WARRANTY/DISCLAIMER OMEGA ENGINEERING, INC. warrants this unit to be free of defects in materials and workmanship for a period of 61 months from date of purchase. OMEGA’s WARRANTY adds an additional one (1) month grace period to the normal five (5) year product warranty to cover handling and shipping time. This ensures that OMEGA’s customers receive maximum coverage on each product. If the unit malfunctions, it must be returned to the factory for evaluation. OMEGA’s Customer Service Department will issue an Authorized Return (AR) number immediately upon phone or written request. Upon examination by OMEGA, if the unit is found to be defective, it will be repaired or replaced at no charge. OMEGA’s WARRANTY does not apply to defects resulting from any action of the purchaser, including but not limited to mishandling, improper interfacing, operation outside of design limits, improper repair, or unauthorized modification. This WARRANTY is VOID if the unit shows evidence of having been tampered with or shows evidence of having been damaged as a result of excessive corrosion; or current, heat, moisture or vibration; improper specification; misapplication; misuse or other operating conditions outside of OMEGA’s control. Components in which wear is not warranted, include but are not limited to contact points, fuses, and triacs. OMEGA is pleased to offer suggestions on the use of its various products. However, OMEGA neither assumes responsibility for any omissions or errors nor assumes liability for any damages that result from the use if its products in accordance with information provided by OMEGA, either verbal or written. OMEGA warrants only that the parts manufactured by the company will be as specified and free of defects. OMEGA MAKES NO OTHER WARRANTIES OR REPRESENTATIONS OF ANY KIND WHATSOEVER, EXPRESSED OR IMPLIED, EXCEPT THAT OF TITLE, AND ALL IMPLIED WARRANTIES INCLUDING ANY WARRANTY OF MERCHANTABILITY AND FITNESS FOR A PARTICULAR PURPOSE ARE HEREBY DISCLAIMED. LIMITATION OF LIABILITY: The remedies of purchaser set forth herein are exclusive, and the total liability of OMEGA with respect to this order, whether based on contract, warranty, negligence, indemnification, strict liability or otherwise, shall not exceed the purchase price of the component upon which liability is based. In no event shall OMEGA be liable for consequential, incidental or special damages. CONDITIONS: Equipment sold by OMEGA is not intended to be used, nor shall it be used: (1) as a “Basic Component” under 10 CFR 21 (NRC), used in or with any nuclear installation or activity; or (2) in medical applications or used on humans. Should any Product(s) be used in or with any nuclear installation or activity, medical application, used on humans, or misused in any way, OMEGA assumes no responsibility as set forth in our basic WARRANTY/DISCLAIMER language, and, additionally, purchaser will indemnify OMEGA and hold OMEGA harmless from any liability or damage whatsoever arising out of the use of the Product(s) in such a manner. ARTWORK/PRODUCT ART/ DWGS/START HERE ARROW ARTWORK/PRODUCT ART/ DWGS/START HERE ARROW 5 6 For complete product manual: www.omega.com/manuals/manualpdf/M1676.pdf To Change To Change Scale Factor ScaleValue Factor Value To Change To the Change Decimal the Decimal Point Position: Point Position: 1. 1. Press1.MENU Press untilMENU the until the display shows display SCALE shows . SCALE . sCale RECALL 2. /SET Press ADV to show to show Press4. ADV current current scale factorscale factor value. value. RECALL 5. /SET /ADV Press SET to select Press3. SET to select the setting the setting from IN/SC from IN/SC I n or IN .SC . or IN .SC . RECALL 4. /ADV MENU RECALL RESET /SET Press2. SET Press SET to show to show the current mathmatical In/SC the current mathmatical operation. operation. RECALL 3. /SET /ADV MENU MENU RECALL RESET /SET /ADV RESET 2. In/SC MENU RESET 3. . C In S S . C MENU RECALL RESET /SET 60 . 00 /ADV MENU RESET /ADV MENU RECALL RESET /SET Press1.MENU Press untilMENU the until the display shows display dECPt shows . dECPt .deCpt RECALL /SET /ADV MENU RECALL RESET /SET /ADV MENU RECALL RESET /SET /ADV MENU TM RESET SET to show Press2. SET Press to show . F F F FF F . FFFF current the currentthe location of location ofF F decimal point. the decimalthe point. RECALL /SET /ADV MENU RECALL RESET /SET /ADV MENU RESET Press Press3. SET to SET to . F F FF f F . FFF move the decimal F f F move the decimal point to thepoint right.to the right. Continue toContinue press to press SET until the meter SET until the meter displays AutO for AutO for displays auto ranging. auto ranging. RECALL /ADV MENU RESET /SET /ADV MENU RECALL RESET /SET Press4.MENU to store Press MENU to store the decimalthe point into point into store decimal volatile memory. volatile memory. RECALL 6. 0 00 /SET deCpt /ADV MENU RESET 60 . 00 4. /SET to select Press ADV to select Press5. ADV the digit youthe want toyou want to digit change. change. RECALL /ADV sCale /SET /ADV DPF700 store MENU RECALL RESET /SET /ADV MENU RESET 6. 0 00 /ADV MENU Low-Cost Ratemeter/ Totalizer/Controller RESET RETURN REQUESTS/INQUIRIES Direct all warranty and repair requests/inquiries to the OMEGA Customer Service Department. BEFORE RETURNING ANY PRODUCT(S) TO OMEGA, PURCHASER MUST OBTAIN AN AUTHORIZED RETURN (AR) NUMBER FROM OMEGA’S CUSTOMER SERVICE DEPARTMENT (IN ORDER TO AVOID PROCESSING DELAYS). The assigned AR number should then be marked on the outside of the return package and on any correspondence. FOR WARRANTY RETURNS, please have the following information available BEFORE contacting OMEGA: 1. Purchase Order number under which the product was PURCHASED, 2. Model and serial number of the product under warranty, and 3. Repair instructions and/or specific problems relative to the product. FOR NON-WARRANTY REPAIRS, consult OMEGA for current repair charges. Have the following information available BEFORE contacting OMEGA: 1. Purchase Order number to cover the COST of the repair or calibration, 2. Model and serial number of the product, and 3.Repair instructions and/or specific problems relative to the product. OMEGA’s policy is to make running changes, not model changes, whenever an improvement is possible. This affords our customers the latest in technology and engineering. OMEGA is a trademark of OMEGA ENGINEERING, INC. © Copyright 2018 OMEGA ENGINEERING, INC. All rights reserved. This document may not be copied, photocopied, reproduced, translated, or reduced to any electronic medium or machine-readable form, in whole or in part, without the prior written consent of OMEGA ENGINEERING, INC. MQS1766/0218 6. Press6. SET Press SET to to increase theincrease value ofthe thevalue of the 6. 2 00 flashing digit. flashing digit. RECALL 7. /SET /ADV MENU RECALL RESET /SET 6. 2 00It is the policy ofIt OMEGA is the policy to comply of OMEGA with all to worldwide comply with safety all worldwide and EMC/EMI safety and EMC/EMI regulations thatregulations apply. OMEGA that apply. is constantly OMEGApursuing is constantly certification pursuing of certification its of its products to the European products toNew the Approach EuropeanDirectives. New Approach Directives. /ADV MENU This device is marked This device with the is marked international with the caution international symbol. caution It is symbol. It is important to read important the Setup to Guide read before the Setup installing Guideorbefore commissioning installing or this commissioning this device as it contains device important as it contains information important relating information to safety and relating EMC. to safety and EMC. Press MENU to store Press7.MENU to store thevolatile value into volatile the value into memory. memory. RECALL RESET WARNING: These WARNING: products are These not designed products for areuse not in, designed and should for use not in, beand used should for, not be used for, patient connected applications. patient connected applications. /SET /ADV MENU RECALL RESET /SET /ADV MENU The information contained The information in this contained document in is this believed document to be is correct believed but to OMEGA be correct but OMEGA Engineering, Inc. Engineering, accepts no liability Inc. accepts for any no errors liability it contains, for any errors and reserves it contains, the right and reserves the right toRESET alter specifications to alter without specifications notice. without notice. omega.com email@example.com Servicing North America: U.S.A. Omega Engineering, Inc. Headquarters: Toll-Free: 1-800-826-6342 (USA & Canada only) The information contained in this document is believed to be correct, but OMEGA accepts no liability for any errors it contains, and reserves the right to alter specifications without notice. Customer Service: 1-800-622-2378 (USA & Canada only) Engineering Service: 1-800-872-9436 (USA & Canada only) Tel: (203) 359-1660 Fax: (203) 359-7700 e-mail: firstname.lastname@example.org For Other Locations Visit omega.com/worldwide ARTWORK/PRODUCT ART/ DWGS/START HERE ARROW ARTWORK/PRODUCT ART/ DWGS/START HERE ARROW ARTWORK/PRODUCT ART/ DWGS/START HERE ARROW ARTWORK/PRODUCT ART/ DWGS/START HERE ARROW START HERE 2 3 4 Using this Quick Reference Use this Quick Reference with your controller for information on wiring and to make changes to the operating mode, scale factor, and decimal point position. For detailed instructions, refer to the appropriate section in the Reference Manual. Safety Consideration Wiring Examples TB2 EXC 12.4 V OUTPUT RTN This device is marked with the international “Caution, risk of danger” symbol. SIG IN Menu Prompt Menu Item Prompt is starting point: 3 SIG RTN 4 Func RST IN Function 5 EXT GATE to select rate or totalize mode. 6 BATTERY SCALE Scale Factor to select scale factor as a divider or multiplier and to enter scale factor value. Offset Value to select offset value. Reading on meter = input x scale + offset dEC Pt Decimal Point to select position of decimal point in the digital display. SP LO SP HI Setpoints Low and High to enter setpoints. Shows only if the relay option board is installed. COnFIG Configuration to program internal software switches to specify analog, output, RS-232 input or alarm conditions An LO Analog Low to obtain reading for low analog output. An HI Analog High to obtain reading for high analog output. EXCITATION = 12.4V POSITIVE THRESHOLD 1K PULL DOWN RESISTOR TO RTN 243K PULL UP RESISTOR TO +EXC OFFSEt EXC OUT 2 SIG IN 3 SIG RTN 4 RST IN 5 EXT GATE 6 BATTERY 12.4V EXC 1 2 3 4 5 6 7 8 9 10 OPEN HIGH LEVEL SIGNAL POSITIVE THRESHOLD LOW PASS FILTER 1K PULL DOWN RESISTOR 243K PULL UP RESISTOR TO +EXC OPEN OPEN OPEN CLOSED CLOSED CLOSED OPEN OPEN OPEN OPEN 2-Wire Contact Closure S1 TB2 1 EXC OUT 2 SIG IN 3 SIG RTN 4 5 EXT GATE 6 BATTERY EXCITATION= 5.0V 243K PULL-UP RESISTOR TO +EXC 1 2 3 4 5 6 7 8 9 10 OPEN LOW LEVEL SIGNAL ZERO THRESHOLD NO FILTER NoStor No Store to save to the volatile (active) memory StorE Store to save to the nonvolatile (permanent) memory. RST IN CLOSED Time-delay, UL 248-14 listed 125 mA (115 Vac line) 63 mA (230 Vac line) Time-lag, IEC 127-2 recognized 100 mA (115 Vac line) AC Powered Unit Connections 50 mA (230 Vac line) A S . C A/SC TB2 1 Mount the Unit External Fuse Required: OPEN OPEN OPEN CLOSED OPEN CLOSED OPEN OPEN OPEN OPEN 3-Wire Sensor Open Collector (PNP) S1 CAUTION: Do not connect AC power to your device until you have completed all input and output connections. This device must only be installed by a specially trained electrician with corresponding qualifications. Failure to follow all instructions and warnings may result in injury! 1. Cut a panel opening using the dimensions shown to the right. 2. Position the unit in the opening, making sure the front bezel is flush with the panel. 3. Install retaining clips on both sides of the meter and tighten against the panel. 1 2 3 4 5 6 7 8 9 10 OPEN 1. Remove the panel at the back of the unit. 2. Locate the TB1 connector. 3. Insert the correct wire in each terminal as shown in the following figure and tighten the lockdown screws. 4. Tug gently on the wires to verify the connections. EXC OUT 2 CLOSED Wiring 1 CLOSED The instrument is a panel mount device protected in accordance with EN61010-1 (Safety requirements for electrical equipment for measurement, control, and laboratory standard). Remember that the unit has no power-on switch. Building installation should include a switch or circuit-breaker that must be compliant to IEC 947-1 and 947-3. SAFETY: • Do not exceed voltage rating on the label located on the top of the instrument housing. • Always disconnect power before changing signal and power connections. • Do not use this instrument on a work bench without its case for safety reasons. • Do not operate this instrument in flammable or explosive atmospheres. • Do not expose this instrument to rain or moisture. EMC: • Whenever EMC is an issue, always use shielded cables. • Never run signal and power wires in the same conduit. • Use signal wire connections with twisted-pair cables. • Install Ferrite Bead(s) on signal wire close to the instrument if EMC problems persist. Using the Menus CLOSED OPEN CLOSED OPEN OPEN OPEN OPEN OPEN OPEN OPEN 2-Wire Magnetic Pickup S1 For complete wiring information on all setup options, please refer to the Operator’s Manual. To Select Operating Mode (Rate or Total): 1. Press MENU until the display shows Func . Func RECALL 2. Press SET to show the current mode. SET to Press select between rAtE and totAL . MENU RESET /SET /ADV MENU RESET totAL RECALL 4. /ADV rate RECALL 3. /SET Press MENU to store the mode into volatile memory. /SET /ADV MENU RESET store RECALL /SET /ADV MENU RESET To store the operating mode (or any value) into non-volatile memory: • Press MENU until the meter displays noStor • Press /SET once to display StorE . • Press MENU again. The value is stored in non-volatile memory. .
* Your assessment is very important for improving the work of artificial intelligence, which forms the content of this project