minimum Compliance Test freMaximum separation IEC / EN60601 Bandwidth a) a) b) level quency power Service Modulation distance test level (V/m) (MHz) (V/m) (MHz) (W) (m) 385 380 - 390 TETRA 400 450 430 - 470 GMRS 460, FRS 460 710 745 780 810 870 930 704 - 787 LTE Band 13, 17 800 - 960 GSM 800 / 900, TETRA 800, iDEN 820 CDMA 850, LTE Band 5 1720 1845 1970 1700 - 1990 GSM 1800; CDMA 1900; GSM 1900; DECT; LTE Band 1, 3, 4, 25; UMTS 2400 - 2570 Bluetooth, WLAN, 802.11 b/g/n, RFID 2450, LTE Band 7 5100 - 5800 WLAN 802.11 a/n 2450 Pulse modulation b) 1.8 18 Hz FM 2 ±5 kHz deviation 1 kHz sine Pulse modulation b) 0.2 217 Hz 0.3 27 27 0.3 28 28 0.3 9 9 Pulse modulation b) 2 18 Hz 0.3 28 28 Pulse modulation b) 2 217 Hz 0.3 28 28 Pulse modulation b) 2 217 Hz 0.3 28 28 0.3 9 9 5240 Pulse modulation b) 0.2 217 Hz 5500 5785 a) For some services, only the uplink frequencies are included. b) carrier are modulated using a 50 % duty cycle square wave signal. The RadiForce series is intended for use in an electromagnetic environment in which radiated RF disturbances are controlled. For other portable and mobile RF communication equipments (transmitters), minimum distance between portable and mobile RF communications equipment (transmitters) and the RadiForce series as recommended below, according to the maximum output power of the communications equipment. Rated maximum output power of transmitter (W) Separation distance according to frequency of transmitter (m) 150 kHz to 80 MHz 80 MHz to 800 MHz 800 MHz to 2.7 GHz d = 1.2√P d = 1.2√P d = 2.3√P 0.01 0.12 0.12 0.23 0.1 0.38 0.38 0.73 1 1.2 1.2 2.3 10 3.8 3.8 7.3 100 12 12 23 For transmitters rated at a maximum output power not listed above, the recommended separation distance “d” in meters (m) can be estimated using the equation applicable to the frequency of the transmitter, where “P” is the maximum output power rating of the transmitter in watts (W) according to the transmitter manufacturer. Note 1 Note 2 At 80 MHz and 800 MHz, the separation distance for a higher frequency range applies. These guidelines may not apply in all situations. Electromagnetic propagation is affected by absorption and reflection from structures, objects and people. Cable Length Signal cable: PP300 3m Signal cable: DD300DL 3m USB cable: UU300 3m Power cord (with earth) 26 Appendix 3m 153 Shimokashiwano, Hakusan, Ishikawa 924-8566 Japan 中国苏州市苏州工业园区展业路 8 号中新科技工业坊 5B Carl-Benz-Straße 3, 76761 Rülzheim, Germany http://www.eizoglobal.com Copyright © 2017-2019 EIZO Corporation. All rights reserved. 00N0N013C1 IFU-RX560-6 3rd Edition-May, 2019 Printed in Japan.
* Your assessment is very important for improving the work of artificial intelligence, which forms the content of this project