WARRANTY/DISCLAIMER FOR NON-WARRANTY REPAIRS, consult OMEGA for current repair charges. Have the following information available BEFORE contacting OMEGA: 1. Purchase Order number to cover the COST of the repair or calibration, 2. Model and serial number of the product, and 3. Repair instructions and/or specific problems relative to the product. OMEGA’s policy is to make running changes, not model changes, whenever an improvement is possible. This affords our customers the latest in technology and engineering. OMEGA is a trademark of OMEGA ENGINEERING, INC. © Copyright 2018 OMEGA EN GIN EERIN G, IN C. All rights reserved. This document may not be copied, photocopied, reproduced, translated, or reduced to any electronic medium or machine-readable form, in whole or in part, without the prior written consent of OMEGA ENGINEERING, INC. The information contained in this document is believed to be correct, but OMEGA accepts no liability for any errors it contains, and reserves the right to alter specifications without notice. MQS4088-0918 MODE QUICK START DISPLAY DISPLAY LCK LCK LCK MODE ––– Store temp data ––– ☞ FUNC * * ☞ The information contained in this document is believed to be correct but OMEGA Engineering, Inc. accepts no liability for any errors it contains, and reserves the right to alter specifications without notice. WARNING: These products are not designed for use in, and should not be used for, patient connected applications. PATENT NOTICE This product may be protected by one or more of the following patents: U.S. PAT. D357,194; B1 5,368,392; 5,524,984; 5,727,880; 5,465,838; 5,823,678; 5,823,679; 6,123,453; 6,267,500 B1; 6,341,891 B1; 6,377,400 B1; 6,540,398 B2 ; 6,614,830 B1; 6,633,434 B2; 6,659,639; / Canada 2,114,806; 2,116,055; 75811 D OMEGA ENGINEERING, INC./ Czech Republic 25372/ France 2 756 920; 2 767 921; 2,773,213; 0378411 to 0378446; 2 773 213 B1 / Germany M 94 06 478.4; G 94 22 197.9; G 94 22 203.7/ Italy RM940000913/ Japan 988,378/Holland 1007752; 25009-00/ Spain mod. ut. 0133292/ Slovak Republic 24565/ U.K. Registered 2041153; 9726133.3/ EPO 0 644,408 B2; EP 1 085 307 A1. Other U.S. and Foreign Patents Pending. Other U.S. and International Patents Pending. (Models OS530LE, OS530HRE) * Shown with Built-in Distance Measuring Option and Digital Video Camera Attachment FUNC ☞ FUNC (Model OS532E) LCK OS530LE OS532E OS533E OS534E OS530HRE OS53x-CF OS523E OS524E LCK ☞ * LAL FUNC ☞ LCK FUNC LCK ATC ☞ * FUNC ☞ LCK FUNC LCK PRN ☞ * OMEGASCOPETM Handheld FUNC ☞ LCK ☞ Infrared Thermometer FUNC (Model OS533E) LCK FUNC ☞ LCK ☞ It is the policy of OMEGA to comply with all worldwide safety and EMC/EMI regulations that apply. OMEGA is constantly pursuing certification of its products to the European New Approach Directives. OMEGA will add the CE mark to every appropriate device upon certification. FUNC LCK * OS530LE, OS530HRE OS532E OS533E OS534E, OS523E, OS524E HAL LCK NOTE: The unit of measure (°F or °C) flashes during Real Time Mode. Go to Turn on/off Logging Go to Go to Go to Set Data Transmission interval (Logging) Set Ambient Target temp Set Low Alarm set point Enable/Disable LAL Enable/Disable ATC Enable/Disable PRN Go to Go to Enable/Disable HAL –––––– Go to Go to Go to Turn on LCD Back lite Change °F to °C Set High Alarm set point F Reset Maximum, Minimum, Diff. Average temps Go to Go to Set laser to Flashing or On Go to Go to Press C to change from °F to °C or feet to meter and vise versa to turn on/off LCD backlight Press Set Emissivity Lock or unlock Trigger LCK Go to ☞ Emissivity Current temperature Distance (feet or meter) Current temperature Laser status Current temperature Maximum temperature Current temperature Minimum temperature Current temperature Differential temp Current temperature Average temperature Current temperature High alarm set point Current temperature Thermocouple temp Current temperature Low alarm set point Current temperature Ambient target temp Current temperature Data Trans. Interval Current Temperature Memory location Current temperature Logging Current temperature Display shows: Press FUNC to... Press LOCK to... Press C F or to... Functional Flow Chart when the Trigger is Pulled (Real Time Mode) FUNC omega.com firstname.lastname@example.org FUNC Servicing North America: (Models OS534E, OS523E, OS524E) on LCK ☞ CAUTION FUNC • There should be a clean, open line of sight from the distance device to the target, otherwise an erroneous reading will result. • For accurate distance measurement readings, the surface must be hard, flat, and reflective to ultrasonic pulse. • Distance measurement can not be taken through glass, or off of soft and padded surfaces, or through smoke or fog. • The distance measurement unit must be held perpendicular to the target surface. • The distance measurement unit is designed for indoor use only. • Accuracy of the distance measurement unit will vary depending on environmental conditions. • Do not aim the distance measurement unit. AVOID EXPOSURE LASER RADIATION IS EMITTED FROM THIS APERTURE FOR WARRANTY RETURNS, please have the following information available BEFORE contacting OMEGA: 1. Purchase Order number under which the product was PURCHASED, 2. Model and serial number of the product under warranty, and 3. Repair instructions and/or specific problems relative to the product. 6 DISPLAY MODE: RETURN REQUESTS/INQUIRIES Direct all warranty and repair requests/inquiries to the OMEGA Customer Service Department. BEFORE RETURNING ANY PRODUCT(S) TO OMEGA, PURCHASER MUST OBTAIN AN AUTHORIZED RETURN (AR) N UMBER FROM OMEGA’S CUSTOMER SERVICE DEPARTMEN T (IN ORDER TO AVOID PROCESSIN G DELAYS). The assigned AR number should then be marked on the outside of the return package and on any correspondence. 5 Real Time Modes OMEGA EN GIN EERIN G, IN C. warrants this unit to be free of defects in materials and workmanship for a period of 13 months from date of purchase. OMEGA’s WARRAN TY adds an additional one (1) month grace period to the normal one (1) year product warranty to cover handling and shipping time. This ensures that OMEGA’s customers receive maximum coverage on each product. If the unit malfunctions, it must be returned to the factory for evaluation. OMEGA’s Customer Service Department will issue an Authorized Return (AR) number immediately upon phone or written request. Upon examination by OMEGA, if the unit is found to be defective, it will be repaired or replaced at no charge. OMEGA’s WARRAN TY does not apply to defects resulting from any action of the purchaser, including but not limited to mishandling, improper interfacing, operation outside of design limits, improper repair, or unauthorized modification. This WARRANTY is VOID if the unit shows evidence of having been tampered with or shows evidence of having been damaged as a result of excessive corrosion; or current, heat, moisture or vibration; improper specification; misapplication; misuse or other operating conditions outside of OMEGA’s control. Components in which wear is not warranted, include but are not limited to contact points, fuses, and triacs. OMEGA is pleased to offer suggestions on the use of its various products. However, OMEGA neither assumes responsibility for any omissions or errors nor assumes liability for any damages that result from the use if its products in accordance with information provided by OMEGA, either verbal or written. OMEGA warrants only that the parts manufactured by the company will be as specified and free of defects. OMEGA MAKES NO OTHER WARRANTIES OR REPRESENTATIONS OF ANY KIND WHATSOEVER, EXPRESSED OR IMPLIED, EXCEPT THAT OF TITLE, AND ALL IMPLIED WARRANTIES INCLUDING ANY WARRANTY OF MERCHANTABILITY AND FITNESS FOR A PARTICULAR PURPOSE ARE HEREBY DISCLAIMED. LIMITATION OF LIABILITY: The remedies of purchaser set forth herein are exclusive, and the total liability of OMEGA with respect to this order, whether based on contract, warranty, negligence, indemnification, strict liability or otherwise, shall not exceed the purchase price of the component upon which liability is based. In no event shall OMEGA be liable for consequential, incidental or special damages. CONDITIONS: Equipment sold by OMEGA is not intended to be used, nor shall it be used: (1) as a “Basic Component” under 10 CFR 21 (N RC), used in or with any nuclear installation or activity; or (2) in medical applications or used on humans. Should any Product(s) be used in or with any nuclear installation or activity, medical application, used on humans, or misused in any way, OMEGA assumes no responsibility as set forth in our basic WARRAN TY/DISCLAIMER language, and, additionally, purchaser will indemnify OMEGA and hold OMEGA harmless from any liability or damage whatsoever arising out of the use of the Product(s) in such a manner. CAUTION OMEGASCOPE TM LASER RADIATION - DO NOT STARE INTO BEAM OUTPUT <1 mW, WAVELENGTH 630-670 nm CLASS II (2) LASER PRODUCT, COMPLIES WITH FDA 21CFR 1040.10 & EN60825-1/11.2001 Warning and Certification Label U.S.A. Omega Engineering, Inc. Headquarters: Toll-Free: 1-800-826-6342 (USA & Canada only) Customer Service: 1-800-622-2378 (USA & Canada only) Engineering Service: 1-800-872-9436 (USA & Canada only) Tel: (203) 359-1660 Fax: (203) 359-7700 e-mail: email@example.com For Other Locations Visit omega.com/worldwide TM OMEGASCOPE 2 START HERE 4 3 Using this Quick Start Manual OMEGASCOPE Use this Quick Start Manual with your OS530E/ OS520E series Handheld Infrared Thermometer to set up and perform basic operations. These tasks are: 2 Types of Laser Beams TM Laser Dot • Installing the Batteries • Operating the Laser Sight • Taking Temperature Readings • Measuring Distance For detailed information, refer to the User’s Guide (M4088). LCK ATC HAL LOBAT LAL PRN °F °C Laser Circle Figure 5. Two Laser Configurations The thermometer is ready for operation. Laser Power Switch Taking Temperature Readings 1. TM Getting Started FUNC Parts of the Thermometer C Digital/Video Camera (Optional) LOCK Aim at the target. If you are not using the Laser Sighting, use the “V” groove on top of the thermometer to align the target to the thermometer’s field of view. Field of View F Figure 3. Display and Keypad View Lens Rubber Boot Built-in Distance Module (Optional) Laser Beam Aperture Backlit LCD F ac Adapter Input Jack Wrist Strap Figure 1. OS530E/OS520E Series Handheld Infrared Thermometer Front View Pull and hold the trigger. To use the laser sighting, set the laser power switch to the ON position. The Power Indicator LED and the laser beam will turn on. The laser beam will stay on as long as the trigger is pulled. The target temperature and emissivity (E) will be displayed and C on the LCD. You can press the key to increment or decrement the target emissivity. The laser beam is switchable between dot and circle using the switch on the right side. Thermocouple Input Socket (SMP) (standard on OS532E, OS533E, OS534E) Battery Compartment Door (UNACCEPTABLE) Figure 6. Field of View Positions 2. Laser Dot/Circle Switch Trigger (ACCEPTABLE) Invert the unit and install 4 fresh AA size batteries as shown in Figure 4. Display Rubber Boot Distance Power Switch Tripod Mount Target Installing the Batteries “V” Groove Analog Output Jack (1mV/deg) RS-232 Phone Jack (standard on OS533E, OS534E, OS523E, OS524E) Figure 2. OS530E/OS520E Series Handheld Infrared Thermometer Various Views Measuring Distance Figure 4. Installing the Batteries Operating the Laser Sight Function WARNING: DO NOT AIM THE LASER BEAM AT ANYONE’S EYES. CAUTION: Use of controls or adjustments or performance of procedures other than those specified here may result in hazardous radiation exposure. Refer to the Operator’s Manual for complete warnings and cautions. Distance measurement is offered either as field mountable or built-in. The field mountable version (HH-DM) is a stand alone device. Press the key to measure distance. The build-in version (-DM) is an integral part of the thermometer. Go to the d_F or d_M display menu. Set the power switch on the distance module to the ON position. Pull the trigger for about 2 seconds. The upper display will show the distance in feet or meters. Releasing the trigger will turn off distance measurement.
* Your assessment is very important for improving the work of artificial intelligence, which forms the content of this project