Model 1102 Low Capacity Torque Sensor DESCRIPTION A line of low capacity rotary slip ring torque sensors primarily designed for fractional horsepower motor test applications. Instrumental in monitoring start up, running, and stall torque levels during development and qualification of these motors. Capacity range from 50 in-oz to 1000 in-oz for small shaft mounting configurations. FEATURES • 50 in-oz to 1000 in-oz capacity range • 0.1 % non-linearity and hysteresis • Integral foot mount (standard) • Lower cost for general “in-line” applications •Can be used with almost all existing dc and ac signal conditioning equipment Model 1102 PERFORMANCE SPECIFICATIONS Characteristic Measure Torque range 50, 100, 200, 500, 1000 oz-in Non-linearity ±0.1 % of rated output Hysteresis ±0.1 % of rated output Repeatability ±0.05 % of rated output Output @ rated capacity ±2 mV/V (nominal) TYPICAL SYSTEM DIAGRAM Customer Supplied Chart Recorder Alarm Panel Data Acquisition Computer PLC In-Line Amplifiers (Used with unamplified units only) Output Amplifier Universal In-Line Amplifiers +/-5 VDC UV +/-10 VDC UV-10 0 +/-5 VDC UBP 4-20mA (3 wire) U3W 4-20mA (2 wire) U2W DIN rail Mount In-Line Amplifiers 0 +/- 5 VDC (3 wire) DV-05 4-20mA (3 wire) DA-05 0 +/- 10 VDC (3 wire) DV10 Power Supply Honeywell In-Line Amplifier ENVIRONMENTAL SPECIFICATIONS Characteristic Measure Temperature, operating -29 °C to 93 °C [-20 °F to 200 °F] Temperature, compensated 21 °C to 77 °C [70 °F to 170 °F] Temperature effect, zero ±0.002 % of rated output/°F Temperature effect, output ±0.002 % of reading/°F Output Input Serial Number For long cable runs or electrically noisy environments an amplified sensor or locally mounted amplifier is often preferred 2003.1 lb. L1 L2 L3 L4 10070-1 Optional Calibration Arm OR VALUE CLEAR UP DOWN SETUP CHAN TARE ENTER EXIT OR OR Lebow Products MENU Optional Speed Sensor 24368-1 ELECTRICAL SPECIFICATIONS Characteristic Measure Excitation (maximum) 20 Vdc or Vac RMS Insulation resistance > 5000 mOhm @ 50 Vdc Bridge resistance 350 ohm (nominal) Number of bridges 1 Zero balance ±1.0 % of rated output Electrical termination PT02E-10-6P MODEL 1102 TEST TARE HOLD ESC ENTER RESET Display Units SC500 SC2001 GM 7541 SC2000 SC3004 NK 7558 HH 7561-PSD Mating Connectors & Cables 023-LW181-023-LF Mating connector 023-LW180-034-LF Mating connector, standard speed pickup 7205-76-XX* Mating connector & 6 conductor cable (unamplified unit with sense leads but not shunt cal) 7205-75-XX** Mating connector & 4 conductor cable (unamplified unit without sense leads but not shunt cal) 7205-83-XX* Mating connector & 6 conductor cable (for connection to instrument 7541 ) 7205-82-XX** Mating connector & 4 conductor cable (for connection to instrument 7541 ) 7204-00-XX* Speed sensor cable, pigtail leads at instrument 7204-16-XX* Speed sensor cable to instrument 7541 * XX represents length in feet 100ft maximum ** XX represents length in feet 20ft maxim um 2 Honeywell • Sensing and Control VIEW MODEL 7541 Low Capacity Torque Sensor MECHANICAL SPECIFICATIONS oz-in N-m Max speed RPM Overload capacity (% of nominal capacity) Torsional stiffness lb-in/rad Rotating inertia lb-in sec2 Weight (lb) 50 0.35 20000 150 665 1.75 x 10-3 2 100 0.70 20000 150 1070 1.75 x 10-3 2 200 1.50 20000 150 1790 1.76 x 10 -3 2 500 3.50 20000 150 3480 1.77 x 10-3 2 1000 7.00 20000 150 4850 1.78 x 10 2 -3 MOUNTING DIMENSIONS C cm [in] L cm [in] N cm [in] P cm [in] U cm [in] K cm [in] A cm [in] B cm [in] D cm [in] E cm [in] F cm [in] H cm [in] 16,50 [6.50] 5,70 [2.25] 2,50 [1.00] 8,90 [3.50] 0,95 [0.37] 1,90 flat [0.75 flat] 0,86 [0.34] 10,50 [4.13] 4,44 [1.75] 4,44 [1.75] 2,50 [1.00] 0,50 [0.20] P MANUAL BRUSH LIFTER STANDARD PTO2E-10-6P OPTIONAL SPEED SENSOR U DIA. K A H 4 MOUNTING HOLES D F N F N E E L C B For reference only Honeywell • Sensing and Control 3 Model 1102 Low Capacity Torque Sensor Warranty. Honeywell warrants goods of its manufacture as being free of defective materials and faulty workmanship. Honeywell’s standard product warranty applies unless agreed to otherwise by Honeywell in writing; please refer to your order acknowledgement or consult your local sales office for specific warranty details. If warranted goods are returned to Honeywell during the period of coverage, Honeywell will repair or replace, at its option, without charge those items it finds defective. The foregoing is buyer’s sole remedy and is in lieu of all warranties, expressed or implied, including those of merchantability and fitness for a particular purpose. In no event shall Honeywell be liable for consequential, special, or indirect damages. While we provide application assistance personally, through our literature and the Honeywell web site, it is up to the customer to determine the suitability of the product in the application. WARNING PERSONAL INJURY • DO NOT USE these products as safety or emergency stop devices or in any other application where failure of the product could result in personal injury. Failure to comply with these instructions could result in death or serious injury. WARNING Find out more Honeywell serves its customers through a worldwide network of sales offices, representatives and distributors. For application assistance, current specifica- MISUSE OF DOCUMENTATION • The information presented in this datasheet is for reference only. DO NOT USE this document as product installation information. • Complete installation, operation and maintenance information is provided in the instructions supplied with each product. Failure to comply with these instructions could result in death or serious injury. tions, pricing or name of the nearest Authorized Distributor, contact your local sales office. To learn more about Honeywell’s test and measurement products, call +1-614-850-5000, visit http://measurementsensors. honeywell.com, or e-mail inquiries to email@example.com Sensing and Control Honeywell 1985 Douglas Drive North Golden Valley, MN 55422 USA honeywell.com 008769-2-EN IL50 GLO July 2014 Copyright © 2014 Honeywell International Inc. All rights reserved.
* Your assessment is very important for improving the work of artificial intelligence, which forms the content of this project