S-Digit mini FEATURES KIT CONSISTS OF: POWER • S-Digit mini On/Off switch • Padded bag ZERO • Batteries Calibrate the unit (see instructions below) • Users‘ manual DEG/GRAD Set reading to degrees (°) or percent (%) HOLD Current reading will be frozen by pressing HOLD. By pressing HOLD again reading will be released. FUNCTIONS TECHNICAL DATA Electronic slope measurer, for measuring • Working range 4 x 90° inclinations, slopes & plane surfaces. • Resolution 0,1 ° or 0,1 % • Accuracy 0° + 90° = ± 0,1° • 1° - 89° = ± 0,2° • Power supply 3 x 1,5 V AA • Operating time 600h • Size 156 x 56 x 31 mm • Weight 0,3 kg incl. batteries IMPORTANT Calibration of inclinometer: • Calibrate before first use • Before important measurements • After a hit or drop • After a large fluctuation in temperature Checking of calibration Position unit on a flat surface (position A) and switch on. Wait 10 sec. and note down measured value. Rotate unit by 180° to position B. Wait 10 sec. and note down measured value. If deviation between the two values is more than 0,1° S-Digit mini should be calibrated. How to calibrate Position unit on a flat surface (position A) and switch on. a) Wait 10 sec. – press ZERO for 5 sec. after display shows „-0-“ wait further 10 sec. b) Press ZERO (display shows „-1“) – rotate by 180° to position B (lay on same position as in A). c) Wait 10 sec. – press ZERO once more to finalize calibration process. Measurement of inclination The 2 arrows in the display indicate which direction S-Digit mini has to be moved in order to reach 0° or 90° position. The exact 0° or 90° position is shown by double arrows. Display has automatic digit inversion for overhead measurements. Batteries Open the battery compartment cover and put in batteries (take care of polarity). Important: Battery symbol flashes when batteries need to be replaced. Connection to camera tripod Special feature within the range of 0° - 30° O-setting in any position required: In the range between 0° With 1/4“-thread (at bottom and 30° the ZERO layer can be defined in any position. side) S-Digit mini can be Start calibration function in the position required (press ZERO connected to a camera longer than 5 sec.) - wait 10 sec. - press ZERO again - wait tripod. 10 sec. - press ZERO again - finalized. Do not turn the S-Digit mini during this process. Now the ZERO-layer is defined in the new position and the application required can be started. ATTENTION: We recommend to define the ZERO-layer in the horizontal position again when you have finished your application. Otherwise it may happen that for the following measurements the results can be completely wrong (e.g. when you work the next day or if another user is working with the unit). Care and cleaning • • • Safety instructions: Please handle measuring Please follow up instructions given in instruments with care. operators’ manual. Clean with soft cloth only after any • Use instrument for measuring jobs only. use. If necessary damp cloth with • Do not open instrument housing. some water. • Repairs should be carried out by If instrument is wet clean and dry it authorized workshops only. Please carefully. contact your local dealer. • Pack it up only if it is perfectly dry. • Transport in original container / case only. • Do not remove warning labels or safety instructions. • Keep instrument away from children. • Do not use instrument in explosive environment. Specific reasons for erroneous measuring results Electromagnetic acceptability (EMC) Measurements through glass or It cannot be completely excluded that this plastic windows; instrument will disturb other instruments • Dirty laser emitting windows; (e.g. navigation systems) will be disturbed by • After instrument has been dropped other instruments (e.g. intensive or hit. electromagnetic radiation nearby industrial Please check accuracy. facilities or radio transmitters). • • Large fluctuation of temperature: If instrument will be used in cold areas after it has been stored in warm areas (or the other way round) please wait some minutes before carrying out measurements. CE-Conformity Instrument has CE-mark according to EN 61326:1997, EN 55022, EN 61000-4-2/-3. Warranty • This product is warranted by the manufacturer to the original purchaser to be free from defects in material and workmanship under normal use for a period of two (2) years from the date of purchase. • During the warranty period, and upon proof of purchase, the product will be repaired or replaced (with the same or similar model at manufacturers option), without charge for either parts or labour. • In case of a defect please contact the dealer where you originally purchased this product. • The warranty will not apply to this product if it has been misused, abused or altered. • Without limiting the foregoing, leakage of the battery, bending or dropping the unit are presumed to be defects resulting from misuse or abuse. Exceptions from responsibility • The user of this product is expected to follow the instructions given in operators’ manual. Although all instruments left our warehouse in perfect condition and adjustment the user is expected to carry out periodic checks of the product’s accuracy and general performance. • The manufacturer, or its representatives, assumes no responsibility of results of a faulty or intentional usage or misuse including any direct, indirect, consequential damage, and loss of profits. • The manufacturer, or its representatives, assumes no responsibility for consequential damage, and loss of profits by any disaster (earthquake, storm, flood etc.), fire, accident, or an act of a third party and/or a usage in other than usual conditions. • The manufacturer, or its representatives, assumes no responsibility for any damage, and loss of profits due to a change of data, loss of data and interruption of business etc., caused by using the product or an unusable product. • The manufacturer, or its representatives, assumes no responsibility for any damage, and loss of profits caused by usage other than explained in the users‘ manual. The manufacturer, or its representatives, assumes no responsibility for damage caused by wrong movement or action due to connecting with other products.
* Your assessment is very important for improving the work of artificial intelligence, which forms the content of this project