Liability for Material Defects Digital Communication Software (Optional) All components of the device have been checked and tested for functionality at the factory. However, if defects occur despite our careful quality control, MICRO-EPSILON or your dealer must be notified immediately. For a digital communication the optional USB programming kit is required. Installation Insert the installation CD into the according drive on your computer The liability for material defects is 12 months from delivery. Within this period, defective parts, except for wearing parts, will be repaired or replaced free of charge, if the device is returned to MICRO-EPSILON with shipping costs prepaid. Any damage that is caused by improper handling, the use of force or by repairs or modifications by third parties is not covered by the liability for material defects. Repairs are carried out exclusively by MICRO-EPSILON. The sensor is offering two ways of digital communication: -- Bidirectional communication (sending and receiving data) -- Unidirectional communication (burst mode – the sensor is sending data only) If the autorun option is activated the installation wizard will start automatically. Otherwise please start setup.exe from the CD-ROM. Follow the instructions of the wizard until the installation is finished. The installation wizard will place a launch icon on the desktop and in the start menu. If you want to uninstall the software from your system please use the uninstall icon in the start menu. Further claims can not be made. Claims arising from the purchase contract remain unaffected. In particular, MICRO-EPSILON shall not be liable for any consequential, special, indirect or incidental damage. In the interest of further development, MICRO-EPSILON reserves the right to make design changes without notification. Main Features -- Graphic display for temperature trends and automatic data logging for analysis and documentation -- Complete sensor setup and remote controlling -- Adjustment of signal processing functions -- Programming of outputs and functional inputs For translations into other languages, the German version shall prevail. You will find a detailed software manual on the CD. Minimum System Requirements Open Collector Output You can download a detailed instruction manual as PDF file from our web site under http://www.micro-epsilon.de/download/manuals/man--thermoMETERCS--com.pdf The open collector output is an additional alarm output on the thermoMETER CS and can control an external relay e.g. In addition the analog output can be used simultaneously. Direct Connection to an RS232 Interface on the PC MICRO-EPSILON MESSTECHNIK GmbH & Co. KG Königbacher Straße 15 94496 Ortenburg / Germany Tel. +49 8542 / 168-0 / Fax +49 8542 / 168-90 e-mail [email protected] www.micro-epsilon.com X9771203-A021118HDR *X9771203-A02* The thermoMETER CS works with a UART voltage of 3.3 V. For a bidirectional RS232 connection of the sensor the following interface circuits can be used: MAX3381E (manufacturer: Maxim), see Chapter Direct Connection to an RS232 Interface on the PC. ------ Windows XP, Windows Vista, Windows 7, Windows 8, Windows 10 USB interface Hard disc with at least 30 MByte free space At least 128 MByte RAM CD-ROM drive Assembly Instructions thermoMETER CS Functions The sensors of the CS series are non-contact infrared temperature sensors. They calculate the surface temperature based on the emitted infrared energy of objects. Unpacking, Included in Delivery ----- 1 Sensor thermoMETER CS 1 Connection cable 2 Mounting nuts 1 Assembly instructions Warnings Connect the power supply and the display/output device in accordance with the safety regulations for electrical equipment. >> Risk of injury, damage to or destruction of the sensor Technical Data Ambient temperature -20 ... +80 °C (-4 ... +176 °F) Storage temperature -40 ... +85 °C (-40 ... +185 °F) Relative humidity 10 ... 95 %, non condensing Dimensions M12x1, 85 mm long Cable length 1 m (standard), 3 m, 8 m, 15 m Output impedance min. 10 kΩ load impedance Current draw 10 mA Power supply 5 ... 30 VDC -40 ... 400 °C (scalable via software) 1 The output impedance must be ≥ 10 kΩ. 8 ... 14 µm Optical resolution 15:1 LED Functions >> Damage to or destruction of the sensor Warm-up time Protect the sensor cable against damage. >> Destruction of the sensor, failure of the measuring device Do not kink the sensor cable and bend the sensor cable in tight radius. The minimum bending radius is 14 mm (static). A dynamic movement is not allowed. >> Damage to the sensor cable, failure of the measuring device No solvent-based cleaning agents may have an effect on the sensor (neither for the optics nor the housing) ± 1.5 °C or ± 1.5 % of reading (whichever is greater) 10 min 1) Optional up to 1030 °C 2) At ambient temperature 23 ± 5 °C and object temperatures > 0 °C 3) Accuracy for thermocouple output ±2.5 °C or ±1 % Optical Chart (15:1) Electrical Installation i i Please use a stabilized power supply unit with an output voltage in the range of 5 – 30 VDC which can supply 100 mA. The residual ripple should be max. 200 mV. Supply the thermoMETER CS either via USB or externally via power supply and not simultaneously! - Use shielded cables only. The sensor shield has to be grounded. - The shield (black) on the CS is not connected to GND (brown). In any case it is necessary to connect the shield to ground or GND (whichever works best)! Analog Device (mV Output on OUT Pin) >> Damage to or destruction of the sensor The measuring system is designed for use in industrial environments and meets the requirements. The green LED can be programmed for the following functions. For the programming the USB adapter cable incl. software (option) is necessary. The factory default setting for the LED is self diagnostic. LED Alarm LED lights up if the object temperature exceeds or deceeds an alarm threshold. Automatic aiming support Sighting feature for an accurate aiming of the CS to hot or cold objects Self diagnostic LED is indicating different states of the sensor. Temperature code indication Indication of the object temperature via the LED Off LED deactivated. Factory Settings The sensor is supplied with factory settings, see also the Chapter Factory Settings in the Operating Instructions. The default settings can be changed with the optional USB kit (USB adapter cable + software). Notes on CE Marking The following apply to the thermoMETER CS: -- EU Directive 2014/30/EU -- EU-Directive 2011/65/EU, “RoHS“ Analog Device (Thermocouple Type K Output on OUT t/c K pins /only at Model CSTK Output on OUT-Pin) Spectral range Avoid shocks and impacts to the sensor. >> Damage to or destruction of the sensor The thermoMETER CS is equipped with a metric M12x1 thread and can be installed either directly via the sensor thread or with the help of the both hex nuts (standard) to the mounting bracket available. Temperature range Accuracy 2 3 The supply voltage must not exceed the specified limits. Mechanical Installation The output impedance must be ≥ 10 kΩ.
* Your assessment is very important for improving the work of artificial intelligence, which forms the content of this project