OM-CP-PRTEMP140 Xxxxx Xxxxxx High Temperature and Pressure Xxxxx Xxxxxx Xxxxxxxx Data Logger INSTRUCTION SHEET MQS5093/1216 M0000/0016 Shop online at omega.com SM e-mail: firstname.lastname@example.org For latest product manuals: www.omegamanual.info NPT Top Flush Top OM-CP-PRTEMP140 Product Overview The OM-CP-PRTEMP140 is a pressure and high temperature data logger all in one, created specifically for use in autoclave validation, mapping studies and pressurized processes. Featuring the familiar body style of the 140 series of data loggers and made of rugged stainless steel, this dual purpose device is less than 2.5 inches in length, allowing it to fit discretely inside small spaces. The OM-CP-PRTEMP140 is able to measure and record temperatures from -20°C to +140°C, and pressure from 0 bar to 5 bar, accommodating a wide range of applications even in the harshest conditions. Submergibility The OM-CP-PRTEMP140 is fully submersible and is rated IP68. O-Rings O-ring maintenance is a key factor when properly caring for the OM-CP-PRTEMP140. The O-rings ensure a tight seal and prevent liquid from entering the inside of the device. Note: This product is rated for use up to 140 °C. Please heed the battery warning. The product will explode if exposed to temperatures above 140 °C. 2 OM-CP-PRTEMP140 Troubleshooting Tips Why is the data logger not appearing in the software? If the OM-CP-PRTEMP140 doesn’t appear in the Connected Devices panel, or an error message is received while using the OM-CP-PRTEMP140, try the following: • Check that the OM-CP-IFC400 is properly connected. For more information, see Troubleshooting Interface Cable problems (below). • Ensure that the battery is not discharged. For best voltage accuracy, use a voltage meter connected to the battery of the device. If possible, try switching the battery with a new OM-CP-BAT110. • Ensure that no other Omega software is running in the background. • Ensure that Omega Software is being used. • Ensure that the Connected Devices panel is large enough to display devices. This can be verified by positioning the cursor on the edge of the Connected Devices panel until the resize cursor appears, then dragging the edge of the panel to resize it. The screen layout may also be reset in the options menu by selecting File, Options, and scrolling to the bottom. Troubleshooting Interface Cable problems Check that the software properly recognizes the connected OM-CP-IFC400. If the data logger is not appearing in the Connected Devices list, it may be that the OM-CP-IFC400 is not properly connected. 1. In the software, click the File button, then click Options. 2. In the Options window, click Communications. 3. The Detected Interfaces box will list all of the available communication interfaces. If the OM-CP IFC400 is listed here, then the software has correctly recognized and is ready to use it. Check that Windows recognizes the connected OM-CP-IFC400. If the software does not recognize the OM-CP-IFC400, there may be a problem with Windows or the USB drivers. 1. In Windows, click Start, right-click Computer and choose Properties or press Windows+Break as a keyboard shortcut. 2. Click Device Manager in the left hand column. 3. Double click Universal Serial Bus Controllers. 4. Look for an entry for Data logger Interface. 5. If the entry is present, and there are no warning messages or icons, then windows has correctly recognized the connected OM-CP-IFC400. 6. If the entry is not present, or has an exclamation point icon next to it, the USB drivers may need to be installed. These are available on the software flash drive included with the OM-CP-IFC400. Ensure that the USB end of the OM-CP-IFC400 is securely connected to the computer. 1. Locate the USB-A plug of the OM-CP-IFC400. 2. If the interface cable is connected to the PC, unplug it. Wait ten seconds. 3. Reconnect the cable to the PC. 4. Check to make sure that the red LED is lit, indicating a successful connection. 3 OM-CP-PRTEMP140 Installation Guide Installing the Interface cable - OM-CP-IFC400 or OM-CP-IFC406 Refer to the “Quick Start Guide” included in the package. Installing the software Insert the Omega Software Flash Drive in an open USB port. If the autorun does not appear, locate the drive on the computer and double click on Autorun.exe. Follow the instructions provided in the Installation Wizard. Device Operation Connecting and Starting the data logger 1. Once the software is installed and running, plug the interface cable into the docking station. 2. Connect the USB end of the interface cable into an open USB port on the computer. Place the data logger into the docking station. 3. The data logger will automatically appear under Connected Devices within the software. 4. For most applications, select “Custom Start” from the menu bar and choose the desired start method, reading rate and other parameters appropriate for the data logging application and click “Start”. (“Quick Start” applies the most recent custom start options, “Batch Start” is used for managing multiple loggers at once, “Real Time Start” stores the dataset as it records while connected to the logger.) 5. The status of the device will change to “Running”, “Waiting to Start” or “Waiting to Manual Start”, depending upon your start method. 6. Disconnect the data logger from the docking station and place it in the environment to measure. Note: The device will stop recording data when the end of memory is reached or the device is stopped. At this point the device cannot be restarted until it has been re-armed by the computer. Downloading data from a data logger 1. Connect the logger to the docking station. 2. Highlight the data logger in the Connected Devices list. Click “Stop” on the menu bar. 3. Once the data logger is stopped, with the logger highlighted, click “Download”. You will be prompted to name your report. 4. Downloading will offload and save all the recorded data to the PC. Product Maintenance Battery Replacement Materials: OM-CP-BAT110 1. Unscrew the bottom of the logger and remove the battery. 2. Place the new battery into the logger. Note the polarity of the battery. 3. Screw the cover back onto the logger. Recalibration The OM-CP-PRTEMP140 standard calibration points are +30°C and +140°C for the temperature channel and 1 bar, 2.5 bar and 5 bar for the pressure channel. Recalibration is recommended annually for any Omega data logger; a reminder is automatically displayed in the software when the device is due. 4 OM-CP-PRTEMP140 OM-CP-PRTEMP140 General Specifications Pressure Sensor Strain Gauge Pressure Range 0 bar to 5 bar Pressure Resolution 0.0001 bar Calibrated Accuracy ±0.03 bar (+20°C to +140°C) Temperature Sensor Resistance Temperature Detector (RTD) Temperature Range -20°C to +140°C (-4°F to +284°F) Temperature Resolution 0.01°C Calibrated Accuracy ±0.1°C (+20°C to +140°C) Memory 32,767 readings Reading Rate 1 reading every second up to 1 reading every 24 hours Battery Life 2 years typical (1 minute reading rate) Required Interface Package OM-CP-IFC400 or OM-CP-IFC406 USB docking station required Operating System XP SP3/Vista/Windows 7/Windows 8 /10 Compatibility Omega Software Standard software version 126.96.36.199 or later Compatibility Secure software version 188.8.131.52 or later Operating Environment -20°C to +140°C (-4°F to +284°F), 0%RH to 100%RH, 0.002 PSIA to 100 PSIA Material 316 Stainless Steel Dimensions (Body) Weight Flush Top: 1.98” x 0.97” x 0.97” (50.2mm x 24.6mm x 24.6mm) NPT Top: 2.3” x 0.97” x 0.97” (58.2mm x 24.6mm x 24.6mm) Flush Top: 90g (3.2oz) NPT Top: 100g (3.5oz) IP Rating IP68 Approvals CE Battery Warning WARNING: FIRE, EXPLOSION, AND SEVERE BURN HAZARD. DO NOT SHORT CIRCUIT, CHARGE, FORCE OVER DISCHARGE, CRUSH, PENETRATE OR INCINERATE. BATTERY MAY LEAK OR EXPLODE IF HEATED ABOVE 150°C (302°F). Specifications subject to change. See Omega’s terms and conditions at www.omega.com 5 omega.com email@example.com U.S.A. Headquarters: Servicing North America: Omega Engineering, Inc. Toll-Free: 1-800-826-6342 (USA & Canada only) Customer Service: 1-800-622-2378 (USA & Canada only) Engineering Service: 1-800-872-9436 (USA & Canada only) Tel: (203) 359-1660 Fax: (203) 359-7700 e-mail: firstname.lastname@example.org For Other Locations Visit omega.com/worldwide The information contained in this document is believed to be correct, but OMEGA accepts no liability for any errors it contains, and reserves the right to alter specifications without notice. WARRANTY/DISCLAIMER OMEGA ENGINEERING, INC. warrants this unit to be free of defects in materials and workmanship for a period of 13 months from date of purchase. OMEGA’s WARRANTY adds an additional one (1) month grace period to the normal one (1) year product warranty to cover handling and shipping time. This ensures that OMEGA’s customers receive maximum coverage on each product. If the unit malfunctions, it must be returned to the factory for evaluation. OMEGA’s Customer Service Department will issue an Authorized Return (AR) number immediately upon phone or written request. Upon examination by OMEGA, if the unit is found to be defective, it will be repaired or replaced at no charge. OMEGA’s WARRANTY does not apply to defects resulting from any action of the purchaser, including but not limited to mishandling, improper interfacing, operation outside of design limits, improper repair, or unauthorized modification. This WARRANTY is VOID if the unit shows evidence of having been tampered with or shows evidence of having been damaged as a result of excessive corrosion; or current, heat, moisture or vibration; improper specification; misapplication; misuse or other operating conditions outside of OMEGA’s control. Components in which wear is not warranted, include but are not limited to contact points, fuses, and triacs. OMEGA is pleased to offer suggestions on the use of its various products. However, OMEGA neither assumes responsibility for any omissions or errors nor assumes liability for any damages that result from the use of its products in accordance with information provided by OMEGA, either verbal or written. OMEGA warrants only that the parts manufactured by the company will be as specified and free of defects. OMEGA MAKES NO OTHER WARRANTIES OR REPRESENTATIONS OF ANY KIND WHATSOEVER, EXPRESSED OR IMPLIED, EXCEPT THAT OF TITLE, AND ALL IMPLIED WARRANTIES INCLUDING ANY WARRANTY OF MERCHANTABILITY AND FITNESS FOR A PARTICULAR PURPOSE ARE HEREBY DISCLAIMED. LIMITATION OF LIABILITY: The remedies of purchaser set forth herein are exclusive, and the total liability of OMEGA with respect to this order, whether based on contract, warranty, negligence, indemnification, strict liability or otherwise, shall not exceed the purchase price of the component upon which liability is based. In no event shall OMEGA be liable for consequential, incidental or special damages. CONDITIONS: Equipment sold by OMEGA is not intended to be used, nor shall it be used: (1) as a “Basic Component” under 10 CFR 21 (NRC), used in or with any nuclear installation or activity; or (2) in medical applications or used on humans. Should any Product(s) be used in or with any nuclear installation or activity, medical application, used on humans, or misused in any way, OMEGA assumes no responsibility as set forth in our basic WARRANTY / DISCLAIMER language, and, additionally, purchaser will indemnify OMEGA and hold OMEGA harmless from any liability or damage whatsoever arising out of the use of the Product(s) in such a manner. RETURN REQUESTS / INQUIRIES Direct all warranty and repair requests/inquiries to the OMEGA Customer Service Department. BEFORE RETURNING ANY PRODUCT(S) TO OMEGA, PURCHASER MUST OBTAIN AN AUTHORIZED RETURN (AR) NUMBER FROM OMEGA’S CUSTOMER SERVICE DEPARTMENT (IN ORDER TO AVOID PROCESSING DELAYS). The assigned AR number should then be marked on the outside of the return package and on any correspondence. The purchaser is responsible for shipping charges, freight, insurance and proper packaging to prevent breakage in transit. FOR WARRANTY RETURNS, please have the following information available BEFORE contacting OMEGA: 1. Purchase Order number under which the product was PURCHASED, 2. Model and serial number of the product under warranty, and 3. Repair instructions and/or specific problems relative to the product. FOR NON-WARRANTY REPAIRS, consult OMEGA for current repair charges. Have the following information available BEFORE contacting OMEGA: 1. Purchase Order number to cover the COST of the repair, 2. Model and serial number of the product, and 3. Repair instructions and/or specific problems relative to the product. OMEGA’s policy is to make running changes, not model changes, whenever an improvement is possible. This affords our customers the latest in technology and engineering. OMEGA is a registered trademark of OMEGA ENGINEERING, INC. © Copyright 2016 OMEGA ENGINEERING, INC. All rights reserved. This document may not be copied, photocopied, reproduced, translated, or reduced to any electronic medium or machine-readable form, in whole or in part, without the prior written consent of OMEGA ENGINEERING, INC.
* Your assessment is very important for improving the work of artificial intelligence, which forms the content of this project