Instruction Manual Professional Weather Center model 02032 / 888143 CONTENTS Unpacking Instructions............ 2 Package Contents................... 2 Product Registration................ 2 Features & Benefits: 5-in-1........ 3 Features: Display.................... 4 Back of Display....................... 6 Setup..................................... 7 5-in-1 Sensor Setup................. 7 Display Setup......................... 8 Set the Time, Date & Units....... 9 Backlight Settings.................... 9 Placement Guidelines.............10 5-in-1 Sensor Installation......... 11 AcuRite Connect....................12 Using the Weather Center......14 Rainfall Tracking....................15 Weather Ticker......................16 Troubleshooting.....................17 Care & Maintenance............. 20 Calibration........................... 20 Specifications........................ 22 FCC Information................... 22 Customer Support................. 23 Warranty............................. 23 Questions? Contact Customer Support at (844) 228-5465 or visit www.AcuRite.com. SAVE THIS MANUAL FOR FUTURE REFERENCE. Congratulations on your new AcuRite product. To ensure the best possible product performance, please read this manual in its entirety and retain it for future reference. Unpacking Instructions Remove the protective film that is applied to the LCD screen prior to using this product. Locate the tab and peel off to remove. Package Contents 1. Display with tabletop stand 2. 5-in-1 sensor 3. Sensor mounting bracket 4. Mounting hardware 5. Power adapter 6. USB cable 7. Instruction Manual 8. 4 Alkaline Batteries IMPORTANT PRODUCT MUST BE REGISTERED TO RECEIVE WARRANTY SERVICE PRODUCT REGISTRATION Register online to receive 1 year warranty protection www.AcuRite.com ►Register a Product 2 Features & Benefits 7 1 2 9 11 3 4 5 8 6 10 5-in-1 SENSOR 5. Temperature & Humidity Sensors (internal) 1. Rainfall Collector Funnel 2. Solar Cell Panel 6. Mounting Point Converts sunlight into power to run internal aspirating fan. Internal Aspirating Fan 7. Wind Speed Anemometer 8. Wind Direction Vane (not shown) Draws ambient air into 9. Mounting Bracket sensor to reduce solar radiation heating, resulting in more accurate temperature measurement. 10. Mounting Hardware Includes anchors & screws. Qty Diameter Length 5 #4 3/4” 1 #4 1/2” 1 #6 1/2” 3. A-B-C Switch ID code that must match display’s A-B-C switch to ensure units synchronize. 11. Debris Filter 4. Battery Compartment Pre-installed to prevent debris from entering the rain gauge. 3 Features & Benefits Display 1 23 4 1 5 33 32 31 30 29 6 7 8 9 10 11 12 13 28 27 26 25 14 15 24 16 17 23 22 21 20 4 19 18 1. Previous 2 Wind Directions 2. Backlight Button 20.SET Button For setup preferences. 21.“V“ Button For momentary backlight and dimmer settings. for setup preferences. 3. Current Wind Speed 4. Current Wind Direction 5. Current Indoor Temperature 22.SELECT Button Press to change the WEATHER SELECT category data being displayed. Arrow icon indicates direction temperature is trending. 23.Weather TickerTM 24.Heat Index/ Wind Chill Alert Indicator 6. Display Low Battery Indicator 7. Peak Wind Speed Activates when heat index rises over 90ºF (32ºC), or when wind chill drops below -17ºF (-27ºC). Highest speed from the last 60 minutes. 8. Current Indoor Humidity 25.Learning Mode Icon Arrow icon indicates direction humidity is trending. Disappears after weather forecast self-calibration is complete. 9. Wind Speed Alert Indicator 26.WEATHER SELECT Activates when wind speed is over 50 mph. Displays heat index, dew point, wind chill, feels like, days with rain this month, and days since last recorded rainfall. 10.Clock 11.Heavy Rainfall Warning Icon 27.Date 28.Current Outdoor Humidity Activated when over 1 inch of rainfall is recorded in 2 hours or less. 12.AUTO DIM Indicator 13.Current Rainfall Arrow icon indicates direction humidity is trending. 29.Average Wind Speed Rainfall recorded since 12:00am. of all speeds from past 2 minutes. 14.12 to 24 Hour Weather Forecast 30.5-in-1 Sensor Signal Strength 31.Sensor Low Battery Indicator 32.Current Outdoor Temperature Arrow icon indicates direction Self-Calibrating Forecasting pulls data from your 5-in-1 sensor to generate your personal forecast. 15.Storm Alert Indicator Activates when barometric pressure drops. temperature is trending. 33.Calibration Indicator 16.Month/Year/All-Time Rainfall 17.Current Barometric Pressure Appears next to calibrated values. Arrow icon indicates direction pressure is trending. 18.RAINFALL Button Press to change the RAINFALL data being displayed (Month, Year, All-Time). 19."T” Button For setup preferences. 5 Back of Display 1 2 4 3 ############ 6 7 5 9 8 BACK OF DISPLAY 7. 1. Plug-in for Power Adapter 2. Plug-in for USB Cable 3. Integrated Hang Hole for easy wall mounting 8. Power Adapter 9. USB Cable for PC Connection 4. MAC ADDRESS 12 digit unique ID required for online connectivity functions 10. Battery Compartment Cover (not shown) 5. Backup Battery Compartment 6. CLEAR RECORD/ RESET BUTTON Press and release to clear data recorded since 12:00 am. Press and HOLD for full reset to factory defaults. A-B-C Switch ID code that must match 5-in-1 sensor’s A-B-C switch to ensure units synchronize 6 SETUP 5-in-1 Sensor Setup 1 Set the A-B-C Switch Locate the A-B-C switch inside the battery compartment. Set the A-B-C switch to A, B or C. You must select the same letter choices for both the sensor and the display in order for the units to synchronize. 2 Install or Replace Batteries Batteries MUST be installed for this product to operate. AcuRite recommends high quality alkaline or lithium batteries for the best product performance. Heavy duty or rechargeable batteries are not recommended. cause alkaline batteries to function improperly. Use lithium batteries in the 5-in-1 sensor for temperatures below -4ºF / -20ºC. 1. Slide off the battery compartment cover. 2. Insert 4 x AA batteries into the battery compartment, as shown. Follow the polarity (+/-) diagram in the battery compartment. 3. Replace the battery cover. 3 Remove Rain Gauge Stabilizer Locate and remove the rain gauge stabilizer (plastic tab) taped into the bottom of the sensor. The rain gauge will not function until this is removed. The 5-in-1 sensor requires lithium batteries in low temperature conditions. Cold temperatures can ABC 1 A-B-C Switch Set to match display 2 3 Install Batteries 4 AA Batteries 7 Rain Gauge Stabilizer Remove and discard Display Setup 3 Write Down the MAC ADDRESS 1 Set the A-B-C Switch The A-B-C switch is located inside the battery compartment. It can be set to A, B or C. However, you must select the same letter choices for both the sensor and the display in order for the units to synchronize. 12 digit unique ID needed for online connectivity functions. 24 C8 6E 4 Install or Replace Backup Batteries (optional) Insert 3 x AA alkaline batteries into the battery compartment, as shown. Follow the polarity (+/-) diagram in the battery compartment. 2 Plug Power Adapter into Electric Outlet 3 1 A-B-C Switch set to match sensor Write Down MAC ADDRESS 12 digit number 4 Install Backup Batteries 3 AA Batteries ABC 2 Plug in Power Adapter IMPORTANT: Batteries are a backup power source to preserve records in the event of a power outage. Power adapter is the recommended primary power source to enjoy the full functionality of this product. PLEASE DISPOSE OF OLD OR DEFECTIVE BATTERIES IN AN ENVIRONMENTALLY SAFE WAY AND IN ACCORDANCE WITH YOUR LOCAL LAWS AND REGULATIONS. BATTERY SAFETY: Clean the battery contacts and also those of the device prior to battery installation. Remove batteries from equipment which is not to be used for an extended period of time. Follow the polarity (+/-) diagram in the battery compartment. Promptly remove dead batteries from the device. Dispose of used batteries properly. Only batteries of the same or equivalent type as recommended are to be used. DO NOT incinerate used batteries. DO NOT dispose of batteries in fire, as batteries may explode or leak. DO NOT mix old and new batteries or types of batteries (alkaline/standard). DO NOT use rechargeable batteries. DO NOT recharge non-rechargeable batteries. DO NOT short-circuit the supply terminals. 8 Set the Time, Date & Units Press the “ SE T ” button, located on the front of the display, to enter SET MODE. Once in set mode, the preference you are currently setting will blink on the display. To adjust the currently selected (flashing) item, press and release the “ T“ or “V “ buttons (press and HOLD to fast adjust). To save your adjustments, press and release the “ SE T ” button again to adjust the next preference. The preference set order is as follows: TOP DISPLAY: CLOCK HOUR CLOCK MINUTE CALENDAR MONTH CALENDAR DATE CALENDAR YEAR TEMPERATURE UNITS (ºF or ºC) WIND SPEED UNITS (MPH, KPH, Knots) RAINFALL UNITS (in or mm) PRESSURE UNITS (inHg or hPa) TICKER DISPLAY: LANGUAGE (English, French, Spanish) WEATHER TICKER SPEED (SLOW, MEDIUM, FAST) PC CONNECT MODE (1, 2, 3, 4) You will automatically exit SET MODE if no buttons are pressed for 20 seconds. Enter setup mode at any time by pressing the “ SE T ” button. Display Backlight Settings This weather station’s color display features three different lighting settings: High (100%) brightness, Medium (60%) brightness and Low (30%) brightness. Using battery power alone, the backlight is available momentarily for 10 seconds by pressing the “ ” button. When display is powered with the power adapter, backlight remains on at 100% brightness. Press “ ” button once to dim to 60% brightness; press again to dim to 30%, press a 3rd time to enter “AUTO DIM” mode. “AUTO DIM” will appear below the time. AUTO DIM MODE: Automatically adjusts display brightness based on time of day & year. MARCH 11- NOVEMBER 4 6:00 a.m. - 9:00 p.m. = 100% brightness 9:01 p.m. - 5:59 a.m.= 30% brightness NOVEMBER 5 - MARCH 10 7:30 a.m. - 7:00 p.m.= 100% brightness 7:01 p.m. - 7:29 a.m. = 30% brightness 9 Placement for Maximum Accuracy AcuRite sensors are sensitive to surrounding environmental conditions. Proper placement of both the display and the sensor are critical to the accuracy and performance of this product. Display Placement Place the display in a dry area free of dirt and dust. To ensure accurate temperature measurement, place out of direct sunlight and away from heat sources or vents. Display stands upright for tabletop use or is wall-mountable. 5-in-1 Sensor Placement The 5-in-1 sensor is designed to remain outdoors all year long. Choose an open location with no obstructions above or around the sensor for the most accurate measurements. Important Placement Guidelines Display and sensor must be within 330 feet (100 m) of each other. MAXIMIZE WIRELESS RANGE lace units away from large metallic items, thick walls, metal surfaces, or other P objects that may limit wireless communication. PREVENT WIRELESS INTERFERENCE lace both units at least 3 feet (.9 m) away from electronic devices (TV, P computer, microwave, radio, etc.). LOCATE AWAY FROM HEAT SOURCES Position sensor away from heaters, air conditioners, chimneys, exhaust vents, asphalt and concrete (surfaces that radiate heat). LOCATE AWAY FROM HUMIDITY SOURCES Avoid installing the sensor near pools, spas, or other bodies of water. Water sources may impact humidity accuracy. LOCATE AWAY FROM SPRINKLER HEADS DO NOT install the sensor where it will be sprayed by a sprinkler system. This may force water inside the sensor. LOCATE AWAY FROM WIND & RAIN OBSTRUCTIONS DO NOT mount the sensor with obstructions around it. Consider a location that is a wide open area, with few structures around to ensure accurate wind measurement. Visit us online to view installation photos and video, or learn more about AcuRite technology: www.AcuRite.com/5in1 10 View Video 5-in-1 Sensor Installation Guidelines INSTALLATION HEIGHT Mount the sensor at least 5 feet (1.5 meters) off the ground in an open area. Higher is better for wind measurement- the National Weather Service recommends 33 feet (10 meters) high! MOUNTING OPTIONS The included mounting bracket is designed to screw directly onto wooden posts or surfaces that are 2”x 4” or larger. The sensor can also be mounted directly to 3/4” steel pipe (available at hardware stores and home centers). LEVEL INSTALLATION Use the bubble level on the top of the sensor to ensure level installation for accurate wind and rain measurement. SOLAR CELL INSTALLATION Install the NO OBSTRUCTIONS ABOVE OR AROUND TH SOU sensor with the solar cell facing SOUTH. This ensures the cell receives as much sun as possible and orients wind direction. 5-in-1 Sensor Installation 1. F asten mounting base (included) to a post or pole (not included) using the 4 longer screws included in the hardware bag. 2. Insert the mounting base into the hole on the bottom of the sensor. 3. Make sure the arrows on the top of the sensor are pointed in the proper direction and the bubble level is centered. The solar cell should be facing south to properly orient the wind direction. 4. F asten the sensor into the mounting base using the 2 shorter screws included in the hardware bag. (1.5 meters) The 5-in-1 sensor is now ready to use. Clear Data Collected During Installation When installation is complete, clear erroneous data from the display unit by pressing the “CLEAR RECORD/ RESET” button, on the back of the display inside the battery compartment. This button will clear data recorded since 12:00 am. Basic Setup is Complete The 5-in-1 sensor will now synchronize with the display. It may take a few minutes for synchronization to complete. If both or one of the units appear to be functioning improperly, please refer to the troubleshooting section. 11 AcuRite Connect AcuRite Connect works with Windows PC’s to provide access to your Weather Center data in the following ways: • Data File: Display logs (or stores) data so that you can download it to a PC in a data file (CSV, or comma-separated values file). • Web Browser or Smartphone: Monitor sensor data remotely using AcuRite’s free AcuRite online software, or from a mobile device using the free AcuRite app, available from the iOS App Store or Google Play Store. • Weather Underground: This weather station is compatible for streaming data to the Weather Underground Personal Weather Station Network. Navigate to http://www.acurite.com/kbase/downloads/AcuRite-Connect.html for more information. USB Mode To set up AcuRite Connect, you must first select mode 3 on your display to gain access to all functionality. Modes 1, 2 and 4 are not used for the AcuRite Connect software. 1.While in normal viewing mode, press and release the “SET” button until “PC CONNECT MODE” appears on the Weather TickerTM screen. 2.Press and release the “T“ or “V “ buttons to select mode 3. 3.Press the “SET” button to confirm your selection. AcuRite Connect USB Mode is now set. Download the Software The AcuRite Connect software is available online as a free download. A manual (PDF) is included with the software download. Before setting up the AcuRite Connect software, verify that the display is ON with the power adapter plugged in. Next, connect the mini-USB (small end of the cable) into the USB port on the display, located inside the battery compartment. AcuRite Connect software to your computer by visiting: 1 Download http://www.acurite.com/kbase/downloads/ AcuRite-Connect.html Or, navigate to the link above using the following path: www.acurite.com > Support > Downloads > AcuRite Connect the .exe file that was downloaded in step 1, and follow the on-screen 2 Click instructions to install the software. 12 AcuRite Connect is installed on the computer, the initial configuration will 3 After begin automatically. When selecting where to save weather data downloaded 4 from the display, a folder named “AcuRite Weather Station” will be created on the computer. (PC) When prompted by the software to plug in the display, connect the USB (larger end of the cable) into a USB port on the computer. AcuRite Connect is installed and ready for use. Follow the instructions within the AcuRite Connect app to complete setup and begin using the software. 13 OPERATION Using the Professional Weather Center Learning Mode Self-Calibrating Forecasting use a unique algorithm to analyze changes in pressure over a time period (called Learning Mode) to determine your altitude. After 14 days, the Learning Mode icon disappears from the display screen. At this point, the self-calibrated pressure is tuned in to your location and the unit is ready for superior weather prediction. Weather Forecast AcuRite’s patented Self-Calibrating Forecasting provides your personal forecast of weather conditions for the next 12 to 24 hours by collecting data from the sensor in your backyard. It generates a forecast with pinpoint accuracy personalized for your exact location. STORM LIKELY PARTLY CLOUDY RAIN LIKELY CLOUDY SNOW LIKELY CHANCE OF MIXED RAIN & SNOW View the complete list of icons at www.AcuRite.com/acurite-icons Weather Select Weather Select displays data including wind chill, dew point, heat index, feels like, days with rain this month, and days since last recorded rainfall. To change the “Weather Select” category shown, press the “SELECT” button on the front of the display. Barometric Pressure Subtle variations in barometric pressure greatly affect the weather. This weather center displays the current pressure with an arrow icon to indicate the direction the pressure is trending (FALLING, STEADY, or RISING). 14 Rainfall Tracking This weather station features enhanced tracking of historical rainfall data. Rain accumulation data is recorded for today, over all-time, by year, by month, and by rainfall event. icon is shown Press the “RAINFALL” button to review rainfall records. The on the display while historical data is being viewed. Records are shown in the following order: PREVIOUS RAINFALL #1* PREVIOUS RAINFALL #2* PREVIOUS RAINFALL #3* CURRENT MONTH TOTAL RAINFALL PREVIOUS MONTH TOTAL RAINFALL 2 MONTHS AGO TOTAL RAINFALL CURRENT YEAR TOTAL RAINFALL PREVIOUS YEAR TOTAL RAINFALL ALL-TIME TOTAL RAINFALL (date shown is start date of the all-time total; the date the display was powered on) TOTAL RAINFALL FOR DATE SHOWN CURRENT YEAR TOTAL RAINFALL TOTAL RAINFALL FOR MONTH SHOWN PREVIOUS YEAR TOTAL RAINFALL ALL-TIME TOTAL RAINFALL *Previous rainfall is the amount of rain recorded from 12:00am to 11:59pm on the date shown. 15 Weather Ticker ™ The Weather Ticker automatically flashes your real-time weather information and alerts as text in the lower part of the display screen. The possible WEATHER TICKER messages are as follows: HEAT INDEX-XX WINDCHILL-XX DEW POINT-XX IT FEELS LIKE XX OUTSIDE 7 DAY HIGH TEMP. XX- MM/DD 7 DAY LOW TEMP. XX- MM/DD 30 DAY HIGH TEMP. XX- MM/DD 30 DAY LOW TEMP. XX- MM/DD ALL TIME HIGH TEMP. XXX...RECORDED MM/DD/YY ALL TIME LOW TEMP. XXX...RECORDED MM/DD/YY 24 HOUR TEMP. CHANGE +XX ALL TIME HIGH WIND XX MPH...RECORDED MM/DD/YY 7 DAY AVERAGE WIND XX MPH TODAY’S AVERAGE WIND XX MPH MOON- NEW MOON- WAXING CRESCENT MOON- FIRST QUARTER MOON- WAXING GIBBOUS MOON- FULL MOON- WANING GIBBOUS MOON- LAST QUARTER MOON- WANING CRESCENT INDOOR HUMIDITY OK INDOOR HUMIDITY HIGH INDOOR HUMIDITY LOW NEW LOW TEMP. RECORD XX NEW HIGH TEMP. RECORD XX NEW WIND RECORD TODAY XX CURRENT RAINFALL X.XX/HR. RAIN EVENT STARTED XX HRS. AGO SENSOR BATTERIES LOW DISPLAY BATTERIES LOW SENSOR SIGNAL LOST...CHECK BATTERIES AND PLACEMENT CAUTION- HEAT INDEX IS XXX CAUTION- WIND CHILL IS XXX CALIBRATION DATA LOGGER FULL...CONNECT TO PC TO TRANSFER 16 Troubleshooting Problem No outdoor sensor reception no bars Outdoor temperature is flashing or showing dashes Inaccurate forecast Possible Solution •R elocate the display and/or the 5-in-1 sensor. The units must be within 330 ft (100 m) of each other. •M ake sure both units are placed at least 3 feet (.9 m) away from electronics that may interfere with the wireless communication (such as TVs, microwaves, computers, etc). •U se standard alkaline batteries (or lithium batteries in sensor when temperature is below -4ºF/-20ºC). Do not use heavy duty or rechargeable batteries. NOTE: It may take up to 20 minutes for display and sensor to synchronize after batteries are replaced. •S ynchronize the units: 1. B ring both the sensor and display indoors and remove power adapter/batteries from each. 2. Reinstall batteries in outdoor sensor. 3. Reinstall power adapter/batteries in display. 4. L et the units sit within a couple feet of each other for about 20 minutes to gain a strong connection. Flashing of the outdoor temperature may be an indication of wireless interference. •M ake sure the A-B-C switch in the battery compartments of both the display and sensor are switched to the same letter. You may choose A, B or C; but both units must match to sync up. Sometimes changing to a different channel can help. •W eather Forecast icon predicts conditions for the next 12 to 24 hours, not current conditions. •H as Learning Mode icon disappeared from the display? Learning Mode must complete before forecast and pressure will be accurate. •A llow unit to run continuously for 33 days. Battery removal or resetting the display will restart Learning Mode. After 14 days, forecast should be fairly accurate, however Learning Mode calibrates for a total of 33 days. 17 Troubleshooting Problem Possible Solution Inaccurate temperature or humidity •M ake sure both the display and 5-in-1 sensor are placed away from any heat sources or vents (see page 10). •M ake sure both units are positioned away from moisture sources (see page 10). •M ake sure 5-in-1 sensor is mounted at least 5 ft off of the ground. • Calibrate indoor and outdoor temperature and humidity (see page 20). No rainfall • Check to ensure the rain gauge stabilizer (plastic tab) has been removed from the bottom of the sensor (see page 7). • Clear debris, such as leaves, out of the rain collector funnel and debris screen. • Calibrate the Rain Gauge (see page 21). Inaccurate wind readings •W hat is wind reading being compared to? Pro weather stations are typically mounted at 30 ft high or more. Make sure to compare data using a sensor positioned at the same mounting height. •C heck location of the sensor. Ensure it’s mounted a minimum of 5 ft in the air with no obstructions around it (within several feet). •Ensure wind cups are spinning freely. If they hesitate or stop try lubricating with graphite powder or spray lubricant. USB connectivity not working • Verify display is in USB mode 3 (see page 12). • Unplug USB from display to computer for 20 seconds, and then plug back in. •C heck that the batteries are installed correctly. Batteries may need to be replaced. •R eset the display by pressing the reset button, located on the back of the display. Date and time will need to be entered after a reset. Display screen not working •C heck that the batteries are installed correctly. Batteries may need to be replaced. •R eset the display by pressing and holding the CLEAR RECORD / RESET button, located in the battery compartment of the display. Date and time will need to be entered after a reset. 18 Troubleshooting Problem Possible Solution Cannot access data online or on the AcuRite mobile app • Verify that the status in the AcuRite Connect software on your PC indicates that the data was received from the display. • Verify that you created an account or logged into an existing account at www.acu-link.com or in the AcuRite app, and that you input the display’s MAC ADDRESS as a part of the account setup. • Verify that the USB cable is securely connected to the display AND to the PC. • Verify the display is powered ON. • Verify the PC that the display is connected to is always powered ON. • Verify the PC that the display is connected to is always connected to the internet. • Verify that the display’s MAC ADDRESS is entered correctly in the AcuRite Connect software on your PC. • Verify that “Streaming” is enabled in the AcuRite Connect software on your PC. • Verify that the display’s MAC ADDRESS is correct in your www.acu-link.com and/or AcuRite mobile app account. • Follow on-screen troubleshooting steps in the AcuRite Connect software. AcuRite Connect data is different than the display’s data • Check that the USB cable is securely connected to the display AND to the PC’s USB port. • Restart the AcuRite Connect software. • Unplug the USB cable from the PC for 1 minute, and then plug the cable back in again. • USB port may be malfunctioning, try using another USB port. If your AcuRite product does not operate properly after trying the troubleshooting steps, visit www.AcuRite.com or call (844) 228-5465 for assistance. 19 Care & Maintenance Display Care Clean with a soft, damp cloth. Do not use caustic cleaners or abrasives. Keep away from dust, dirt and moisture. Clean ventilation ports regularly with a gentle puff of air. 5-in-1 Sensor Care Clean the Sensor Clean with a soft damp cloth. Do not use caustic cleaners or abrasives that will mar the polished surfaces of the rain collection funnel or the solar cell. Scratches will result in decreased performance and reliability. Insect Prevention Insects may cause obstructions and interrupt data by nesting in or on the 5-in-1 sensor. To limit this problem, spray sensor with a household insect repellent. Consult the insect repellent instructions prior to use. Snow & Freezing Weather The 5-in-1 sensor will not be damaged by freezing conditions. NOTE: If the rain collector cup fills with snow and then melts, it will register as rain on the display. Clean the Rain Collector Cup Remove and empty rain collector debris filter. The debris filter is located in the rain collector funnel. Remove from the top by gently squeezing and pulling out. Clean the Wind Vane & Anemometer Remove foreign matter from the outside of the sensor for free movement of the wind vane and anemometer. If needed, use a small amount of spray lubricant, clear silicone or graphite powder on the anemometer for improved movement. Calibration The indoor / outdoor temperature and humidity readings, and barometric pressure can be calibrated on the display to improve accuracy. Calibration can improve accuracy when sensor placement or environmental factors impact the data accuracy. 1. To access calibration mode, press AND HOLD the “ T”, “SE T ”, and “V ” buttons simultaneously for at least 5 seconds. 2. To adjust the currently selected (flashing) item, press and release the “ T“ or “V “ buttons to calibrate the data value higher or lower from the actual reading. 3. To save your adjustments, press and release the “ SE T ” button to adjust the next preference. The “ ” icon will remain illuminated next to calibrated values. The preference set order is as follows: OUTDOOR TEMPERATURE OUTDOOR HUMIDITY 20 INDOOR TEMPERATURE INDOOR HUMIDITY BAROMETRIC PRESSURE (must be set to MANUAL mode to calibrate)* *To change from AUTO to MANUAL pressure mode and vice versa, press AND HOLD the “SET” button for at least 10 seconds. The display indicates the current pressure mode selected,“AUTO” or “MANUAL”. After 20 seconds of inactivity, the display will save the adjustments and exit calibration mode. Note: Calibrations will be erased if the display is reset or if batteries are removed and the power adapter is unplugged. Calibrate the Rain Gauge The rain gauge on the 5-in-1 sensor can be calibrated to improve accuracy. Items Needed: 5-in-1 sensor, display, plastic cup, pin, screw driver 1. First, ensure 5-in-1 sensor is perfectly level using built-in bubble level. 2. Place display close so you can monitor it during calibration. 3. Make a pin hole in the bottom of a plastic cup. Hold the cup over the rain gauge and fill it with exactly 1 cup (8oz) of water, allowing the water to drip into the rain gauge. You should hear the internal buckets tip and see water drain through the rain gauge. 4. A few seconds after each bucket tip, the display displays rainfall in approximately 0.01” or more increments. 5. The cup of water should take more than 20 minutes to empty; a quicker period will result in inaccurate calibration. Try to simulate a normal steady rainfall. When cup is empty of water, display should register 1.06” Tips • There should be nearly an equal number of water drops (about 25 water drops) between bucket tips. If not, adjust the calibration screws on the bottom of the 5-in-1 sensor until an equal number of water drops are tipping the buckets. Then, restart the calibration procedure. • If you don’t hear the buckets tipping and see water dripping alternately out of each drain, there may be an issue with the rain gauge or it’s adjustment. See Troubleshooting on page 18. Adjustment If the rain gauge doesn’t register close to 1.06”, make an EQUAL adjustment to the two calibration screws on the bottom of the 5-in-1 sensor. Turning screws clockwise increases rainfall; counter clockwise decreases rainfall. • To adjust the rainfall reading by 2% turn both screws 1/8 of a turn. • To adjust the rainfall reading by 4% turn both screws 1/4 of a turn. • To adjust the rainfall reading by 8% turn both screws 1/2 of a turn. Watch the video at www.acurite.com/5n1 21 Specifications TEMPERATURE RANGE Outdoor: -40ºF to 158ºF; -40ºC to 70ºC Indoor: 32ºF to 122ºF; 0ºC to 50ºC HUMIDITY RANGE Outdoor: 1% to 99% Indoor: 1% to 99% WIND SPEED 0 to 99 mph; 0 to 159 kph WIND DIRECTION INDICATORS 16 points RAINFALL 0 to 393 in; 0 to 9999mm WIRELESS RANGE 330ft / 100m depending on home construction materials OPERATING FREQUENCY 433 MHz POWER Display: 4.5VDC, 250mA adapter 3 x AA alkaline batteries (optional) Sensor: 4 x AA alkaline or lithium batteries DATA REPORTING Wind Speed: 18 second updates; Direction: 30 seconds Outdoor temperature & humidity: 36 second updates Indoor temperature & humidity: 60 second updates AcuRite Connect CSV Data Logging: 12 minute intervals AcuRite Connect to Online Portal/App: 18 seconds DISPLAY MEMORY 512 kilobytes (not expandable) ACURITE CONNECT SYSTEM Windows 8.1 / 8 / 7; 1 GHz or faster processor; 1 GB REQUIREMENTS (32-bit) or 2 GB (64-bit) of RAM; available USB port High speed internet is required for some advanced features of AcuRite Connect, such as the AcuRite software and app. FCC Information This device complies with part 15 of FCC rules. Operation is subject to the following two conditions: 1- This device may NOT cause harmful interference, and 2- This device must accept any interference received, including interference that may cause undesired operation. This equipment has been tested and found to comply with the limits for a Class B digital device, pursuant to Part 15 of the FCC rules. These limits are designed to provide reasonable protection against harmful interference in a residential installation. This equipment generates, uses and can radiate radio frequency energy and, if not installed and used in accordance with the instructions, may cause harmful interference to radio communications. However, there is no guarantee that interference will not occur in a particular installation. If this equipment does cause harmful interference to radio or television reception, which can be determined by turning the equipment off and on, the user is encouraged to try to correct the interference by one or more of the following measures: • Reorient or relocate the receiving antenna. • Increase the separation between the equipment and the receiver. • Connect the equipment into an outlet on a circuit different from that to which the receiver is connected. • Consult the dealer or an experienced radio/TV technician for help. NOTE: The manufacturer is not responsible for any radio or TV interference caused by unauthorized modifications to this equipment. Such modifications could void the user authority to operate the equipment. This device complies with Industry Canada licence-exempt RSS standard(s). Operation is subject to the following two conditions: (1) This device may not cause interference, and (2) This device must accept any interference received, including interference that may cause undesired operation of the device. 22 Customer Support AcuRite customer support is committed to providing you with best-inclass service. For assistance, please have the model number of this product available and contact us in any of the following ways: (844) 228-5465 [email protected] 24/7 support at www.AcuRite.com ► Installation Videos ► Register your Product ► Instruction Manuals ► Support User Forum ► Replacement Parts ► Submit IMPORTANT Feedback & Ideas PRODUCT MUST BE REGISTERED TO RECEIVE WARRANTY SERVICE PRODUCT REGISTRATION Register online to receive 1 year warranty protection www.AcuRite.com ►Register a Product Limited One Year Warranty At AcuRite, we proudly uphold our commitment to quality technology. Chaney Instrument Co. warrants that all products it manufactures to be of good material and workmanship, and to be free of defects when properly installed and operated for a period of one year from the date of purchase. not be breached, and Chaney will give no credit for products it manufactures which have received normal wear and tear, been damaged (including by acts of nature), tampered, abused, improperly installed, damaged in shipping, or repaired or altered by others than authorized representatives of Chaney. The above-described warranty is expressly in lieu of all other warranties, express or implied, and all other warranties are hereby expressly disclaimed, including without limitation the implied warranty of merchantability and the implied warranty of fitness for a particular purpose. Chaney expressly disclaims all liability for special, consequential or incidental damages, whether arising in tort or by contract from any breach of this warranty. Some states do not allow the exclusion or limitation of incidental or consequential damages, so the above limitation or exclusion may not apply to you. Chaney further disclaims all liability from personal injury relating to its products to the extent permitted by law. By acceptance of any of Chaney’s products, the purchaser assumes all liability for the consequences arising from their use or misuse. No person, firm or corporation is authorized to assume for Chaney any other liability in connection with the sale of its products. Furthermore, no person, firm or corporation is authorized to modify or waive the terms of this paragraph, and the preceding paragraph, unless done in writing and signed by a duly authorized agent of Chaney. This warranty gives you specific legal rights, and you may also have other rights which vary from state to state. We recommend that you visit us at www.AcuRite.com for the fastest way to register your product. However, product registration does not eliminate the need to retain your original proof of purchase in order to obtain warranty benefits. Chaney Instrument Co. warrants that all products it manufactures to be of good material and workmanship, and to be free of defects when properly installed and operated for a period of one year from the date of purchase. Remedy for breach of this warranty is limited to repair or replacement of the defective item(s). Any product which, under normal use and service, is proven to breach the warranty contained herein within ONE YEAR from date of sale will, upon examination by Chaney, and at its sole option, be repaired or replaced by Chaney. Transportation costs and charges for returned goods shall be paid for by the purchaser. Chaney hereby disclaims all responsibility for such transportation costs and charges. This warranty will For in-warranty claims: Chaney Instrument Co. | 965 Wells St. | Lake Geneva, WI 53147 23 Weather Stations Temperature & Humidity Weather Alert Radio Kitchen Thermometers & Timers Clocks It’s More than Accurate, it’s AcuRite offers an extensive assortment of precision instruments, designed to provide you with information you can depend on to Plan your day with confidence™. www.AcuRite.com Printed in China 02032C2 INST 090315 ©Chaney Instrument Co. All rights reserved. AcuRite is a registered trademark of the Chaney Instrument Co., Lake Geneva, WI 53147. All other trademarks and copyrights are the property of their respective owners. AcuRite uses patented technology. Visit www.AcuRite.com/patents for details.